-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Sep 28 00:46:47 2024 UTC
-- File 20240822_230239_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.729968 to 21.930418
-each separately from 2s to 21s and from 729968us to 930418us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.199872 in 1 packets: 18
MIU packet time from 2024-08-22T23:02:44.729.968.000 to 2024-08-22T23:03:03.930.418.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.199872 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.5659200 to 21.5658547
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999347 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240822_230239_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2958891 to 3.2961781
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1998070  -- OK for SL --
   MIU time from 2024-08-22T23:02:44.729.968.000 to 2024-08-22T23:02:44.730.257.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.199872

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.4959851 to 22.4962741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-22T23:03:03.930.129.000 to 2024-08-22T23:03:03.930.418.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 30 XO calibration points at 1Hz counter from 0 to 29
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.295889100 corrected by 40bit 10MHz counter of 2.565920000 obtained at 1Hz counter of 2. Result: 2.729971582
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-22T23:02:44.729.827.725
   Max 1st sample MIU time among all SL packets: 2024-08-22T23:03:03.930.277.725
   Min 1st sample XO time among all SL packets: 2.729831306
   Max 1st sample XO time among all SL packets: 21.930282381
   TM values from -274 to 209 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-22T23:02:44.729.827.725 (XO: 2.729831306 - 2.730137307, t=38.8 degC)  Rank: 00089
   TM values: Max =   209  Min =  -258  Ave =    -4.41  Std =    20.17
   Min TM value:-258 at an offset of 128.175 usec -> 2024-08-22T23:02:44.729.955.900 (XO: 2.729959482)
   Max TM value:209 at an offset of 126.000 usec -> 2024-08-22T23:02:44.729.953.725 (XO: 2.729957307)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 130.525 usec -> 2024-08-22T23:02:44.729.958.250 (XO: 2.729961832)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 127.988 usec -> 2024-08-22T23:02:44.729.955.712 (XO: 2.729959294)

 - Block number: 2 starting at 2024-08-22T23:03:03.929.988.725 (XO: 21.929993380 - 21.930299381, t=38.8 degC)  Rank: 00080
   TM values: Max =   203  Min =  -274  Ave =    -4.47  Std =    21.97
   Min TM value:-274 at an offset of 116.075 usec -> 2024-08-22T23:03:03.930.104.800 (XO: 21.930109455)
   Max TM value:203 at an offset of 103.913 usec -> 2024-08-22T23:03:03.930.092.637 (XO: 21.930097293)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 101.825 usec -> 2024-08-22T23:03:03.930.090.550 (XO: 21.930095205)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 104.038 usec -> 2024-08-22T23:03:03.930.092.762 (XO: 21.930097418)
