-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Aug  6 02:11:08 2024 UTC
-- File 20240709_135611_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000097 to 15.011286
-each separately from 0s to 15s and from 97us to 11286us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
MIU packet time from 2024-07-09T13:56:14.000.097.000 to 2024-07-09T13:56:29.011.286.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4292136 to 16.4291617
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999481 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240709_135611_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4293116 to 1.4296006
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-07-09T13:56:14.000.097.000 to 2024-07-09T13:56:14.000.386.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4401596 to 16.4404486
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-07-09T13:56:29.010.997.000 to 2024-07-09T13:56:29.011.286.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.429311600 corrected by 40bit 10MHz counter of 1.429213600 obtained at 1Hz counter of 0. Result: 0.000098000
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-07-09T13:56:13.999.956.725
   Max 1st sample MIU time among all SL packets: 2024-07-09T13:56:29.011.145.725
   Min 1st sample XO time among all SL packets: -0.000042275
   Max 1st sample XO time among all SL packets: 15.011146664
   TM values from -6 to -3 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-07-09T13:56:13.999.956.725 (XO: -0.000042275 - 0.000263726, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.51  Std =     0.50
   Min TM value:-6 at an offset of 25.700 usec -> 2024-07-09T13:56:13.999.982.425 (XO: -0.000016575)
   Max TM value:-3 at an offset of 29.250 usec -> 2024-07-09T13:56:13.999.985.975 (XO: -0.000013025)
   Max dTM/dt [1/us]:1.10e+01 at an offset of 0.700 usec -> 2024-07-09T13:56:13.999.957.425 (XO: -0.000041575)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 256.488 usec -> 2024-07-09T13:56:14.000.213.213 (XO: 0.000214213)

 - Block number: 2 starting at 2024-07-09T13:56:29.010.856.725 (XO: 15.010857663 - 15.011163664, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.65  Std =     0.48
   Min TM value:-6 at an offset of 33.975 usec -> 2024-07-09T13:56:29.010.890.700 (XO: 15.010891638)
   Max TM value:-3 at an offset of 99.988 usec -> 2024-07-09T13:56:29.010.956.712 (XO: 15.010957651)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 49.425 usec -> 2024-07-09T13:56:29.010.906.150 (XO: 15.010907088)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 233.476 usec -> 2024-07-09T13:56:29.011.090.200 (XO: 15.011091139)
