-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Sep 28 23:20:31 2024 UTC
-- File 20240823_182352_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.427141 to 25.447233
-each separately from 8s to 25s and from 427141us to 447233us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.019514 in 1 packets: 18
MIU packet time from 2024-08-23T18:24:03.427.141.000 to 2024-08-23T18:24:20.447.233.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.019514 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.6118332 to 26.6117741
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999409 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240823_182352_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0389753 to 10.0392643
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0194550  -- OK for SL --
   MIU time from 2024-08-23T18:24:03.427.141.000 to 2024-08-23T18:24:03.427.430.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.019514

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0587193 to 27.0590083
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-23T18:24:20.446.944.000 to 2024-08-23T18:24:20.447.233.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.038975300 corrected by 40bit 10MHz counter of 9.611833200 obtained at 1Hz counter of 8. Result: 8.427143595
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-08-23T18:24:03.427.000.725
   Max 1st sample MIU time among all SL packets: 2024-08-23T18:24:20.447.092.725
   Min 1st sample XO time among all SL packets: 8.427003320
   Max 1st sample XO time among all SL packets: 25.447095490
   TM values from -309 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-23T18:24:03.427.000.725 (XO: 8.427003320 - 8.427309321, t=38.8 degC)  Rank: 00077
   TM values: Max =   211  Min =  -263  Ave =    -4.41  Std =    21.33
   Min TM value:-263 at an offset of 118.900 usec -> 2024-08-23T18:24:03.427.119.625 (XO: 8.427122220)
   Max TM value:211 at an offset of 116.675 usec -> 2024-08-23T18:24:03.427.117.400 (XO: 8.427119995)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 116.500 usec -> 2024-08-23T18:24:03.427.117.225 (XO: 8.427119820)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 118.713 usec -> 2024-08-23T18:24:03.427.119.437 (XO: 8.427122032)

 - Block number: 2 starting at 2024-08-23T18:24:20.446.803.725 (XO: 25.446806489 - 25.447112490, t=38.8 degC)  Rank: 00084
   TM values: Max =   230  Min =  -309  Ave =    -4.37  Std =    29.80
   Min TM value:-309 at an offset of 127.775 usec -> 2024-08-23T18:24:20.446.931.500 (XO: 25.446934264)
   Max TM value:230 at an offset of 115.650 usec -> 2024-08-23T18:24:20.446.919.375 (XO: 25.446922139)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 113.550 usec -> 2024-08-23T18:24:20.446.917.275 (XO: 25.446920039)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 115.775 usec -> 2024-08-23T18:24:20.446.919.500 (XO: 25.446922264)
