-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Sep 29 00:29:06 2024 UTC
-- File 20240823_193728_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.857999 to 70.361714
-each separately from 0s to 70s and from 60478us to 858288us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.660939 in 1 packets: 18
      17.920718 in 1 packets: 36
      5.619955 in 1 packets: 54
      24.300658 in 1 packets: 72
MIU packet time from 2024-08-23T19:37:31.857.999.000 to 2024-08-23T19:38:41.361.714.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.660939 in 1 packets: 18
      17.920718 in 1 packets: 36
      5.619955 in 1 packets: 54
      24.300658 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7157587 to 71.7155163
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999239 in 1 packets: 18
      17.9999376 in 1 packets: 36
      5.9999792 in 1 packets: 54
      23.9999169 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20240823_193728_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5737589 to 2.5740479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6608630  -- OK for SL --
   MIU time from 2024-08-23T19:37:31.857.999.000 to 2024-08-23T19:37:31.858.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.660939

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2349109 to 24.2351999
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9206550  -- OK for SL --
   MIU time from 2024-08-23T19:37:53.519.227.000 to 2024-08-23T19:37:53.519.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.920718

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.1558549 to 42.1561439
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.6199350  -- OK for SL --
   MIU time from 2024-08-23T19:38:11.440.234.000 to 2024-08-23T19:38:11.440.523.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.619955

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.7760789 to 47.7763679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.3005750  -- OK for SL --
   MIU time from 2024-08-23T19:38:17.060.478.000 to 2024-08-23T19:38:17.060.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.300658

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.0769429 to 72.0772319
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-23T19:38:41.361.425.000 to 2024-08-23T19:38:41.361.714.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.573758900 corrected by 40bit 10MHz counter of 1.715758700 obtained at 1Hz counter of 0. Result: 0.858003117
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-23T19:37:31.857.858.725
   Max 1st sample MIU time among all SL packets: 2024-08-23T19:38:41.361.573.725
   Min 1st sample XO time among all SL packets: 0.857862842
   Max 1st sample XO time among all SL packets: 70.361576591
   TM values from -299 to 234 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-23T19:37:31.857.858.725 (XO: 0.857862842 - 0.858168843, t=38.8 degC)  Rank: 00077
   TM values: Max =   206  Min =  -288  Ave =    -4.44  Std =    25.48
   Min TM value:-288 at an offset of 140.975 usec -> 2024-08-23T19:37:31.857.999.700 (XO: 0.858003817)
   Max TM value:206 at an offset of 109.863 usec -> 2024-08-23T19:37:31.857.968.587 (XO: 0.857972705)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 114.463 usec -> 2024-08-23T19:37:31.857.973.187 (XO: 0.857977305)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 140.825 usec -> 2024-08-23T19:37:31.857.999.550 (XO: 0.858003667)

 - Block number: 2 starting at 2024-08-23T19:37:53.519.086.725 (XO: 22.519089842 - 22.519395843, t=38.8 degC)  Rank: 00086
   TM values: Max =   229  Min =  -265  Ave =    -4.44  Std =    21.86
   Min TM value:-265 at an offset of 120.825 usec -> 2024-08-23T19:37:53.519.207.550 (XO: 22.519210667)
   Max TM value:229 at an offset of 118.613 usec -> 2024-08-23T19:37:53.519.205.337 (XO: 22.519208455)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 123.163 usec -> 2024-08-23T19:37:53.519.209.887 (XO: 22.519213005)
   Min dTM/dt [1/us]:-2.74e+03 at an offset of 120.638 usec -> 2024-08-23T19:37:53.519.207.362 (XO: 22.519210480)

 - Block number: 3 starting at 2024-08-23T19:38:11.440.093.725 (XO: 40.440095965 - 40.440401966, t=38.8 degC)  Rank: 00079
   TM values: Max =   230  Min =  -299  Ave =    -4.46  Std =    26.65
   Min TM value:-299 at an offset of 120.463 usec -> 2024-08-23T19:38:11.440.214.187 (XO: 40.440216428)
   Max TM value:230 at an offset of 108.475 usec -> 2024-08-23T19:38:11.440.202.200 (XO: 40.440204441)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 122.963 usec -> 2024-08-23T19:38:11.440.216.687 (XO: 40.440218928)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 120.275 usec -> 2024-08-23T19:38:11.440.214.000 (XO: 40.440216241)

 - Block number: 4 starting at 2024-08-23T19:38:17.060.337.725 (XO: 46.060339430 - 46.060645431, t=38.8 degC)  Rank: 00079
   TM values: Max =   234  Min =  -252  Ave =    -4.33  Std =    21.31
   Min TM value:-252 at an offset of 121.288 usec -> 2024-08-23T19:38:17.060.459.012 (XO: 46.060460718)
   Max TM value:234 at an offset of 119.125 usec -> 2024-08-23T19:38:17.060.456.850 (XO: 46.060458556)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 118.913 usec -> 2024-08-23T19:38:17.060.456.637 (XO: 46.060458343)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 121.113 usec -> 2024-08-23T19:38:17.060.458.837 (XO: 46.060460543)

 - Block number: 5 starting at 2024-08-23T19:38:41.361.284.725 (XO: 70.361287590 - 70.361593591, t=38.8 degC)  Rank: 00081
   TM values: Max =   176  Min =  -249  Ave =    -4.48  Std =    18.38
   Min TM value:-249 at an offset of 107.925 usec -> 2024-08-23T19:38:41.361.392.650 (XO: 70.361395515)
   Max TM value:176 at an offset of 110.675 usec -> 2024-08-23T19:38:41.361.395.400 (XO: 70.361398265)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 110.488 usec -> 2024-08-23T19:38:41.361.395.212 (XO: 70.361398077)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 107.738 usec -> 2024-08-23T19:38:41.361.392.462 (XO: 70.361395327)
