-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 12 00:04:58 2024 UTC
-- File 20240908_162922_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.709481 to 75.141421
-each separately from 0s to 75s and from 40636us to 709770us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.330866 in 1 packets: 18
      18.526333 in 1 packets: 36
      10.851205 in 1 packets: 54
      12.288137 in 1 packets: 72
      9.433665 in 1 packets: 90
MIU packet time from 2024-09-08T16:29:25.709.481.000 to 2024-09-08T16:30:40.141.421.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.330866 in 1 packets: 18
      18.526333 in 1 packets: 36
      10.851205 in 1 packets: 54
      12.288137 in 1 packets: 72
      9.433665 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7667112 to 76.7664521
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999171 in 1 packets: 18
      17.9999378 in 1 packets: 36
      10.9999620 in 1 packets: 54
      11.9999586 in 1 packets: 72
      9.9999654 in 1 packets: 90
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 1 packets: 73
      3 in 5 packets: 74, 75, 76, 77, 78
      4 in 30 packets: 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, 92, 93, 94, 95, 96, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240908_162922_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      115 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4761934 to 2.4764824
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3307830  -- OK for SL --
   MIU time from 2024-09-08T16:29:25.709.481.000 to 2024-09-08T16:29:25.709.770.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.330866

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8072654 to 25.8075544
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5262710  -- OK for SL --
   MIU time from 2024-09-08T16:29:49.040.636.000 to 2024-09-08T16:29:49.040.925.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.526333

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.3338254 to 44.3341144
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8511670  -- OK for SL --
   MIU time from 2024-09-08T16:30:07.567.258.000 to 2024-09-08T16:30:07.567.547.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.851205

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.1852814 to 55.1855704
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2880950  -- OK for SL --
   MIU time from 2024-09-08T16:30:18.418.752.000 to 2024-09-08T16:30:18.419.041.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.288137

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      215 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.4736654 to 67.4739544
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4336310  -- OK for SL --
   MIU time from 2024-09-08T16:30:30.707.178.000 to 2024-09-08T16:30:30.707.467.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.433665

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.9075854 to 76.9078744
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-08T16:30:40.141.132.000 to 2024-09-08T16:30:40.141.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.476193400 corrected by 40bit 10MHz counter of 1.766711200 obtained at 1Hz counter of 0. Result: 0.709484612
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-09-08T16:29:25.709.340.725
   Max 1st sample MIU time among all SL packets: 2024-09-08T16:30:40.141.280.725
   Min 1st sample XO time among all SL packets: 0.709344337
   Max 1st sample XO time among all SL packets: 75.141282519
   TM values from -618 to 554 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-09-08T16:29:25.709.340.725 (XO: 0.709344337 - 0.709650338, t=38.8 degC)  Rank: 00115
   TM values: Max =   554  Min =  -314  Ave =    -4.35  Std =    64.60
   Min TM value:-314 at an offset of 192.813 usec -> 2024-09-08T16:29:25.709.533.538 (XO: 0.709537150)
   Max TM value:554 at an offset of 92.975 usec -> 2024-09-08T16:29:25.709.433.700 (XO: 0.709437312)
   Max dTM/dt [1/us]:3.11e+03 at an offset of 182.938 usec -> 2024-09-08T16:29:25.709.523.663 (XO: 0.709527275)
   Min dTM/dt [1/us]:-3.01e+03 at an offset of 173.426 usec -> 2024-09-08T16:29:25.709.514.150 (XO: 0.709517762)

 - Block number: 2 starting at 2024-09-08T16:29:49.040.495.725 (XO: 24.040496963 - 24.040802964, t=38.8 degC)  Rank: 00073
   TM values: Max =   229  Min =  -230  Ave =    -4.35  Std =    25.13
   Min TM value:-230 at an offset of 124.613 usec -> 2024-09-08T16:29:49.040.620.337 (XO: 24.040621576)
   Max TM value:229 at an offset of 126.688 usec -> 2024-09-08T16:29:49.040.622.412 (XO: 24.040623651)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 126.563 usec -> 2024-09-08T16:29:49.040.622.287 (XO: 24.040623526)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 126.813 usec -> 2024-09-08T16:29:49.040.622.537 (XO: 24.040623776)

 - Block number: 3 starting at 2024-09-08T16:30:07.567.117.725 (XO: 42.567121010 - 42.567427011, t=38.8 degC)  Rank: 00071
   TM values: Max =   227  Min =  -202  Ave =    -4.40  Std =    20.78
   Min TM value:-202 at an offset of 92.825 usec -> 2024-09-08T16:30:07.567.210.550 (XO: 42.567213835)
   Max TM value:227 at an offset of 94.913 usec -> 2024-09-08T16:30:07.567.212.637 (XO: 42.567215923)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 94.813 usec -> 2024-09-08T16:30:07.567.212.537 (XO: 42.567215823)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 95.025 usec -> 2024-09-08T16:30:07.567.212.750 (XO: 42.567216035)

 - Block number: 4 starting at 2024-09-08T16:30:18.418.611.725 (XO: 53.418614490 - 53.418920491, t=38.8 degC)  Rank: 00071
   TM values: Max =   214  Min =  -206  Ave =    -4.40  Std =    19.43
   Min TM value:-206 at an offset of 86.875 usec -> 2024-09-08T16:30:18.418.698.600 (XO: 53.418701365)
   Max TM value:214 at an offset of 89.000 usec -> 2024-09-08T16:30:18.418.700.725 (XO: 53.418703490)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 88.888 usec -> 2024-09-08T16:30:18.418.700.612 (XO: 53.418703378)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 89.100 usec -> 2024-09-08T16:30:18.418.700.825 (XO: 53.418703590)

 - Block number: 5 starting at 2024-09-08T16:30:30.707.037.725 (XO: 65.707040900 - 65.707346901, t=38.8 degC)  Rank: 00215
   TM values: Max =   473  Min =  -618  Ave =    -4.23  Std =    93.27
   Min TM value:-618 at an offset of 149.138 usec -> 2024-09-08T16:30:30.707.186.863 (XO: 65.707190038)
   Max TM value:473 at an offset of 110.175 usec -> 2024-09-08T16:30:30.707.147.900 (XO: 65.707151075)
   Max dTM/dt [1/us]:4.00e+03 at an offset of 148.101 usec -> 2024-09-08T16:30:30.707.185.825 (XO: 65.707189000)
   Min dTM/dt [1/us]:-2.87e+03 at an offset of 148.376 usec -> 2024-09-08T16:30:30.707.186.100 (XO: 65.707189275)

 - Block number: 6 starting at 2024-09-08T16:30:40.140.991.725 (XO: 75.140993518 - 75.141299520, t=38.8 degC)  Rank: 00075
   TM values: Max =   198  Min =  -200  Ave =    -4.31  Std =    18.66
   Min TM value:-200 at an offset of 129.000 usec -> 2024-09-08T16:30:40.141.120.725 (XO: 75.141122519)
   Max TM value:198 at an offset of 131.175 usec -> 2024-09-08T16:30:40.141.122.900 (XO: 75.141124694)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 131.025 usec -> 2024-09-08T16:30:40.141.122.750 (XO: 75.141124544)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 131.225 usec -> 2024-09-08T16:30:40.141.122.950 (XO: 75.141124744)
