-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 11 23:59:56 2024 UTC
-- File 20240908_162140_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.295582 to 15.072811
-each separately from 7s to 15s and from 72522us to 295871us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.776651 in 1 packets: 18
MIU packet time from 2024-09-08T16:21:50.295.582.000 to 2024-09-08T16:21:58.072.811.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.776651 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7938948 to 16.7938673
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999725 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240908_162140_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.0894784 to 9.0897674
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7766230  -- OK for SL --
   MIU time from 2024-09-08T16:21:50.295.582.000 to 2024-09-08T16:21:50.295.871.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.776651

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8663904 to 16.8666794
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-08T16:21:58.072.522.000 to 2024-09-08T16:21:58.072.811.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.089478400 corrected by 40bit 10MHz counter of 8.793894800 obtained at 1Hz counter of 7. Result: 7.295584605
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-09-08T16:21:50.295.441.725
   Max 1st sample MIU time among all SL packets: 2024-09-08T16:21:58.072.670.725
   Min 1st sample XO time among all SL packets: 7.295444330
   Max 1st sample XO time among all SL packets: 15.072672072
   TM values from -202 to 367 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-09-08T16:21:50.295.441.725 (XO: 7.295444330 - 7.295750331, t=38.8 degC)  Rank: 00077
   TM values: Max =   216  Min =  -202  Ave =    -4.32  Std =    19.48
   Min TM value:-202 at an offset of 114.250 usec -> 2024-09-08T16:21:50.295.555.975 (XO: 7.295558580)
   Max TM value:216 at an offset of 116.325 usec -> 2024-09-08T16:21:50.295.558.050 (XO: 7.295560655)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 116.213 usec -> 2024-09-08T16:21:50.295.557.937 (XO: 7.295560542)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 116.425 usec -> 2024-09-08T16:21:50.295.558.150 (XO: 7.295560755)

 - Block number: 2 starting at 2024-09-08T16:21:58.072.381.725 (XO: 15.072383071 - 15.072689072, t=38.8 degC)  Rank: 00100
   TM values: Max =   367  Min =  -167  Ave =    -4.41  Std =    21.59
   Min TM value:-167 at an offset of 103.450 usec -> 2024-09-08T16:21:58.072.485.175 (XO: 15.072486521)
   Max TM value:367 at an offset of 100.550 usec -> 2024-09-08T16:21:58.072.482.275 (XO: 15.072483621)
   Max dTM/dt [1/us]:2.58e+03 at an offset of 100.463 usec -> 2024-09-08T16:21:58.072.482.187 (XO: 15.072483534)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 94.075 usec -> 2024-09-08T16:21:58.072.475.800 (XO: 15.072477146)
