-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Sep 28 05:31:00 2024 UTC
-- File 20240823_062839_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.655729 to 28.635739
-each separately from 7s to 28s and from 635450us to 656018us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.979432 in 1 packets: 18
MIU packet time from 2024-08-23T06:28:49.655.729.000 to 2024-08-23T06:29:10.635.739.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.979432 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.6025178 to 29.6024448
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999270 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240823_062839_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      98 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2582474 to 9.2585364
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9793590  -- OK for SL --
   MIU time from 2024-08-23T06:28:49.655.729.000 to 2024-08-23T06:28:49.656.018.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.979432

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.2378954 to 30.2381844
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-23T06:29:10.635.450.000 to 2024-08-23T06:29:10.635.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.258247400 corrected by 40bit 10MHz counter of 8.602517800 obtained at 1Hz counter of 7. Result: 7.655731895
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-08-23T06:28:49.655.588.725
   Max 1st sample MIU time among all SL packets: 2024-08-23T06:29:10.635.598.725
   Min 1st sample XO time among all SL packets: 7.655591620
   Max 1st sample XO time among all SL packets: 28.635601550
   TM values from -938 to 1009 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-23T06:28:49.655.588.725 (XO: 7.655591620 - 7.655897621, t=38.8 degC)  Rank: 00098
   TM values: Max =  1009  Min =  -938  Ave =    -4.50  Std =    33.55
   Min TM value:-938 at an offset of 128.288 usec -> 2024-08-23T06:28:49.655.717.012 (XO: 7.655719908)
   Max TM value:1009 at an offset of 128.088 usec -> 2024-08-23T06:28:49.655.716.812 (XO: 7.655719708)
   Max dTM/dt [1/us]:1.21e+04 at an offset of 128.063 usec -> 2024-08-23T06:28:49.655.716.787 (XO: 7.655719683)
   Min dTM/dt [1/us]:-9.20e+03 at an offset of 128.138 usec -> 2024-08-23T06:28:49.655.716.862 (XO: 7.655719758)

 - Block number: 2 starting at 2024-08-23T06:29:10.635.309.725 (XO: 28.635312549 - 28.635618550, t=38.8 degC)  Rank: 00084
   TM values: Max =   969  Min =  -879  Ave =    -4.45  Std =    36.94
   Min TM value:-879 at an offset of 98.963 usec -> 2024-08-23T06:29:10.635.408.687 (XO: 28.635411511)
   Max TM value:969 at an offset of 98.763 usec -> 2024-08-23T06:29:10.635.408.487 (XO: 28.635411311)
   Max dTM/dt [1/us]:1.22e+04 at an offset of 98.738 usec -> 2024-08-23T06:29:10.635.408.462 (XO: 28.635411286)
   Min dTM/dt [1/us]:-9.02e+03 at an offset of 98.825 usec -> 2024-08-23T06:29:10.635.408.550 (XO: 28.635411374)
