-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Sep 29 01:34:32 2024 UTC
-- File 20240823_211606_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.344264 to 21.304441
-each separately from 7s to 21s and from 304152us to 344553us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.959599 in 1 packets: 18
MIU packet time from 2024-08-23T21:16:16.344.264.000 to 2024-08-23T21:16:30.304.441.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.959599 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.6755387 to 22.6754903
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999516 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240823_211606_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.0198038 to 9.0200928
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9595510  -- OK for SL --
   MIU time from 2024-08-23T21:16:16.344.264.000 to 2024-08-23T21:16:16.344.553.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.959599

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9796438 to 22.9799328
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-23T21:16:30.304.152.000 to 2024-08-23T21:16:30.304.441.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.019803800 corrected by 40bit 10MHz counter of 8.675538700 obtained at 1Hz counter of 7. Result: 7.344266305
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-23T21:16:16.344.123.725
   Max 1st sample MIU time among all SL packets: 2024-08-23T21:16:30.304.300.725
   Min 1st sample XO time among all SL packets: 7.344126029
   Max 1st sample XO time among all SL packets: 21.304303260
   TM values from -266 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-23T21:16:16.344.123.725 (XO: 7.344126029 - 7.344432031, t=38.8 degC)  Rank: 00081
   TM values: Max =   221  Min =  -262  Ave =    -4.39  Std =    22.73
   Min TM value:-262 at an offset of 116.288 usec -> 2024-08-23T21:16:16.344.240.012 (XO: 7.344242317)
   Max TM value:221 at an offset of 104.150 usec -> 2024-08-23T21:16:16.344.227.875 (XO: 7.344230180)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 102.100 usec -> 2024-08-23T21:16:16.344.225.825 (XO: 7.344228130)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 104.250 usec -> 2024-08-23T21:16:16.344.227.975 (XO: 7.344230280)

 - Block number: 2 starting at 2024-08-23T21:16:30.304.011.725 (XO: 21.304014259 - 21.304320260, t=38.8 degC)  Rank: 00079
   TM values: Max =   227  Min =  -266  Ave =    -4.43  Std =    19.60
   Min TM value:-266 at an offset of 103.900 usec -> 2024-08-23T21:16:30.304.115.625 (XO: 21.304118159)
   Max TM value:227 at an offset of 103.613 usec -> 2024-08-23T21:16:30.304.115.337 (XO: 21.304117872)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 106.275 usec -> 2024-08-23T21:16:30.304.118.000 (XO: 21.304120534)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 103.725 usec -> 2024-08-23T21:16:30.304.115.450 (XO: 21.304117984)
