-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Sep 27 01:17:44 2024 UTC
-- File 20240822_074403_1229.dat, 7094  packets found.
APID:
      1229 (EW) in 7094 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 7094 packets (all packets)
Total number of packets: 
      36720 in 7094 packets (all packets)
Packet No from 1 to 7094
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 2.416178 to 2.416178
-each separately from 2s to 2s and from 416178us to 416178us.
   Differences in neigboring packets: 
      0.000000 in 7093 packets (all packets)
MIU packet time from 2024-08-22T07:44:08.416.178.000 to 2024-08-22T07:44:08.416.178.000
   Differences in neigboring packets: 
      0.000000 in 7093 packets (all packets)
MIU orbit number: 
      60000 in 7094 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 7094 packets (all packets)
MIU orbit information / number of events: 
      0 in 7094 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 7094 packets (all packets)
Instrument anomalies: 
      0 (OK) in 7094 packets (all packets)
FPGA configuration version: 
      211 in 7094 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7269384 to 3.7269384
   Differences in neigboring packets: 
      0.0000000 in 7093 packets (all packets)
Temperature FPGA: 
      03 -> 51.1 degC in 7094 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 7094 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 7094 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 7094 packets (all packets)
Number of external events: 
      3 in 7094 packets (all packets)
Number of generated alerts: 
      3 in 7094 packets (all packets)
Survey memory error: 
      0 in 7094 packets (all packets)
Event memory error: 
      0 in 7094 packets (all packets)
Configuration error: 
      0 in 7094 packets (all packets)
SDRAM CRC error: 
      0 in 7094 packets (all packets)
Radio gain: 
      0 in 7094 packets (all packets)
Radio bandwidth: 
      0 in 7094 packets (all packets)
Digital gain: 
      0 in 7094 packets (all packets)
Analog gain: 
      0 in 7094 packets (all packets)

EVENT DATA READER
-- File 20240822_074403_1229.dat, 34 EW pages found. Extra packets: 158.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 3 to 3
      3 in 6936 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 4.0876292 to 4.2055242
      Difference between the neighboring packets: 
      0.0000170 in 6935 packets (all packets)
   Memory page number from 28 to 61
      Difference between the neighboring packets: 
      0 in 6902 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 33 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 34
      Difference between the neighboring packets: 
      0 in 6902 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 33 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   MIU event flag 
      1 in 6936 packets (all packets)
   Time triggered event flag 
      0 in 6936 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-08-22T07:44:08.360.689.825
   Min EW header abs. time found among all events            : 2024-08-22T07:44:08.416.178.000
   Max EW event (1st sample) abs. time found among all events: 2024-08-22T07:44:08.360.689.825
   Max EW header abs. time found among all events            : 2024-08-22T07:44:08.416.178.000
   TM values from -271 to 208 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 6936
    Page no. 1:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 34  ordered packets: 6936
   TM values from -271 to 208 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=2 C1MHz=0.416178
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 6935 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 4.0876292
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-08-22T07:44:08.416.178.000
   Event 1st sample abs. time : 2024-08-22T07:44:08.360.689.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2024-08-22T07:44:08.360.689.825 to 2024-08-22T07:44:08.478.601.825 (XO: -333.000000175 - -332.882088175)
   Min TM value:-271 at an offset of 32022.213 usec -> 2024-08-22T07:44:08.392.712.037 (XO: -332.967977963)
   Max TM value:208 at an offset of 32021.912 usec -> 2024-08-22T07:44:08.392.711.737 (XO: -332.967978263)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 52019.025 usec -> 2024-08-22T07:44:08.412.708.850 (XO: -332.947981150)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 32022.038 usec -> 2024-08-22T07:44:08.392.711.862 (XO: -332.967978138)
   Med: -4.00000   Ave: -4.3922287   Std: 4.1220022
