-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Sep 27 05:27:13 2024 UTC
-- File 20240822_205604_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.389263 to 35.969546
-each separately from 9s to 35s and from 289262us to 969546us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.899710 in 1 packets: 18
      17.679706 in 1 packets: 36
MIU packet time from 2024-08-22T20:56:16.389.263.000 to 2024-08-22T20:56:42.969.546.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.899710 in 1 packets: 18
      17.679706 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.5995071 to 36.5994172
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999688 in 1 packets: 18
      16.9999413 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240822_205604_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.9887712 to 10.9890602
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.8996790  -- OK for SL --
   MIU time from 2024-08-22T20:56:16.389.263.000 to 2024-08-22T20:56:16.389.552.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.899710

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8887392 to 19.8890282
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6796470  -- OK for SL --
   MIU time from 2024-08-22T20:56:25.289.262.000 to 2024-08-22T20:56:25.289.551.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.679706

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.5686752 to 37.5689642
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-22T20:56:42.969.257.000 to 2024-08-22T20:56:42.969.546.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.988771200 corrected by 40bit 10MHz counter of 10.599507100 obtained at 1Hz counter of 9. Result: 9.389265462
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-22T20:56:16.389.122.725
   Max 1st sample MIU time among all SL packets: 2024-08-22T20:56:42.969.405.725
   Min 1st sample XO time among all SL packets: 9.389125187
   Max 1st sample XO time among all SL packets: 35.969410118
   TM values from -300 to 231 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-22T20:56:16.389.122.725 (XO: 9.389125187 - 9.389431188, t=38.8 degC)  Rank: 00089
   TM values: Max =   217  Min =  -255  Ave =    -4.52  Std =    20.57
   Min TM value:-255 at an offset of 123.025 usec -> 2024-08-22T20:56:16.389.245.750 (XO: 9.389248212)
   Max TM value:217 at an offset of 120.863 usec -> 2024-08-22T20:56:16.389.243.587 (XO: 9.389246050)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 125.463 usec -> 2024-08-22T20:56:16.389.248.187 (XO: 9.389250650)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 122.888 usec -> 2024-08-22T20:56:16.389.245.612 (XO: 9.389248075)

 - Block number: 2 starting at 2024-08-22T20:56:25.289.121.725 (XO: 18.289124008 - 18.289430009, t=38.8 degC)  Rank: 00080
   TM values: Max =   231  Min =  -300  Ave =    -4.63  Std =    27.03
   Min TM value:-300 at an offset of 152.176 usec -> 2024-08-22T20:56:25.289.273.900 (XO: 18.289276184)
   Max TM value:231 at an offset of 121.538 usec -> 2024-08-22T20:56:25.289.243.262 (XO: 18.289245546)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 126.050 usec -> 2024-08-22T20:56:25.289.247.775 (XO: 18.289250058)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 152.013 usec -> 2024-08-22T20:56:25.289.273.738 (XO: 18.289276021)

 - Block number: 3 starting at 2024-08-22T20:56:42.969.116.725 (XO: 35.969121117 - 35.969427118, t=38.8 degC)  Rank: 00083
   TM values: Max =   201  Min =  -222  Ave =    -4.63  Std =    19.64
   Min TM value:-222 at an offset of 108.125 usec -> 2024-08-22T20:56:42.969.224.850 (XO: 35.969229242)
   Max TM value:201 at an offset of 105.888 usec -> 2024-08-22T20:56:42.969.222.612 (XO: 35.969227005)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 110.500 usec -> 2024-08-22T20:56:42.969.227.225 (XO: 35.969231617)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 107.925 usec -> 2024-08-22T20:56:42.969.224.650 (XO: 35.969229042)
