-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Oct  7 21:52:09 2024 UTC
-- File 20240829_203055_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841253us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2024-08-29T20:30:58.000.000.000 to 2024-08-29T20:31:12.177.027.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5472184 to 15.5471703
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999967 in 1 packets: 2
      0.9999965 in 6 packets: 3, 5, 9, 11, 15, 17
      0.9999966 in 7 packets: 4, 6, 8, 10, 12, 14, 16
   XO PPM: 
      -3.3 in 1 packets: 2
      -3.5 in 6 packets: 3, 5, 9, 11, 15, 17
      -3.4 in 7 packets: 4, 6, 8, 10, 12, 14, 16
Temperature FPGA: 
      04 -> 49.8 degC in 7 packets: 1, 3, 4, 5, 11, 13, 14
      03 -> 51.1 degC in 11 packets: 2, 6, 7, 8, 9, 10, 12, 15, 16, 17, 18
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 2 packets: 1, 2
      1 in 16 packets: 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 2 packets: 1, 2
      1 in 16 packets: 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20240829_203055_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5472185 to 15.7241985
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-29T20:30:58.000.000.000 to 2024-08-29T20:31:12.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.547218500 corrected by 40bit 10MHz counter of 1.547218400 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2024-08-29T20:30:57.999.999.925
   Max 1st sample MIU time among all SR packets: 2024-08-29T20:31:12.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028745
   TM values from -113 to 131 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2024-08-29T20:30:57.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    51  Min =   -57  Ave =    -4.71  Std =    18.09

 - Packet number: 2 starting at 2024-08-29T20:30:58.833.938.925 (XO: 0.833942777 - 0.833959777, t=38.8 degC)
   TM values: Max =    50  Min =   -56  Ave =    -4.55  Std =    16.71

 - Packet number: 3 starting at 2024-08-29T20:30:59.667.881.925 (XO: 1.667885663 - 1.667902663, t=38.8 degC)
   TM values: Max =    48  Min =   -58  Ave =    -4.43  Std =    17.71

 - Packet number: 4 starting at 2024-08-29T20:31:00.501.825.925 (XO: 2.501828531 - 2.501845531, t=38.8 degC)
   TM values: Max =    38  Min =   -48  Ave =    -4.51  Std =    14.88

 - Packet number: 5 starting at 2024-08-29T20:31:01.335.768.925 (XO: 3.335771400 - 3.335788400, t=38.8 degC)
   TM values: Max =    45  Min =   -58  Ave =    -4.52  Std =    16.66

 - Packet number: 6 starting at 2024-08-29T20:31:02.169.712.925 (XO: 4.169714302 - 4.169731302, t=38.8 degC)
   TM values: Max =    43  Min =   -47  Ave =    -4.67  Std =    14.00

 - Packet number: 7 starting at 2024-08-29T20:31:03.003.655.925 (XO: 5.003657137 - 5.003674137, t=38.8 degC)
   TM values: Max =    49  Min =   -57  Ave =    -4.72  Std =    20.33

 - Packet number: 8 starting at 2024-08-29T20:31:03.837.595.925 (XO: 5.837599973 - 5.837616973, t=38.8 degC)
   TM values: Max =    47  Min =   -58  Ave =    -4.70  Std =    19.99

 - Packet number: 9 starting at 2024-08-29T20:31:04.671.539.925 (XO: 6.671542875 - 6.671559875, t=38.8 degC)
   TM values: Max =    54  Min =   -69  Ave =    -4.76  Std =    21.56

 - Packet number: 10 starting at 2024-08-29T20:31:05.505.482.925 (XO: 7.505485744 - 7.505502744, t=38.8 degC)
   TM values: Max =    67  Min =   -67  Ave =    -4.76  Std =    24.69

 - Packet number: 11 starting at 2024-08-29T20:31:06.339.425.925 (XO: 8.339428613 - 8.339445613, t=38.8 degC)
   TM values: Max =    63  Min =   -67  Ave =    -4.64  Std =    20.07

 - Packet number: 12 starting at 2024-08-29T20:31:07.173.369.925 (XO: 9.173371514 - 9.173388515, t=38.8 degC)
   TM values: Max =    63  Min =   -68  Ave =    -4.61  Std =    27.00

 - Packet number: 13 starting at 2024-08-29T20:31:08.007.312.925 (XO: 10.007314350 - 10.007331350, t=38.8 degC)
   TM values: Max =    43  Min =   -56  Ave =    -4.49  Std =    18.51

 - Packet number: 14 starting at 2024-08-29T20:31:08.841.252.925 (XO: 10.841257185 - 10.841274185, t=38.8 degC)
   TM values: Max =    52  Min =   -55  Ave =    -4.64  Std =    19.68

 - Packet number: 15 starting at 2024-08-29T20:31:09.675.196.925 (XO: 11.675200088 - 11.675217088, t=38.8 degC)
   TM values: Max =    68  Min =   -75  Ave =    -4.58  Std =    21.75

 - Packet number: 16 starting at 2024-08-29T20:31:10.509.139.925 (XO: 12.509142956 - 12.509159956, t=38.8 degC)
   TM values: Max =    58  Min =   -66  Ave =    -4.47  Std =    25.63

 - Packet number: 17 starting at 2024-08-29T20:31:11.343.083.925 (XO: 13.343085826 - 13.343102826, t=38.8 degC)
   TM values: Max =    47  Min =   -48  Ave =    -4.63  Std =    16.65

 - Packet number: 18 starting at 2024-08-29T20:31:12.177.026.925 (XO: 14.177028745 - 14.177045745, t=38.8 degC)
   TM values: Max =   131  Min =  -113  Ave =    -4.74  Std =    31.25
