-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu Sep 12 09:47:27 2024 UTC
-- File 20240808_120800_1229.dat, 5781  packets found.
APID:
      1229 (EW) in 5781 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 5781 packets (all packets)
Total number of packets: 
      28764 in 5781 packets (all packets)
Packet No from 1 to 5781
!!!ERROR: Maximum packet number must be = 28764
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 28764
Hz and MHz counters from 1.072717 to 1.072717
-each separately from 1s to 1s and from 72717us to 72717us.
   Differences in neigboring packets: 
      0.000000 in 5780 packets (all packets)
MIU packet time from 2024-08-08T12:08:04.072.717.000 to 2024-08-08T12:08:04.072.717.000
   Differences in neigboring packets: 
      0.000000 in 5780 packets (all packets)
MIU orbit number: 
      60000 in 5781 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 5781 packets (all packets)
MIU orbit information / number of events: 
      0 in 5781 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 5781 packets (all packets)
Instrument anomalies: 
      0 (OK) in 5781 packets (all packets)
FPGA configuration version: 
      211 in 5781 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.0015120 to 3.0015120
   Differences in neigboring packets: 
      0.0000000 in 5780 packets (all packets)
Temperature FPGA: 
      02 -> 52.4 degC in 5781 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 5781 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 5781 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 5781 packets (all packets)
Number of external events: 
      1 in 5781 packets (all packets)
Number of generated alerts: 
      1 in 5781 packets (all packets)
Survey memory error: 
      0 in 5781 packets (all packets)
Event memory error: 
      0 in 5781 packets (all packets)
Configuration error: 
      0 in 5781 packets (all packets)
SDRAM CRC error: 
      0 in 5781 packets (all packets)
Radio gain: 
      0 in 5781 packets (all packets)
Radio bandwidth: 
      0 in 5781 packets (all packets)
Digital gain: 
      0 in 5781 packets (all packets)
Analog gain: 
      0 in 5781 packets (all packets)

EVENT DATA READER
-- File 20240808_120800_1229.dat, 28 EW pages found. Extra packets: 69.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 1
      1 in 5712 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 3.0187420 to 3.1158290
      Difference between the neighboring packets: 
      0.0000170 in 5711 packets (all packets)
   Memory page number from 75 to 102
      Difference between the neighboring packets: 
      0 in 5684 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 27 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 28
      Difference between the neighboring packets: 
      0 in 5684 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 27 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   MIU event flag 
      1 in 5712 packets (all packets)
   Time triggered event flag 
      0 in 5712 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-08-08T12:08:04.017.228.825
   Min EW header abs. time found among all events            : 2024-08-08T12:08:04.072.717.000
   Max EW event (1st sample) abs. time found among all events: 2024-08-08T12:08:04.017.228.825
   Max EW header abs. time found among all events            : 2024-08-08T12:08:04.072.717.000
   TM values from -294 to 261 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 5712
    Page no. 1:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 28  ordered packets: 5712
   TM values from -294 to 261 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=1 C1MHz=0.072717
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 5711 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 3.0187420
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-08-08T12:08:04.072.717.000
   Event 1st sample abs. time : 2024-08-08T12:08:04.017.228.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2024-08-08T12:08:04.017.228.825 to 2024-08-08T12:08:04.114.332.825 (XO: -333.000000175 - -332.902896175)
   Min TM value:-294 at an offset of 33632.612 usec -> 2024-08-08T12:08:04.050.861.437 (XO: -332.966367563)
   Max TM value:261 at an offset of 33632.362 usec -> 2024-08-08T12:08:04.050.861.187 (XO: -332.966367813)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 33630.387 usec -> 2024-08-08T12:08:04.050.859.212 (XO: -332.966369788)
   Min dTM/dt [1/us]:-3.28e+03 at an offset of 33632.475 usec -> 2024-08-08T12:08:04.050.861.300 (XO: -332.966367700)
   Med: -4.00000   Ave: -4.4338473   Std: 4.4043427
