-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Sep 26 23:54:23 2024 UTC
-- File 20240822_052928_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.238513 to 21.288785
-each separately from 7s to 21s and from 238513us to 288785us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.049694 in 1 packets: 18
MIU packet time from 2024-08-22T05:29:38.238.513.000 to 2024-08-22T05:29:52.288.785.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.049694 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.5715964 to 22.5715485
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999521 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240822_052928_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8101101 to 8.8103991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0496470  -- OK for SL --
   MIU time from 2024-08-22T05:29:38.238.513.000 to 2024-08-22T05:29:38.238.802.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.049694

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.8600461 to 22.8603351
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-22T05:29:52.288.496.000 to 2024-08-22T05:29:52.288.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.810110100 corrected by 40bit 10MHz counter of 8.571596400 obtained at 1Hz counter of 7. Result: 7.238514511
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-22T05:29:38.238.372.725
   Max 1st sample MIU time among all SL packets: 2024-08-22T05:29:52.288.644.725
   Min 1st sample XO time among all SL packets: 7.238374235
   Max 1st sample XO time among all SL packets: 21.288647306
   TM values from -266 to 205 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-22T05:29:38.238.372.725 (XO: 7.238374235 - 7.238680237, t=38.8 degC)  Rank: 00081
   TM values: Max =   168  Min =  -231  Ave =    -4.36  Std =    20.68
   Min TM value:-231 at an offset of 121.938 usec -> 2024-08-22T05:29:38.238.494.662 (XO: 7.238496173)
   Max TM value:168 at an offset of 109.038 usec -> 2024-08-22T05:29:38.238.481.762 (XO: 7.238483273)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 124.525 usec -> 2024-08-22T05:29:38.238.497.250 (XO: 7.238498761)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 109.200 usec -> 2024-08-22T05:29:38.238.481.925 (XO: 7.238483436)

 - Block number: 2 starting at 2024-08-22T05:29:52.288.355.725 (XO: 21.288358305 - 21.288664306, t=38.8 degC)  Rank: 00084
   TM values: Max =   205  Min =  -266  Ave =    -4.48  Std =    20.89
   Min TM value:-266 at an offset of 108.813 usec -> 2024-08-22T05:29:52.288.464.537 (XO: 21.288467118)
   Max TM value:205 at an offset of 96.863 usec -> 2024-08-22T05:29:52.288.452.587 (XO: 21.288455168)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 94.838 usec -> 2024-08-22T05:29:52.288.450.562 (XO: 21.288453143)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 96.975 usec -> 2024-08-22T05:29:52.288.452.700 (XO: 21.288455281)
