-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 12 10:13:27 2024 UTC
-- File 20240917_170329_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.061458 to 87.366539
-each separately from 2s to 87s and from 5025us to 813540us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.620512 in 1 packets: 18
      23.130703 in 1 packets: 36
      17.569886 in 1 packets: 54
      8.621310 in 1 packets: 72
      21.360936 in 1 packets: 90
MIU packet time from 2024-09-17T17:03:34.061.458.000 to 2024-09-17T17:04:59.366.539.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.620512 in 1 packets: 18
      23.130703 in 1 packets: 36
      17.569886 in 1 packets: 54
      8.621310 in 1 packets: 72
      21.360936 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7527084 to 88.7524124
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999513 in 1 packets: 18
      22.9999199 in 1 packets: 36
      17.9999374 in 1 packets: 54
      8.9999687 in 1 packets: 72
      20.9999267 in 1 packets: 90
Temperature FPGA: 
      03 -> 51.1 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C2 -> 38.8 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240917_170329_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8141680 to 3.8144570
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6204630  -- OK for SL --
   MIU time from 2024-09-17T17:03:34.061.458.000 to 2024-09-17T17:03:34.061.747.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.620512

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4349200 to 18.4352090
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1306230  -- OK for SL --
   MIU time from 2024-09-17T17:03:48.682.259.000 to 2024-09-17T17:03:48.682.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.130703

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.5658320 to 41.5661210
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5698230  -- OK for SL --
   MIU time from 2024-09-17T17:04:11.813.251.000 to 2024-09-17T17:04:11.813.540.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.569886

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      171 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1359440 to 59.1362330
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6212790  -- OK for SL --
   MIU time from 2024-09-17T17:04:29.383.426.000 to 2024-09-17T17:04:29.383.715.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.621310

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.7575120 to 67.7578010
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3608630  -- OK for SL --
   MIU time from 2024-09-17T17:04:38.005.025.000 to 2024-09-17T17:04:38.005.314.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.360936

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.1186640 to 89.1189530
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-17T17:04:59.366.250.000 to 2024-09-17T17:04:59.366.539.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.814168000 corrected by 40bit 10MHz counter of 3.752708400 obtained at 1Hz counter of 2. Result: 2.061459815
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-09-17T17:03:34.061.317.725
   Max 1st sample MIU time among all SL packets: 2024-09-17T17:04:59.366.398.725
   Min 1st sample XO time among all SL packets: 2.061319540
   Max 1st sample XO time among all SL packets: 87.366401571
   TM values from -230 to 237 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-09-17T17:03:34.061.317.725 (XO: 2.061319540 - 2.061625541, t=38.8 degC)  Rank: 00067
   TM values: Max =   237  Min =  -209  Ave =    -4.46  Std =    25.61
   Min TM value:-209 at an offset of 135.200 usec -> 2024-09-17T17:03:34.061.452.925 (XO: 2.061454740)
   Max TM value:237 at an offset of 134.575 usec -> 2024-09-17T17:03:34.061.452.300 (XO: 2.061454115)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 134.463 usec -> 2024-09-17T17:03:34.061.452.187 (XO: 2.061454003)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 134.688 usec -> 2024-09-17T17:03:34.061.452.412 (XO: 2.061454228)

 - Block number: 2 starting at 2024-09-17T17:03:48.682.118.725 (XO: 16.682122412 - 16.682428413, t=38.8 degC)  Rank: 00077
   TM values: Max =   205  Min =  -186  Ave =    -4.50  Std =    26.26
   Min TM value:-186 at an offset of 117.925 usec -> 2024-09-17T17:03:48.682.236.650 (XO: 16.682240338)
   Max TM value:205 at an offset of 117.363 usec -> 2024-09-17T17:03:48.682.236.087 (XO: 16.682239775)
   Max dTM/dt [1/us]:1.32e+03 at an offset of 117.200 usec -> 2024-09-17T17:03:48.682.235.925 (XO: 16.682239613)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 117.475 usec -> 2024-09-17T17:03:48.682.236.200 (XO: 16.682239888)

 - Block number: 3 starting at 2024-09-17T17:04:11.813.110.725 (XO: 39.813114971 - 39.813420972, t=38.8 degC)  Rank: 00071
   TM values: Max =   208  Min =  -230  Ave =    -4.59  Std =    26.16
   Min TM value:-230 at an offset of 118.225 usec -> 2024-09-17T17:04:11.813.228.950 (XO: 39.813233196)
   Max TM value:208 at an offset of 117.663 usec -> 2024-09-17T17:04:11.813.228.387 (XO: 39.813232634)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 120.313 usec -> 2024-09-17T17:04:11.813.231.037 (XO: 39.813235284)
   Min dTM/dt [1/us]:-1.42e+03 at an offset of 117.775 usec -> 2024-09-17T17:04:11.813.228.500 (XO: 39.813232746)

 - Block number: 4 starting at 2024-09-17T17:04:29.383.285.725 (XO: 57.383288067 - 57.383594068, t=38.8 degC)  Rank: 00171
   TM values: Max =   208  Min =  -177  Ave =    -4.58  Std =    35.27
   Min TM value:-177 at an offset of 129.013 usec -> 2024-09-17T17:04:29.383.414.737 (XO: 57.383417079)
   Max TM value:208 at an offset of 124.325 usec -> 2024-09-17T17:04:29.383.410.050 (XO: 57.383412392)
   Max dTM/dt [1/us]:5.37e+02 at an offset of 121.550 usec -> 2024-09-17T17:04:29.383.407.275 (XO: 57.383409617)
   Min dTM/dt [1/us]:-6.13e+02 at an offset of 202.251 usec -> 2024-09-17T17:04:29.383.487.975 (XO: 57.383490317)

 - Block number: 5 starting at 2024-09-17T17:04:38.004.884.725 (XO: 66.004886042 - 66.005192043, t=38.8 degC)  Rank: 00076
   TM values: Max =   196  Min =  -188  Ave =    -4.48  Std =    18.88
   Min TM value:-188 at an offset of 129.263 usec -> 2024-09-17T17:04:38.005.013.987 (XO: 66.005015305)
   Max TM value:196 at an offset of 128.688 usec -> 2024-09-17T17:04:38.005.013.412 (XO: 66.005014730)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 128.563 usec -> 2024-09-17T17:04:38.005.013.287 (XO: 66.005014605)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 128.800 usec -> 2024-09-17T17:04:38.005.013.525 (XO: 66.005014843)

 - Block number: 6 starting at 2024-09-17T17:04:59.366.109.725 (XO: 87.366112570 - 87.366418571, t=38.8 degC)  Rank: 00078
   TM values: Max =   203  Min =  -212  Ave =    -4.50  Std =    22.44
   Min TM value:-212 at an offset of 125.350 usec -> 2024-09-17T17:04:59.366.235.075 (XO: 87.366237920)
   Max TM value:203 at an offset of 127.488 usec -> 2024-09-17T17:04:59.366.237.212 (XO: 87.366240058)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 127.388 usec -> 2024-09-17T17:04:59.366.237.112 (XO: 87.366239958)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 127.638 usec -> 2024-09-17T17:04:59.366.237.362 (XO: 87.366240208)
