-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Apr 19 19:58:33 2024 UTC
-- File 20240405_235705_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000089 to 75.055163
-each separately from 0s to 75s and from 89us to 55163us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010610 in 1 packets: 18
      15.010707 in 2 packets: 36, 72
      15.010658 in 2 packets: 54, 90
MIU packet time from 2024-04-05T23:57:08.000.089.000 to 2024-04-05T23:58:23.055.163.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010610 in 1 packets: 18
      15.010707 in 2 packets: 36, 72
      15.010658 in 2 packets: 54, 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4310760 to 76.4308183
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999484 in 3 packets: 18, 54, 90
      14.9999485 in 1 packets: 36
      14.9999486 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240405_235705_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4311658 to 1.4314548
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-04-05T23:57:08.000.089.000 to 2024-04-05T23:57:08.000.378.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010610

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4420138 to 16.4423028
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-05T23:57:23.010.988.000 to 2024-04-05T23:57:23.011.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4529578 to 31.4532468
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-05T23:57:38.021.984.000 to 2024-04-05T23:57:38.022.273.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4638538 to 46.4641428
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-05T23:57:53.032.931.000 to 2024-04-05T23:57:53.033.220.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4747978 to 61.4750868
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-05T23:58:08.043.927.000 to 2024-04-05T23:58:08.044.216.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4856938 to 76.4859828
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-04-05T23:58:23.054.874.000 to 2024-04-05T23:58:23.055.163.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.431165800 corrected by 40bit 10MHz counter of 1.431076000 obtained at 1Hz counter of 0. Result: 0.000089800
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-04-05T23:57:07.999.948.725
   Max 1st sample MIU time among all SL packets: 2024-04-05T23:58:23.055.022.725
   Min 1st sample XO time among all SL packets: -0.000050475
   Max 1st sample XO time among all SL packets: 75.055024412
   TM values from -6 to -3 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-04-05T23:57:07.999.948.725 (XO: -0.000050475 - 0.000255526, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.56  Std =     0.50
   Min TM value:-6 at an offset of 43.675 usec -> 2024-04-05T23:57:07.999.992.400 (XO: -0.000006800)
   Max TM value:-3 at an offset of 104.725 usec -> 2024-04-05T23:57:08.000.053.450 (XO: 0.000054250)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 243.801 usec -> 2024-04-05T23:57:08.000.192.525 (XO: 0.000193326)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 67.213 usec -> 2024-04-05T23:57:08.000.015.937 (XO: 0.000016738)

 - Block number: 2 starting at 2024-04-05T23:57:23.010.847.725 (XO: 15.010849162 - 15.011155163, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.46  Std =     0.50
   Min TM value:-6 at an offset of 109.150 usec -> 2024-04-05T23:57:23.010.956.875 (XO: 15.010958312)
   Max TM value:-3 at an offset of 18.563 usec -> 2024-04-05T23:57:23.010.866.287 (XO: 15.010867724)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 137.400 usec -> 2024-04-05T23:57:23.010.985.125 (XO: 15.010986562)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 109.125 usec -> 2024-04-05T23:57:23.010.956.850 (XO: 15.010958287)

 - Block number: 3 starting at 2024-04-05T23:57:38.021.843.725 (XO: 30.021844701 - 30.022150703, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.34  Std =     0.48
   Min TM value:-6 at an offset of 44.463 usec -> 2024-04-05T23:57:38.021.888.187 (XO: 30.021889164)
   Max TM value:-3 at an offset of 23.238 usec -> 2024-04-05T23:57:38.021.866.962 (XO: 30.021867939)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 262.188 usec -> 2024-04-05T23:57:38.022.105.913 (XO: 30.022106890)
   Min dTM/dt [1/us]:-1.80e+01 at an offset of 33.313 usec -> 2024-04-05T23:57:38.021.877.037 (XO: 30.021878014)

 - Block number: 4 starting at 2024-04-05T23:57:53.032.790.725 (XO: 45.032792336 - 45.033098338, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -5  Ave =    -4.43  Std =     0.50
   Min TM value:-5 at an offset of 0.000 usec -> 2024-04-05T23:57:53.032.790.725 (XO: 45.032792336)
   Max TM value:-3 at an offset of 24.763 usec -> 2024-04-05T23:57:53.032.815.487 (XO: 45.032817099)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 60.325 usec -> 2024-04-05T23:57:53.032.851.050 (XO: 45.032852662)
   Min dTM/dt [1/us]:-1.10e+01 at an offset of 1.713 usec -> 2024-04-05T23:57:53.032.792.437 (XO: 45.032794049)

 - Block number: 5 starting at 2024-04-05T23:58:08.043.786.725 (XO: 60.043787774 - 60.044093775, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -5  Ave =    -4.40  Std =     0.49
   Min TM value:-5 at an offset of 0.000 usec -> 2024-04-05T23:58:08.043.786.725 (XO: 60.043787774)
   Max TM value:-3 at an offset of 13.075 usec -> 2024-04-05T23:58:08.043.799.800 (XO: 60.043800849)
   Max dTM/dt [1/us]:1.10e+01 at an offset of 10.650 usec -> 2024-04-05T23:58:08.043.797.375 (XO: 60.043798424)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 190.963 usec -> 2024-04-05T23:58:08.043.977.688 (XO: 60.043978737)

 - Block number: 6 starting at 2024-04-05T23:58:23.054.733.725 (XO: 75.054735411 - 75.055041412, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.41  Std =     0.49
   Min TM value:-6 at an offset of 116.100 usec -> 2024-04-05T23:58:23.054.849.825 (XO: 75.054851511)
   Max TM value:-3 at an offset of 18.588 usec -> 2024-04-05T23:58:23.054.752.312 (XO: 75.054753999)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 244.438 usec -> 2024-04-05T23:58:23.054.978.163 (XO: 75.054979849)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 129.975 usec -> 2024-04-05T23:58:23.054.863.700 (XO: 75.054865387)
