-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 11 05:22:21 2024 UTC
-- File 20240905_054258_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.585949 to 84.825210
-each separately from 6s to 84s and from 405558us to 825210us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.839586 in 1 packets: 18
      20.139764 in 1 packets: 36
      9.239646 in 1 packets: 54
      9.599457 in 1 packets: 72
      24.419074 in 1 packets: 90
MIU packet time from 2024-09-05T05:43:07.585.949.000 to 2024-09-05T05:44:25.825.210.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.839586 in 1 packets: 18
      20.139764 in 1 packets: 36
      9.239646 in 1 packets: 54
      9.599457 in 1 packets: 72
      24.419074 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.6949712 to 85.6947031
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999486 in 1 packets: 18
      19.9999312 in 1 packets: 36
      8.9999691 in 1 packets: 54
      9.9999657 in 1 packets: 72
      23.9999173 in 1 packets: 90
Temperature FPGA: 
      02 -> 52.4 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C2 -> 38.8 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240905_054258_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2809211 to 8.2812101
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8395350  -- OK for SL --
   MIU time from 2024-09-05T05:43:07.585.949.000 to 2024-09-05T05:43:07.586.238.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.839586

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1207451 to 23.1210341
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1396950  -- OK for SL --
   MIU time from 2024-09-05T05:43:22.425.824.000 to 2024-09-05T05:43:22.426.113.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.139764

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.2607291 to 43.2610181
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2396150  -- OK for SL --
   MIU time from 2024-09-05T05:43:42.565.877.000 to 2024-09-05T05:43:42.566.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.239646

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.5006331 to 52.5009221
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5994230  -- OK for SL --
   MIU time from 2024-09-05T05:43:51.805.812.000 to 2024-09-05T05:43:51.806.101.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.599457

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.1003451 to 62.1006341
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.4189910  -- OK for SL --
   MIU time from 2024-09-05T05:44:01.405.558.000 to 2024-09-05T05:44:01.405.847.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.419074

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.5196251 to 86.5199141
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-05T05:44:25.824.921.000 to 2024-09-05T05:44:25.825.210.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.280921100 corrected by 40bit 10MHz counter of 7.694971200 obtained at 1Hz counter of 6. Result: 6.585951892
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-09-05T05:43:07.585.808.725
   Max 1st sample MIU time among all SL packets: 2024-09-05T05:44:25.825.069.725
   Min 1st sample XO time among all SL packets: 6.585811617
   Max 1st sample XO time among all SL packets: 84.825073530
   TM values from -212 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-09-05T05:43:07.585.808.725 (XO: 6.585811617 - 6.586117618, t=38.8 degC)  Rank: 00073
   TM values: Max =   199  Min =  -185  Ave =    -4.41  Std =    18.83
   Min TM value:-185 at an offset of 111.250 usec -> 2024-09-05T05:43:07.585.919.975 (XO: 6.585922867)
   Max TM value:199 at an offset of 113.338 usec -> 2024-09-05T05:43:07.585.922.062 (XO: 6.585924955)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 113.225 usec -> 2024-09-05T05:43:07.585.921.950 (XO: 6.585924842)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 113.463 usec -> 2024-09-05T05:43:07.585.922.187 (XO: 6.585925080)

 - Block number: 2 starting at 2024-09-05T05:43:22.425.683.725 (XO: 21.425686472 - 21.425992473, t=38.8 degC)  Rank: 00070
   TM values: Max =   197  Min =  -204  Ave =    -4.54  Std =    20.95
   Min TM value:-204 at an offset of 109.375 usec -> 2024-09-05T05:43:22.425.793.100 (XO: 21.425795848)
   Max TM value:197 at an offset of 97.300 usec -> 2024-09-05T05:43:22.425.781.025 (XO: 21.425783773)
   Max dTM/dt [1/us]:1.29e+03 at an offset of 95.250 usec -> 2024-09-05T05:43:22.425.778.975 (XO: 21.425781723)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 97.438 usec -> 2024-09-05T05:43:22.425.781.162 (XO: 21.425783910)

 - Block number: 3 starting at 2024-09-05T05:43:42.565.736.725 (XO: 41.565739749 - 41.566045750, t=38.8 degC)  Rank: 00068
   TM values: Max =   204  Min =  -212  Ave =    -4.56  Std =    20.94
   Min TM value:-212 at an offset of 98.613 usec -> 2024-09-05T05:43:42.565.835.337 (XO: 41.565838361)
   Max TM value:204 at an offset of 109.625 usec -> 2024-09-05T05:43:42.565.846.350 (XO: 41.565849374)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 98.100 usec -> 2024-09-05T05:43:42.565.834.825 (XO: 41.565837849)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 109.750 usec -> 2024-09-05T05:43:42.565.846.475 (XO: 41.565849499)

 - Block number: 4 starting at 2024-09-05T05:43:51.805.671.725 (XO: 50.805675464 - 50.805981465, t=38.8 degC)  Rank: 00068
   TM values: Max =   192  Min =  -188  Ave =    -4.58  Std =    24.18
   Min TM value:-188 at an offset of 108.975 usec -> 2024-09-05T05:43:51.805.780.700 (XO: 50.805784440)
   Max TM value:192 at an offset of 111.013 usec -> 2024-09-05T05:43:51.805.782.737 (XO: 50.805786477)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 110.913 usec -> 2024-09-05T05:43:51.805.782.637 (XO: 50.805786377)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 111.125 usec -> 2024-09-05T05:43:51.805.782.850 (XO: 50.805786590)

 - Block number: 5 starting at 2024-09-05T05:44:01.405.417.725 (XO: 60.405420444 - 60.405726445, t=38.8 degC)  Rank: 00059
   TM values: Max =   202  Min =  -203  Ave =    -4.58  Std =    22.78
   Min TM value:-203 at an offset of 85.663 usec -> 2024-09-05T05:44:01.405.503.387 (XO: 60.405506107)
   Max TM value:202 at an offset of 85.313 usec -> 2024-09-05T05:44:01.405.503.037 (XO: 60.405505757)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 85.275 usec -> 2024-09-05T05:44:01.405.503.000 (XO: 60.405505719)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 85.438 usec -> 2024-09-05T05:44:01.405.503.162 (XO: 60.405505882)

 - Block number: 6 starting at 2024-09-05T05:44:25.824.780.725 (XO: 84.824784529 - 84.825090530, t=38.8 degC)  Rank: 00067
   TM values: Max =   195  Min =  -205  Ave =    -4.64  Std =    27.01
   Min TM value:-205 at an offset of 110.088 usec -> 2024-09-05T05:44:25.824.890.812 (XO: 84.824894617)
   Max TM value:195 at an offset of 112.163 usec -> 2024-09-05T05:44:25.824.892.887 (XO: 84.824896692)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 110.200 usec -> 2024-09-05T05:44:25.824.890.925 (XO: 84.824894730)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 112.288 usec -> 2024-09-05T05:44:25.824.893.012 (XO: 84.824896817)
