-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Oct  7 21:36:11 2024 UTC
-- File 20240829_195721_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.482576 to 5.482865
-each separately from 5s to 5s and from 482576us to 482865us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2024-08-29T19:57:29.482.576.000 to 2024-08-29T19:57:29.482.865.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.5667897 to 6.5667897
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20240829_195721_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0493670 to 7.0496560
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-29T19:57:29.482.576.000 to 2024-08-29T19:57:29.482.865.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.049367000 corrected by 40bit 10MHz counter of 6.566789700 obtained at 1Hz counter of 5. Result: 5.482578941
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-29T19:57:29.482.435.725
   Max 1st sample MIU time among all SL packets: 2024-08-29T19:57:29.482.724.725
   Min 1st sample XO time among all SL packets: 5.482438665
   Max 1st sample XO time among all SL packets: 5.482727666
   TM values from -310 to 241 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-29T19:57:29.482.435.725 (XO: 5.482438665 - 5.482744666, t=38.8 degC)  Rank: 00080
   TM values: Max =   241  Min =  -310  Ave =    -4.53  Std =    27.17
   Min TM value:-310 at an offset of 136.975 usec -> 2024-08-29T19:57:29.482.572.700 (XO: 5.482575641)
   Max TM value:241 at an offset of 125.250 usec -> 2024-08-29T19:57:29.482.560.975 (XO: 5.482563916)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 139.513 usec -> 2024-08-29T19:57:29.482.575.237 (XO: 5.482578178)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 114.325 usec -> 2024-08-29T19:57:29.482.550.050 (XO: 5.482552991)
