-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Oct  8 11:23:51 2024 UTC
-- File 20240902_202839_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 15.177027
-each separately from 0s to 15s and from 0us to 841253us.
   Differences in neigboring packets: 
      1.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2024-09-02T20:28:42.000.000.000 to 2024-09-02T20:28:57.177.027.000
   Differences in neigboring packets: 
      1.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.5707934 to 15.5707420
   Differences in neigboring packets: 
      0.9999966 in 9 packets: 1, 3, 4, 5, 8, 12, 15, 16, 17
      0.9999965 in 5 packets: 2, 6, 9, 11, 14
      0.0000000 in 2 packets: 7, 13
      0.9999967 in 1 packets: 10
   XO PPM: 
      -3.4 in 9 packets: 1, 3, 4, 5, 8, 12, 15, 16, 17
      -3.5 in 5 packets: 2, 6, 9, 11, 14
      -3.3 in 1 packets: 10
Temperature FPGA: 
      02 -> 52.4 degC in 13 packets: 1, 2, 3, 7, 8, 9, 10, 11, 12, 15, 16, 17, 18
      04 -> 49.8 degC in 1 packets: 4
      03 -> 51.1 degC in 4 packets: 5, 6, 13, 14
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 3 packets: 1, 2, 3
      1 in 15 packets: 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 3 packets: 1, 2, 3
      1 in 15 packets: 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20240902_202839_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5707900 to 15.7477700
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2024-09-02T20:28:42.000.000.000 to 2024-09-02T20:28:57.177.027.000
      Difference between neigboring packets: 
      1.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 15 XO calibration points at 1Hz counter from 0 to 14
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.570790000 corrected by 40bit 10MHz counter of 0.570793400 obtained at 1Hz counter of 0. Result: 1.000000000
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2024-09-02T20:28:41.999.999.925
   Max 1st sample MIU time among all SR packets: 2024-09-02T20:28:57.177.026.925
   Min 1st sample XO time among all SR packets: 0.999999925
   Max 1st sample XO time among all SR packets: 15.177028527
   TM values from -144 to 177 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2024-09-02T20:28:41.999.999.925 (XO: 0.999999925 - 1.000016925, t=38.8 degC)
   TM values: Max =    62  Min =   -71  Ave =    -4.25  Std =    27.20

 - Packet number: 2 starting at 2024-09-02T20:28:43.833.938.925 (XO: 1.833942844 - 1.833959844, t=38.8 degC)
   TM values: Max =    50  Min =   -66  Ave =    -4.71  Std =    22.71

 - Packet number: 3 starting at 2024-09-02T20:28:44.667.881.925 (XO: 2.667885696 - 2.667902696, t=38.8 degC)
   TM values: Max =    59  Min =   -66  Ave =    -4.57  Std =    22.14

 - Packet number: 4 starting at 2024-09-02T20:28:45.501.825.925 (XO: 3.501828531 - 3.501845531, t=38.8 degC)
   TM values: Max =    75  Min =   -87  Ave =    -4.61  Std =    32.97

 - Packet number: 5 starting at 2024-09-02T20:28:46.335.768.925 (XO: 4.335771367 - 4.335788367, t=38.8 degC)
   TM values: Max =   100  Min =  -108  Ave =    -4.59  Std =    38.56

 - Packet number: 6 starting at 2024-09-02T20:28:47.169.712.925 (XO: 5.169714219 - 5.169731219, t=38.8 degC)
   TM values: Max =    59  Min =   -72  Ave =    -4.44  Std =    23.37

 - Packet number: 7 starting at 2024-09-02T20:28:48.003.655.925 (XO: 6.003657137 - 6.003674137, t=38.8 degC)
   TM values: Max =   177  Min =  -144  Ave =    -3.61  Std =    37.98

 - Packet number: 8 starting at 2024-09-02T20:28:48.837.595.925 (XO: 6.837599973 - 6.837616973, t=38.8 degC)
   TM values: Max =   100  Min =  -116  Ave =    -4.39  Std =    47.49

 - Packet number: 9 starting at 2024-09-02T20:28:49.671.538.925 (XO: 7.671542875 - 7.671559875, t=38.8 degC)
   TM values: Max =    62  Min =   -72  Ave =    -4.58  Std =    26.10

 - Packet number: 10 starting at 2024-09-02T20:28:50.505.482.925 (XO: 8.505485693 - 8.505502693, t=38.8 degC)
   TM values: Max =    72  Min =   -75  Ave =    -4.26  Std =    24.44

 - Packet number: 11 starting at 2024-09-02T20:28:51.339.425.925 (XO: 9.339428513 - 9.339445513, t=38.8 degC)
   TM values: Max =   104  Min =  -105  Ave =    -4.49  Std =    35.36

 - Packet number: 12 starting at 2024-09-02T20:28:52.173.369.925 (XO: 10.173371414 - 10.173388415, t=38.8 degC)
   TM values: Max =    73  Min =   -88  Ave =    -4.53  Std =    28.59

 - Packet number: 13 starting at 2024-09-02T20:28:53.007.312.925 (XO: 11.007314251 - 11.007331251, t=38.8 degC)
   TM values: Max =    83  Min =   -96  Ave =    -4.53  Std =    34.05

 - Packet number: 14 starting at 2024-09-02T20:28:53.841.252.925 (XO: 11.841257169 - 11.841274169, t=38.8 degC)
   TM values: Max =    61  Min =   -68  Ave =    -4.51  Std =    25.99

 - Packet number: 15 starting at 2024-09-02T20:28:54.675.196.925 (XO: 12.675200021 - 12.675217021, t=38.8 degC)
   TM values: Max =    58  Min =   -69  Ave =    -4.52  Std =    25.44

 - Packet number: 16 starting at 2024-09-02T20:28:55.509.139.925 (XO: 13.509142856 - 13.509159856, t=38.8 degC)
   TM values: Max =    77  Min =   -88  Ave =    -4.36  Std =    32.72

 - Packet number: 17 starting at 2024-09-02T20:28:56.343.082.925 (XO: 14.343085691 - 14.343102692, t=38.8 degC)
   TM values: Max =    67  Min =   -68  Ave =    -4.36  Std =    18.76

 - Packet number: 18 starting at 2024-09-02T20:28:57.177.026.925 (XO: 15.177028527 - 15.177045527, t=38.8 degC)
   TM values: Max =    78  Min =   -87  Ave =    -4.46  Std =    31.58
