-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 12 23:26:47 2024 UTC
-- File 20240917_202935_1229.dat, 8822  packets found.
APID:
      1229 (EW) in 8822 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 8822 packets (all packets)
Total number of packets: 
      36720 in 8822 packets (all packets)
Packet No from 1 to 8822
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 319.029238 to 323.749200
-each separately from 319s to 323s and from 29238us to 749200us.
   Differences in neigboring packets: 
      0.000000 in 8820 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.719962 in 1 packets: 2448
MIU packet time from 2024-09-17T20:34:57.029.238.000 to 2024-09-17T20:35:01.749.200.000
   Differences in neigboring packets: 
      0.000000 in 8820 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.719962 in 1 packets: 2448
MIU orbit number: 
      60000 in 8822 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 8822 packets (all packets)
MIU orbit information / number of events: 
      0 in 8822 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 8822 packets (all packets)
Instrument anomalies: 
      0 (OK) in 8822 packets (all packets)
FPGA configuration version: 
      211 in 8822 packets (all packets)
10 MHz counter at the last 1Hz pulse from 320.6685392 to 324.6685252
   Differences in neigboring packets: 
      0.0000000 in 8820 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3.9999860 in 1 packets: 2448
Temperature FPGA: 
      02 -> 52.4 degC in 8822 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 8822 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 8822 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 8822 packets (all packets)
Number of external events: 
      2 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 6374 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
Number of generated alerts: 
      2 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 6374 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
Survey memory error: 
      0 in 8822 packets (all packets)
Event memory error: 
      0 in 8822 packets (all packets)
Configuration error: 
      0 in 8822 packets (all packets)
SDRAM CRC error: 
      0 in 8822 packets (all packets)
Radio gain: 
      0 in 8822 packets (all packets)
Radio bandwidth: 
      0 in 8822 packets (all packets)
Digital gain: 
      0 in 8822 packets (all packets)
Analog gain: 
      0 in 8822 packets (all packets)

EVENT DATA READER
-- File 20240917_202935_1229.dat, 43 EW pages found. Extra packets: 50.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 3
      2 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 6324 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 320.8087539 to 325.4697289
      Difference between the neighboring packets: 
      0.0000170 in 8770 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.5118850 in 1 packets: 2448
   Memory page number from 0 to 45
      Difference between the neighboring packets: 
      0 in 8729 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 41 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
      4 in 1 packets: 2448
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 8729 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 41 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
      -59 in 1 packets: 2448
   MIU event flag 
      1 in 8772 packets (all packets)
   Time triggered event flag 
      0 in 8772 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-09-17T20:34:56.973.749.631
   Min EW header abs. time found among all events            : 2024-09-17T20:34:57.029.238.000
   Max EW event (1st sample) abs. time found among all events: 2024-09-17T20:35:01.693.711.631
   Max EW header abs. time found among all events            : 2024-09-17T20:35:01.749.200.000
   TM values from -229 to 221 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 2448
    Page no. 49:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 12  ordered packets: 2448
   TM values from -229 to 221 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=319 C1MHz=0.029238
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 2447 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 320.8087539
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-09-17T20:34:57.029.238.000
   Event 1st sample abs. time : 2024-09-17T20:34:56.973.749.631)
   --- Rel. time correction for 1 points based on 314 XO calibration points at 1Hz counter from 0 to 313
   --- First point: 1Hz counter = 319, 32bit 10MHz counter = 320.808753900 corrected by 40bit 10MHz counter of 314.668560400 obtained at 1Hz counter of 313. Result: 319.140214991
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             319.140214816
   Event data descriptor XO time (reconstructed seconds from XP start)    :             319.140214991
   Sampling frequency correction factor for this event:     0.999996500

 Event no. 3 total number of packets: 6324
    Page no. 1:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 31  ordered packets: 6324
   TM values from -226 to 207 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=323 C1MHz=0.749200
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 6323 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 325.3622379
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-09-17T20:35:01.749.200.000
   Event 1st sample abs. time : 2024-09-17T20:35:01.693.711.631)
   --- Rel. time correction for 1 points based on 314 XO calibration points at 1Hz counter from 0 to 313
   --- First point: 1Hz counter = 323, 32bit 10MHz counter = 325.362237900 corrected by 40bit 10MHz counter of 314.668560400 obtained at 1Hz counter of 313. Result: 323.693714928
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             323.693714753
   Event data descriptor XO time (reconstructed seconds from XP start)    :             323.693714928
   Sampling frequency correction factor for this event:     0.999996500
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2024-09-17T20:34:56.973.749.631 to 2024-09-17T20:34:57.015.365.776 (XO: 319.140214816 - 319.181830961)
   Min TM value:-229 at an offset of 6603.648 usec -> 2024-09-17T20:34:56.980.353.279 (XO: 319.146818464)
   Max TM value:221 at an offset of 6591.561 usec -> 2024-09-17T20:34:56.980.341.191 (XO: 319.146806376)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 6591.448 usec -> 2024-09-17T20:34:56.980.341.079 (XO: 319.146806264)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 6626.898 usec -> 2024-09-17T20:34:56.980.376.529 (XO: 319.146841714)
   Med: -4.00000   Ave: -4.4286578   Std: 16.267391

 - Event number: 3 from 2024-09-17T20:35:01.693.711.631 to 2024-09-17T20:35:01.801.220.007 (XO: 323.693714753 - 323.801223129)
   Min TM value:-226 at an offset of 43474.415 usec -> 2024-09-17T20:35:01.737.186.045 (XO: 323.737189168)
   Max TM value:207 at an offset of 13431.522 usec -> 2024-09-17T20:35:01.707.143.153 (XO: 323.707146275)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 53464.437 usec -> 2024-09-17T20:35:01.747.176.068 (XO: 323.747179190)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 13431.647 usec -> 2024-09-17T20:35:01.707.143.278 (XO: 323.707146400)
   Med: -4.00000   Ave: -4.4346895   Std: 11.297707
