-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 11 07:49:32 2024 UTC
-- File 20240905_140344_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.771911 to 24.520633
-each separately from 5s to 24s and from 520344us to 772200us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.748144 in 1 packets: 18
MIU packet time from 2024-09-05T14:03:52.771.911.000 to 2024-09-05T14:04:11.520.633.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.748144 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.6643342 to 25.6642693
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999351 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240905_140344_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4362468 to 7.4365358
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7480790  -- OK for SL --
   MIU time from 2024-09-05T14:03:52.771.911.000 to 2024-09-05T14:03:52.772.200.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.748144

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1846148 to 26.1849038
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-05T14:04:11.520.344.000 to 2024-09-05T14:04:11.520.633.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.436246800 corrected by 40bit 10MHz counter of 6.664334200 obtained at 1Hz counter of 5. Result: 5.771915225
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-09-05T14:03:52.771.770.725
   Max 1st sample MIU time among all SL packets: 2024-09-05T14:04:11.520.492.725
   Min 1st sample XO time among all SL packets: 5.771774949
   Max 1st sample XO time among all SL packets: 24.520495995
   TM values from -217 to 219 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-09-05T14:03:52.771.770.725 (XO: 5.771774949 - 5.772080950, t=38.8 degC)  Rank: 00080
   TM values: Max =   197  Min =  -191  Ave =    -4.42  Std =    19.60
   Min TM value:-191 at an offset of 128.188 usec -> 2024-09-05T14:03:52.771.898.912 (XO: 5.771903137)
   Max TM value:197 at an offset of 130.363 usec -> 2024-09-05T14:03:52.771.901.087 (XO: 5.771905312)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 133.025 usec -> 2024-09-05T14:03:52.771.903.750 (XO: 5.771907974)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 130.463 usec -> 2024-09-05T14:03:52.771.901.187 (XO: 5.771905412)

 - Block number: 2 starting at 2024-09-05T14:04:11.520.203.725 (XO: 24.520206994 - 24.520512995, t=38.8 degC)  Rank: 00087
   TM values: Max =   219  Min =  -217  Ave =    -4.43  Std =    18.15
   Min TM value:-217 at an offset of 119.663 usec -> 2024-09-05T14:04:11.520.323.387 (XO: 24.520326657)
   Max TM value:219 at an offset of 119.363 usec -> 2024-09-05T14:04:11.520.323.087 (XO: 24.520326357)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 121.950 usec -> 2024-09-05T14:04:11.520.325.675 (XO: 24.520328944)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 119.463 usec -> 2024-09-05T14:04:11.520.323.187 (XO: 24.520326457)
