-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Sep 13 00:56:06 2024 UTC
-- File 20240808_202541_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.342857 to 51.783438
-each separately from 4s to 51s and from 122880us to 903396us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.779734 in 1 packets: 18
      24.779938 in 1 packets: 36
      11.879753 in 1 packets: 54
MIU packet time from 2024-08-08T20:25:48.342.857.000 to 2024-08-08T20:26:35.783.438.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.779734 in 1 packets: 18
      24.779938 in 1 packets: 36
      11.879753 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.5752301 to 52.5750670
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999618 in 1 packets: 18
      23.9999167 in 1 packets: 36
      11.9999584 in 1 packets: 54
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240808_202541_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.9180887 to 5.9183777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7796950  -- OK for SL --
   MIU time from 2024-08-08T20:25:48.342.857.000 to 2024-08-08T20:25:48.343.146.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.779734

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6980727 to 16.6983617
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.7798550  -- OK for SL --
   MIU time from 2024-08-08T20:25:59.122.880.000 to 2024-08-08T20:25:59.123.169.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.779938

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4782167 to 41.4785057
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8797110  -- OK for SL --
   MIU time from 2024-08-08T20:26:23.903.107.000 to 2024-08-08T20:26:23.903.396.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.879753

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.3582167 to 53.3585057
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-08T20:26:35.783.149.000 to 2024-08-08T20:26:35.783.438.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.918088700 corrected by 40bit 10MHz counter of 5.575230100 obtained at 1Hz counter of 4. Result: 4.342859800
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-08-08T20:25:48.342.716.725
   Max 1st sample MIU time among all SL packets: 2024-08-08T20:26:35.783.297.725
   Min 1st sample XO time among all SL packets: 4.342719525
   Max 1st sample XO time among all SL packets: 51.783301167
   TM values from -306 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-08T20:25:48.342.716.725 (XO: 4.342719525 - 4.343025526, t=38.8 degC)  Rank: 00077
   TM values: Max =   248  Min =  -259  Ave =    -4.52  Std =    27.81
   Min TM value:-259 at an offset of 109.350 usec -> 2024-08-08T20:25:48.342.826.075 (XO: 4.342828875)
   Max TM value:248 at an offset of 107.213 usec -> 2024-08-08T20:25:48.342.823.937 (XO: 4.342826737)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 111.688 usec -> 2024-08-08T20:25:48.342.828.412 (XO: 4.342831212)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 109.200 usec -> 2024-08-08T20:25:48.342.825.925 (XO: 4.342828725)

 - Block number: 2 starting at 2024-08-08T20:25:59.122.739.725 (XO: 15.122740955 - 15.123046956, t=38.8 degC)  Rank: 00075
   TM values: Max =   239  Min =  -264  Ave =    -4.51  Std =    28.10
   Min TM value:-264 at an offset of 119.463 usec -> 2024-08-08T20:25:59.122.859.187 (XO: 15.122860418)
   Max TM value:239 at an offset of 117.313 usec -> 2024-08-08T20:25:59.122.857.037 (XO: 15.122858268)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 121.813 usec -> 2024-08-08T20:25:59.122.861.537 (XO: 15.122862768)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 119.275 usec -> 2024-08-08T20:25:59.122.859.000 (XO: 15.122860230)

 - Block number: 3 starting at 2024-08-08T20:26:23.902.966.725 (XO: 39.902970895 - 39.903276896, t=38.8 degC)  Rank: 00082
   TM values: Max =   221  Min =  -306  Ave =    -4.49  Std =    35.03
   Min TM value:-306 at an offset of 94.900 usec -> 2024-08-08T20:26:23.903.061.625 (XO: 39.903065795)
   Max TM value:221 at an offset of 106.388 usec -> 2024-08-08T20:26:23.903.073.112 (XO: 39.903077283)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 109.238 usec -> 2024-08-08T20:26:23.903.075.962 (XO: 39.903080133)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 94.750 usec -> 2024-08-08T20:26:23.903.061.475 (XO: 39.903065645)

 - Block number: 4 starting at 2024-08-08T20:26:35.783.008.725 (XO: 51.783012166 - 51.783318167, t=38.8 degC)  Rank: 00083
   TM values: Max =   236  Min =  -298  Ave =    -4.44  Std =    25.15
   Min TM value:-298 at an offset of 119.238 usec -> 2024-08-08T20:26:35.783.127.962 (XO: 51.783131403)
   Max TM value:236 at an offset of 118.950 usec -> 2024-08-08T20:26:35.783.127.675 (XO: 51.783131116)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 105.038 usec -> 2024-08-08T20:26:35.783.113.762 (XO: 51.783117203)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 119.050 usec -> 2024-08-08T20:26:35.783.127.775 (XO: 51.783131216)
