-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Aug  7 10:17:10 2024 UTC
-- File 20240714_105425_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000097 to 75.055173
-each separately from 0s to 75s and from 97us to 55173us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010612 in 1 packets: 18
      15.010706 in 1 packets: 36
      15.010658 in 1 packets: 54
      15.010707 in 1 packets: 72
      15.010659 in 1 packets: 90
MIU packet time from 2024-07-14T10:54:28.000.097.000 to 2024-07-14T10:55:43.055.173.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010612 in 1 packets: 18
      15.010706 in 1 packets: 36
      15.010658 in 1 packets: 54
      15.010707 in 1 packets: 72
      15.010659 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4308111 to 76.4305519
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999482 in 3 packets: 18, 72, 90
      14.9999481 in 2 packets: 36, 54
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C0 -> 35.6 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240714_105425_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4309090 to 1.4311980
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-07-14T10:54:28.000.097.000 to 2024-07-14T10:54:28.000.386.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010612

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4417570 to 16.4420460
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-07-14T10:54:43.010.998.000 to 2024-07-14T10:54:43.011.287.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4527010 to 31.4529900
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-07-14T10:54:58.021.993.000 to 2024-07-14T10:54:58.022.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4635970 to 46.4638860
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-07-14T10:55:13.032.940.000 to 2024-07-14T10:55:13.033.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4745410 to 61.4748300
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-07-14T10:55:28.043.936.000 to 2024-07-14T10:55:28.044.225.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4854370 to 76.4857260
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-07-14T10:55:43.054.884.000 to 2024-07-14T10:55:43.055.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.430909000 corrected by 40bit 10MHz counter of 1.430811100 obtained at 1Hz counter of 0. Result: 0.000097900
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-07-14T10:54:27.999.956.725
   Max 1st sample MIU time among all SL packets: 2024-07-14T10:55:43.055.032.725
   Min 1st sample XO time among all SL packets: -0.000042375
   Max 1st sample XO time among all SL packets: 75.055034018
   TM values from -9 to -1 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-07-14T10:54:27.999.956.725 (XO: -0.000042375 - 0.000263626, t=34.0 degC)  Rank: 00001
   TM values: Max =    -1  Min =    -9  Ave =    -4.37  Std =     0.57
   Min TM value:-9 at an offset of 136.038 usec -> 2024-07-14T10:54:28.000.092.762 (XO: 0.000093663)
   Max TM value:-1 at an offset of 133.550 usec -> 2024-07-14T10:54:28.000.090.275 (XO: 0.000091175)
   Max dTM/dt [1/us]:3.50e+01 at an offset of 135.775 usec -> 2024-07-14T10:54:28.000.092.500 (XO: 0.000093400)
   Min dTM/dt [1/us]:-4.50e+01 at an offset of 124.625 usec -> 2024-07-14T10:54:28.000.081.350 (XO: 0.000082250)

 - Block number: 2 starting at 2024-07-14T10:54:43.010.857.725 (XO: 15.010857463 - 15.011163464, t=35.6 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.45  Std =     0.50
   Min TM value:-6 at an offset of 93.263 usec -> 2024-07-14T10:54:43.010.950.987 (XO: 15.010950726)
   Max TM value:-3 at an offset of 9.300 usec -> 2024-07-14T10:54:43.010.867.025 (XO: 15.010866763)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 9.263 usec -> 2024-07-14T10:54:43.010.866.987 (XO: 15.010866726)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 112.838 usec -> 2024-07-14T10:54:43.010.970.562 (XO: 15.010970301)

 - Block number: 3 starting at 2024-07-14T10:54:58.021.852.725 (XO: 30.021853401 - 30.022159403, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.43  Std =     0.50
   Min TM value:-6 at an offset of 116.375 usec -> 2024-07-14T10:54:58.021.969.100 (XO: 30.021969777)
   Max TM value:-3 at an offset of 89.338 usec -> 2024-07-14T10:54:58.021.942.062 (XO: 30.021942739)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 157.351 usec -> 2024-07-14T10:54:58.022.010.075 (XO: 30.022010752)
   Min dTM/dt [1/us]:-1.10e+01 at an offset of 4.300 usec -> 2024-07-14T10:54:58.021.857.025 (XO: 30.021857702)

 - Block number: 4 starting at 2024-07-14T10:55:13.032.799.725 (XO: 45.032801337 - 45.033107338, t=35.6 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.40  Std =     0.49
   Min TM value:-6 at an offset of 138.963 usec -> 2024-07-14T10:55:13.032.938.687 (XO: 45.032940299)
   Max TM value:-3 at an offset of 8.163 usec -> 2024-07-14T10:55:13.032.807.887 (XO: 45.032809499)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 139.013 usec -> 2024-07-14T10:55:13.032.938.737 (XO: 45.032940349)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 86.413 usec -> 2024-07-14T10:55:13.032.886.137 (XO: 45.032887749)

 - Block number: 5 starting at 2024-07-14T10:55:28.043.795.725 (XO: 60.043797174 - 60.044103175, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.57  Std =     0.50
   Min TM value:-6 at an offset of 46.013 usec -> 2024-07-14T10:55:28.043.841.737 (XO: 60.043843187)
   Max TM value:-3 at an offset of 138.850 usec -> 2024-07-14T10:55:28.043.934.575 (XO: 60.043936024)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 167.413 usec -> 2024-07-14T10:55:28.043.963.138 (XO: 60.043964587)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 291.801 usec -> 2024-07-14T10:55:28.044.087.525 (XO: 60.044088975)

 - Block number: 6 starting at 2024-07-14T10:55:43.054.743.725 (XO: 75.054745017 - 75.055051018, t=35.6 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.39  Std =     0.49
   Min TM value:-6 at an offset of 84.088 usec -> 2024-07-14T10:55:43.054.827.812 (XO: 75.054829104)
   Max TM value:-3 at an offset of 60.538 usec -> 2024-07-14T10:55:43.054.804.262 (XO: 75.054805554)
   Max dTM/dt [1/us]:1.20e+01 at an offset of 162.451 usec -> 2024-07-14T10:55:43.054.906.175 (XO: 75.054907467)
   Min dTM/dt [1/us]:-1.10e+01 at an offset of 1.500 usec -> 2024-07-14T10:55:43.054.745.225 (XO: 75.054746517)
