-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Sep 27 05:34:16 2024 UTC
-- File 20240822_210421_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.231877 to 21.853647
-each separately from 0s to 21s and from 231877us to 853647us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.621192 in 1 packets: 18
MIU packet time from 2024-08-22T21:04:24.231.877.000 to 2024-08-22T21:04:45.853.647.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.621192 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4766055 to 22.4765330
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999275 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240822_210421_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7084840 to 1.7087730
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6211190  -- OK for SL --
   MIU time from 2024-08-22T21:04:24.231.877.000 to 2024-08-22T21:04:24.232.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.621192

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.3298920 to 23.3301810
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-22T21:04:45.853.358.000 to 2024-08-22T21:04:45.853.647.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.708484000 corrected by 40bit 10MHz counter of 1.476605500 obtained at 1Hz counter of 0. Result: 0.231879312
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-08-22T21:04:24.231.736.725
   Max 1st sample MIU time among all SL packets: 2024-08-22T21:04:45.853.506.725
   Min 1st sample XO time among all SL packets: 0.231739036
   Max 1st sample XO time among all SL packets: 21.853510712
   TM values from -282 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-22T21:04:24.231.736.725 (XO: 0.231739036 - 0.232045037, t=38.8 degC)  Rank: 00079
   TM values: Max =   226  Min =  -280  Ave =    -4.50  Std =    23.06
   Min TM value:-280 at an offset of 140.063 usec -> 2024-08-22T21:04:24.231.876.787 (XO: 0.231879099)
   Max TM value:226 at an offset of 127.950 usec -> 2024-08-22T21:04:24.231.864.675 (XO: 0.231866987)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 125.913 usec -> 2024-08-22T21:04:24.231.862.637 (XO: 0.231864949)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 128.063 usec -> 2024-08-22T21:04:24.231.864.787 (XO: 0.231867099)

 - Block number: 2 starting at 2024-08-22T21:04:45.853.217.725 (XO: 21.853221711 - 21.853527712, t=38.8 degC)  Rank: 00090
   TM values: Max =   200  Min =  -282  Ave =    -4.55  Std =    22.40
   Min TM value:-282 at an offset of 123.200 usec -> 2024-08-22T21:04:45.853.340.925 (XO: 21.853344912)
   Max TM value:200 at an offset of 111.000 usec -> 2024-08-22T21:04:45.853.328.725 (XO: 21.853332712)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 125.763 usec -> 2024-08-22T21:04:45.853.343.487 (XO: 21.853347474)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 111.125 usec -> 2024-08-22T21:04:45.853.328.850 (XO: 21.853332837)
