-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Oct  7 20:38:12 2024 UTC
-- File 20240829_173542_1229.dat, 27060  packets found.
APID:
      1229 (EW) in 27060 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 27060 packets (all packets)
Total number of packets: 
      36720 in 27060 packets (all packets)
Packet No from 1 to 27060
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 2.454007 to 9.313736
-each separately from 2s to 9s and from 313736us to 454007us.
   Differences in neigboring packets: 
      0.000000 in 27056 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -0.939832 in 1 packets: 3672
      0.939832 in 1 packets: 15912
      -6.859729 in 1 packets: 23052
MIU packet time from 2024-08-29T17:35:47.454.007.000 to 2024-08-29T17:35:54.313.736.000
   Differences in neigboring packets: 
      0.000000 in 27056 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -0.939832 in 1 packets: 3672
      0.939832 in 1 packets: 15912
      -6.859729 in 1 packets: 23052
MIU orbit number: 
      60000 in 27060 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 27060 packets (all packets)
MIU orbit information / number of events: 
      0 in 27060 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 27060 packets (all packets)
Instrument anomalies: 
      0 (OK) in 27060 packets (all packets)
FPGA configuration version: 
      211 in 27060 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7130135 to 10.7129895
   Differences in neigboring packets: 
      0.0000000 in 27056 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -0.9999966 in 1 packets: 3672
      0.9999966 in 1 packets: 15912
      -6.9999760 in 1 packets: 23052
Temperature FPGA: 
      03 -> 51.1 degC in 10812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 16248 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
Temperature XO: 
      C2 -> 38.8 degC in 27060 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 27060 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 27060 packets (all packets)
Number of external events: 
      3 in 10812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
      1 in 4008 packets: 23053, 23054, 23055, 23056, 23057, 23058, 23059, 23060, 23061, 23062, 23063, 23064, 23065, 23066, 23067, 23068, 23069, 23070, ...
Number of generated alerts: 
      3 in 10812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
      1 in 4008 packets: 23053, 23054, 23055, 23056, 23057, 23058, 23059, 23060, 23061, 23062, 23063, 23064, 23065, 23066, 23067, 23068, 23069, 23070, ...
Survey memory error: 
      0 in 27060 packets (all packets)
Event memory error: 
      0 in 27060 packets (all packets)
Configuration error: 
      0 in 27060 packets (all packets)
SDRAM CRC error: 
      0 in 27060 packets (all packets)
Radio gain: 
      0 in 27060 packets (all packets)
Radio bandwidth: 
      0 in 27060 packets (all packets)
Digital gain: 
      0 in 27060 packets (all packets)
Analog gain: 
      0 in 27060 packets (all packets)

EVENT DATA READER
-- File 20240829_173542_1229.dat, 132 EW pages found. Extra packets: 132.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 10812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
      1 in 3876 packets: 23053, 23054, 23055, 23056, 23057, 23058, 23059, 23060, 23061, 23062, 23063, 23064, 23065, 23066, 23067, 23068, 23069, 23070, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 4.1115342 to 11.1793012
      Difference between the neighboring packets: 
      0.0000170 in 26924 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -1.0265110 in 1 packets: 3672
      0.8184650 in 1 packets: 15912
      -7.0677670 in 1 packets: 23052
   Memory page number from 0 to 169
      Difference between the neighboring packets: 
      0 in 26796 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 130 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      39 in 1 packets: 23052
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 26796 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 128 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3876, ...
      -24 in 1 packets: 3672
      -34 in 1 packets: 15912
      -59 in 1 packets: 23052
   MIU event flag 
      1 in 26928 packets (all packets)
   Time triggered event flag 
      0 in 26928 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-08-29T17:35:47.398.518.825
   Min EW header abs. time found among all events            : 2024-08-29T17:35:47.454.007.000
   Max EW event (1st sample) abs. time found among all events: 2024-08-29T17:35:54.258.247.825
   Max EW header abs. time found among all events            : 2024-08-29T17:35:54.313.736.000
   TM values from -339 to 252 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 3876
    Page no. 1:  204 packets  -- OK;  memory page(s):
      151 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      152 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      153 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      154 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      155 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      156 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      157 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      158 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      159 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      160 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      161 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      162 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      163 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      164 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      165 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      166 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      167 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      168 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      169 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 19  ordered packets: 3876
   TM values from -329 to 231 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=2 C1MHz=0.454007
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3875 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 4.1115342
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-08-29T17:35:47.454.007.000
   Event 1st sample abs. time : 2024-08-29T17:35:47.398.518.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -339 to 252 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=8 C1MHz=0.373904
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 10.0314102
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-08-29T17:35:53.373.904.000
   Event 1st sample abs. time : 2024-08-29T17:35:53.318.415.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 3 total number of packets: 10812
    Page no. 8:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 53  ordered packets: 10812
   TM values from -330 to 251 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=9 C1MHz=0.313736
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 10811 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 10.9955142
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-08-29T17:35:54.313.736.000
   Event 1st sample abs. time : 2024-08-29T17:35:54.258.247.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2024-08-29T17:35:47.398.518.825 to 2024-08-29T17:35:47.464.410.825 (XO: -333.000000175 - -332.934108175)
   Min TM value:-329 at an offset of 32801.363 usec -> 2024-08-29T17:35:47.431.320.187 (XO: -332.967198813)
   Max TM value:231 at an offset of 52818.750 usec -> 2024-08-29T17:35:47.451.337.575 (XO: -332.947181425)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 52821.525 usec -> 2024-08-29T17:35:47.451.340.350 (XO: -332.947178650)
   Min dTM/dt [1/us]:-3.15e+03 at an offset of 12824.600 usec -> 2024-08-29T17:35:47.411.343.425 (XO: -332.987175575)
   Med: -4.00000   Ave: -4.4896072   Std: 17.735454

 - Event number: 2 from 2024-08-29T17:35:53.318.415.825 to 2024-08-29T17:35:53.526.495.825 (XO: -333.000000175 - -332.791920175)
   Min TM value:-339 at an offset of 153063.013 usec -> 2024-08-29T17:35:53.471.478.837 (XO: -332.846937163)
   Max TM value:252 at an offset of 53020.938 usec -> 2024-08-29T17:35:53.371.436.762 (XO: -332.946979238)
   Max dTM/dt [1/us]:2.84e+03 at an offset of 53025.488 usec -> 2024-08-29T17:35:53.371.441.312 (XO: -332.946974688)
   Min dTM/dt [1/us]:-3.27e+03 at an offset of 153062.900 usec -> 2024-08-29T17:35:53.471.478.725 (XO: -332.846937275)
   Med: -4.00000   Ave: -4.4053938   Std: 15.284083

 - Event number: 3 from 2024-08-29T17:35:54.258.247.825 to 2024-08-29T17:35:54.442.051.825 (XO: -333.000000175 - -332.816196175)
   Min TM value:-330 at an offset of 48936.062 usec -> 2024-08-29T17:35:54.307.183.887 (XO: -332.951064113)
   Max TM value:251 at an offset of 88899.175 usec -> 2024-08-29T17:35:54.347.147.000 (XO: -332.911101000)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 28923.900 usec -> 2024-08-29T17:35:54.287.171.725 (XO: -332.971076275)
   Min dTM/dt [1/us]:-3.06e+03 at an offset of 88899.275 usec -> 2024-08-29T17:35:54.347.147.100 (XO: -332.911100900)
   Med: -4.00000   Ave: -4.4518125   Std: 14.936553
