-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Apr 19 18:19:50 2024 UTC
-- File 20240402_211507_1227.dat, 378  packets found.
APID:
      1227 (SL) in 378 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 378 packets (all packets)
Total number of packets: 
      18 in 378 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000095 to 300.219768
-each separately from 0s to 300s and from 95us to 219768us.
   Differences in neigboring packets: 
      0.000017 in 357 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010707 in 5 packets: 36, 72, 288, 324, 360
      15.010658 in 3 packets: 54, 306, 342
      15.010659 in 6 packets: 90, 126, 162, 198, 234, 270
      15.010706 in 5 packets: 108, 144, 180, 216, 252
MIU packet time from 2024-04-02T21:15:10.000.095.000 to 2024-04-02T21:20:10.219.768.000
   Differences in neigboring packets: 
      0.000017 in 357 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010707 in 5 packets: 36, 72, 288, 324, 360
      15.010658 in 3 packets: 54, 306, 342
      15.010659 in 6 packets: 90, 126, 162, 198, 234, 270
      15.010706 in 5 packets: 108, 144, 180, 216, 252
MIU orbit number: 
      60000 in 378 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 378 packets (all packets)
MIU orbit information / number of events: 
      0 in 378 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 378 packets (all packets)
Instrument anomalies: 
      0 (OK) in 378 packets (all packets)
FPGA configuration version: 
      211 in 378 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4373469 to 301.4363158
   Differences in neigboring packets: 
      0.0000000 in 357 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999483 in 4 packets: 18, 54, 90, 162
      14.9999484 in 6 packets: 36, 108, 144, 198, 270, 306
      14.9999486 in 3 packets: 72, 126, 180
      14.9999485 in 7 packets: 216, 234, 252, 288, 324, 342, 360
Temperature FPGA: 
      06 -> 47.2 degC in 270 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      07 -> 45.9 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      05 -> 48.5 degC in 54 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      BF -> 34.0 degC in 378 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 378 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 378 packets (all packets)
Number of external events: 
      0 in 378 packets (all packets)
Number of generated alerts: 
      0 in 378 packets (all packets)
Survey memory error: 
      0 in 378 packets (all packets)
Event memory error: 
      0 in 378 packets (all packets)
Configuration error: 
      0 in 378 packets (all packets)
SDRAM CRC error: 
      0 in 378 packets (all packets)
Radio gain: 
      0 in 378 packets (all packets)
Radio bandwidth: 
      0 in 378 packets (all packets)
Digital gain: 
      0 in 378 packets (all packets)
Analog gain: 
      0 in 378 packets (all packets)
 
SURVEY DATA READER
-- File 20240402_211507_1227.dat, 21 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4374434 to 1.4377324
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-04-02T21:15:10.000.095.000 to 2024-04-02T21:15:10.000.384.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4482914 to 16.4485804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:15:25.010.995.000 to 2024-04-02T21:15:25.011.284.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4592354 to 31.4595244
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:15:40.021.991.000 to 2024-04-02T21:15:40.022.280.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4701314 to 46.4704204
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:15:55.032.938.000 to 2024-04-02T21:15:55.033.227.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4810754 to 61.4813644
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:16:10.043.934.000 to 2024-04-02T21:16:10.044.223.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4919714 to 76.4922604
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:16:25.054.882.000 to 2024-04-02T21:16:25.055.171.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 91.5029154 to 91.5032044
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:16:40.065.877.000 to 2024-04-02T21:16:40.066.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 8 from the sequence of 18 packets.
   Block numbers in the data:
      8 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 106.5138114 to 106.5141004
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:16:55.076.825.000 to 2024-04-02T21:16:55.077.114.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 9 from the sequence of 18 packets.
   Block numbers in the data:
      9 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 121.5247554 to 121.5250444
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:17:10.087.820.000 to 2024-04-02T21:17:10.088.109.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 10 from the sequence of 18 packets.
   Block numbers in the data:
      10 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 136.5356514 to 136.5359404
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:17:25.098.768.000 to 2024-04-02T21:17:25.099.057.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 11 from the sequence of 18 packets.
   Block numbers in the data:
      11 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 151.5465954 to 151.5468844
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:17:40.109.763.000 to 2024-04-02T21:17:40.110.052.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 12 from the sequence of 18 packets.
   Block numbers in the data:
      12 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 166.5574914 to 166.5577804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:17:55.120.711.000 to 2024-04-02T21:17:55.121.000.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 13 from the sequence of 18 packets.
   Block numbers in the data:
      13 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 181.5684354 to 181.5687244
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:18:10.131.706.000 to 2024-04-02T21:18:10.131.995.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 14 from the sequence of 18 packets.
   Block numbers in the data:
      14 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 196.5793314 to 196.5796204
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:18:25.142.654.000 to 2024-04-02T21:18:25.142.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 15 from the sequence of 18 packets.
   Block numbers in the data:
      15 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 211.5902754 to 211.5905644
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:18:40.153.649.000 to 2024-04-02T21:18:40.153.938.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 16 from the sequence of 18 packets.
   Block numbers in the data:
      16 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 226.6011714 to 226.6014604
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:18:55.164.597.000 to 2024-04-02T21:18:55.164.886.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 17 from the sequence of 18 packets.
   Block numbers in the data:
      17 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 241.6121154 to 241.6124044
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:19:10.175.593.000 to 2024-04-02T21:19:10.175.882.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 18 from the sequence of 18 packets.
   Block numbers in the data:
      18 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 256.6230114 to 256.6233004
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:19:25.186.540.000 to 2024-04-02T21:19:25.186.829.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 19 from the sequence of 18 packets.
   Block numbers in the data:
      19 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 271.6339554 to 271.6342444
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-02T21:19:40.197.536.000 to 2024-04-02T21:19:40.197.825.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010658

 - Block number: 20 from the sequence of 18 packets.
   Block numbers in the data:
      20 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 286.6448514 to 286.6451404
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-02T21:19:55.208.483.000 to 2024-04-02T21:19:55.208.772.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 21 from the sequence of 18 packets.
   Block numbers in the data:
      21 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 301.6557954 to 301.6560844
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-04-02T21:20:10.219.479.000 to 2024-04-02T21:20:10.219.768.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 378 points based on 314 XO calibration points at 1Hz counter from 0 to 313
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.437443400 corrected by 40bit 10MHz counter of 1.437346900 obtained at 1Hz counter of 0. Result: 0.000096500
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-04-02T21:15:09.999.954.725
   Max 1st sample MIU time among all SL packets: 2024-04-02T21:20:10.219.627.725
   Min 1st sample XO time among all SL packets: -0.000043775
   Max 1st sample XO time among all SL packets: 300.219629094
   TM values from -6 to -3 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-04-02T21:15:09.999.954.725 (XO: -0.000043775 - 0.000262226, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.51  Std =     0.50
   Min TM value:-6 at an offset of 118.388 usec -> 2024-04-02T21:15:10.000.073.112 (XO: 0.000074613)
   Max TM value:-3 at an offset of 40.200 usec -> 2024-04-02T21:15:09.999.994.925 (XO: -0.000003575)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 292.764 usec -> 2024-04-02T21:15:10.000.247.488 (XO: 0.000248988)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 123.738 usec -> 2024-04-02T21:15:10.000.078.462 (XO: 0.000079963)

 - Block number: 2 starting at 2024-04-02T21:15:25.010.854.725 (XO: 15.010855963 - 15.011161964, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.57  Std =     0.50
   Min TM value:-6 at an offset of 33.438 usec -> 2024-04-02T21:15:25.010.888.162 (XO: 15.010889401)
   Max TM value:-3 at an offset of 142.513 usec -> 2024-04-02T21:15:25.010.997.237 (XO: 15.010998476)
   Max dTM/dt [1/us]:1.20e+01 at an offset of 33.475 usec -> 2024-04-02T21:15:25.010.888.200 (XO: 15.010889438)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 180.776 usec -> 2024-04-02T21:15:25.011.035.500 (XO: 15.011036739)

 - Block number: 3 starting at 2024-04-02T21:15:40.021.850.725 (XO: 30.021851601 - 30.022157603, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.53  Std =     0.50
   Min TM value:-6 at an offset of 80.488 usec -> 2024-04-02T21:15:40.021.931.212 (XO: 30.021932089)
   Max TM value:-3 at an offset of 107.450 usec -> 2024-04-02T21:15:40.021.958.175 (XO: 30.021959052)
   Max dTM/dt [1/us]:1.80e+01 at an offset of 257.463 usec -> 2024-04-02T21:15:40.022.108.188 (XO: 30.022109065)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 80.438 usec -> 2024-04-02T21:15:40.021.931.162 (XO: 30.021932039)

 - Block number: 4 starting at 2024-04-02T21:15:55.032.797.725 (XO: 45.032799337 - 45.033105338, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.55  Std =     0.50
   Min TM value:-6 at an offset of 14.700 usec -> 2024-04-02T21:15:55.032.812.425 (XO: 45.032814037)
   Max TM value:-3 at an offset of 32.200 usec -> 2024-04-02T21:15:55.032.829.925 (XO: 45.032831537)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 178.476 usec -> 2024-04-02T21:15:55.032.976.200 (XO: 45.032977812)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 14.650 usec -> 2024-04-02T21:15:55.032.812.375 (XO: 45.032813987)

 - Block number: 5 starting at 2024-04-02T21:16:10.043.793.725 (XO: 60.043794778 - 60.044100779, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.39  Std =     0.49
   Min TM value:-6 at an offset of 84.363 usec -> 2024-04-02T21:16:10.043.878.087 (XO: 60.043879141)
   Max TM value:-3 at an offset of 5.513 usec -> 2024-04-02T21:16:10.043.799.237 (XO: 60.043800291)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 107.550 usec -> 2024-04-02T21:16:10.043.901.275 (XO: 60.043902329)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 107.650 usec -> 2024-04-02T21:16:10.043.901.375 (XO: 60.043902429)

 - Block number: 6 starting at 2024-04-02T21:16:25.054.741.725 (XO: 75.054742511 - 75.055048512, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.55  Std =     0.50
   Min TM value:-6 at an offset of 21.800 usec -> 2024-04-02T21:16:25.054.763.525 (XO: 75.054764311)
   Max TM value:-3 at an offset of 70.700 usec -> 2024-04-02T21:16:25.054.812.425 (XO: 75.054813211)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 247.001 usec -> 2024-04-02T21:16:25.054.988.725 (XO: 75.054989512)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 257.738 usec -> 2024-04-02T21:16:25.054.999.463 (XO: 75.055000250)

 - Block number: 7 starting at 2024-04-02T21:16:40.065.736.725 (XO: 90.065738149 - 90.066044150, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.42  Std =     0.50
   Min TM value:-6 at an offset of 55.813 usec -> 2024-04-02T21:16:40.065.792.537 (XO: 90.065793961)
   Max TM value:-3 at an offset of 16.675 usec -> 2024-04-02T21:16:40.065.753.400 (XO: 90.065754824)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 174.613 usec -> 2024-04-02T21:16:40.065.911.338 (XO: 90.065912762)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 170.876 usec -> 2024-04-02T21:16:40.065.907.600 (XO: 90.065909024)

 - Block number: 8 starting at 2024-04-02T21:16:55.076.684.725 (XO: 105.076685586 - 105.076991587, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.39  Std =     0.49
   Min TM value:-6 at an offset of 90.325 usec -> 2024-04-02T21:16:55.076.775.050 (XO: 105.076775911)
   Max TM value:-3 at an offset of 30.500 usec -> 2024-04-02T21:16:55.076.715.225 (XO: 105.076716086)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 199.663 usec -> 2024-04-02T21:16:55.076.884.388 (XO: 105.076885249)
   Min dTM/dt [1/us]:-1.20e+01 at an offset of 115.113 usec -> 2024-04-02T21:16:55.076.799.837 (XO: 105.076800699)

 - Block number: 9 starting at 2024-04-02T21:17:10.087.679.725 (XO: 120.087681232 - 120.087987233, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.50  Std =     0.50
   Min TM value:-6 at an offset of 32.513 usec -> 2024-04-02T21:17:10.087.712.237 (XO: 120.087713744)
   Max TM value:-3 at an offset of 161.113 usec -> 2024-04-02T21:17:10.087.840.838 (XO: 120.087842345)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 274.563 usec -> 2024-04-02T21:17:10.087.954.288 (XO: 120.087955795)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 240.738 usec -> 2024-04-02T21:17:10.087.920.463 (XO: 120.087921970)

 - Block number: 10 starting at 2024-04-02T21:17:25.098.627.725 (XO: 135.098628960 - 135.098934961, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.60  Std =     0.49
   Min TM value:-6 at an offset of 1.088 usec -> 2024-04-02T21:17:25.098.628.812 (XO: 135.098630048)
   Max TM value:-3 at an offset of 22.950 usec -> 2024-04-02T21:17:25.098.650.675 (XO: 135.098651910)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 81.375 usec -> 2024-04-02T21:17:25.098.709.100 (XO: 135.098710336)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 159.251 usec -> 2024-04-02T21:17:25.098.786.975 (XO: 135.098788211)

 - Block number: 11 starting at 2024-04-02T21:17:40.109.622.725 (XO: 150.109624409 - 150.109930410, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.45  Std =     0.50
   Min TM value:-6 at an offset of 35.463 usec -> 2024-04-02T21:17:40.109.658.187 (XO: 150.109659871)
   Max TM value:-3 at an offset of 7.200 usec -> 2024-04-02T21:17:40.109.629.925 (XO: 150.109631609)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 237.351 usec -> 2024-04-02T21:17:40.109.860.075 (XO: 150.109861760)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 35.413 usec -> 2024-04-02T21:17:40.109.658.137 (XO: 150.109659821)

 - Block number: 12 starting at 2024-04-02T21:17:55.120.570.725 (XO: 165.120572035 - 165.120878036, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.41  Std =     0.49
   Min TM value:-6 at an offset of 69.238 usec -> 2024-04-02T21:17:55.120.639.962 (XO: 165.120641273)
   Max TM value:-3 at an offset of 1.775 usec -> 2024-04-02T21:17:55.120.572.500 (XO: 165.120573810)
   Max dTM/dt [1/us]:1.20e+01 at an offset of 1.738 usec -> 2024-04-02T21:17:55.120.572.462 (XO: 165.120573772)
   Min dTM/dt [1/us]:-1.20e+01 at an offset of 169.076 usec -> 2024-04-02T21:17:55.120.739.800 (XO: 165.120741111)

 - Block number: 13 starting at 2024-04-02T21:18:10.131.565.725 (XO: 180.131567585 - 180.131873587, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.57  Std =     0.50
   Min TM value:-6 at an offset of 8.225 usec -> 2024-04-02T21:18:10.131.573.950 (XO: 180.131575811)
   Max TM value:-3 at an offset of 99.338 usec -> 2024-04-02T21:18:10.131.665.062 (XO: 180.131666923)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 251.938 usec -> 2024-04-02T21:18:10.131.817.663 (XO: 180.131819524)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 45.413 usec -> 2024-04-02T21:18:10.131.611.137 (XO: 180.131612998)

 - Block number: 14 starting at 2024-04-02T21:18:25.142.513.725 (XO: 195.142515110 - 195.142821111, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.54  Std =     0.50
   Min TM value:-6 at an offset of 15.200 usec -> 2024-04-02T21:18:25.142.528.925 (XO: 195.142530310)
   Max TM value:-3 at an offset of 92.675 usec -> 2024-04-02T21:18:25.142.606.400 (XO: 195.142607785)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 203.288 usec -> 2024-04-02T21:18:25.142.717.013 (XO: 195.142718398)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 209.588 usec -> 2024-04-02T21:18:25.142.723.313 (XO: 195.142724698)

 - Block number: 15 starting at 2024-04-02T21:18:40.153.508.725 (XO: 210.153510662 - 210.153816663, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.48  Std =     0.50
   Min TM value:-6 at an offset of 29.300 usec -> 2024-04-02T21:18:40.153.538.025 (XO: 210.153539962)
   Max TM value:-3 at an offset of 20.825 usec -> 2024-04-02T21:18:40.153.529.550 (XO: 210.153531487)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 129.175 usec -> 2024-04-02T21:18:40.153.637.900 (XO: 210.153639838)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 20.875 usec -> 2024-04-02T21:18:40.153.529.600 (XO: 210.153531537)

 - Block number: 16 starting at 2024-04-02T21:18:55.164.456.725 (XO: 225.164458284 - 225.164764285, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.45  Std =     0.50
   Min TM value:-6 at an offset of 55.488 usec -> 2024-04-02T21:18:55.164.512.212 (XO: 225.164513772)
   Max TM value:-3 at an offset of 16.863 usec -> 2024-04-02T21:18:55.164.473.587 (XO: 225.164475147)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 55.538 usec -> 2024-04-02T21:18:55.164.512.262 (XO: 225.164513822)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 65.513 usec -> 2024-04-02T21:18:55.164.522.237 (XO: 225.164523797)

 - Block number: 17 starting at 2024-04-02T21:19:10.175.452.725 (XO: 240.175453839 - 240.175759840, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.42  Std =     0.49
   Min TM value:-6 at an offset of 1.050 usec -> 2024-04-02T21:19:10.175.453.775 (XO: 240.175454889)
   Max TM value:-3 at an offset of 9.900 usec -> 2024-04-02T21:19:10.175.462.625 (XO: 240.175463739)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 120.025 usec -> 2024-04-02T21:19:10.175.572.750 (XO: 240.175573865)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 60.675 usec -> 2024-04-02T21:19:10.175.513.400 (XO: 240.175514514)

 - Block number: 18 starting at 2024-04-02T21:19:25.186.399.725 (XO: 255.186401459 - 255.186707460, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.40  Std =     0.49
   Min TM value:-6 at an offset of 18.463 usec -> 2024-04-02T21:19:25.186.418.187 (XO: 255.186419921)
   Max TM value:-3 at an offset of 24.300 usec -> 2024-04-02T21:19:25.186.424.025 (XO: 255.186425759)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 141.900 usec -> 2024-04-02T21:19:25.186.541.625 (XO: 255.186543359)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 58.225 usec -> 2024-04-02T21:19:25.186.457.950 (XO: 255.186459684)

 - Block number: 19 starting at 2024-04-02T21:19:40.197.395.725 (XO: 270.197397016 - 270.197703017, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.57  Std =     0.50
   Min TM value:-6 at an offset of 20.838 usec -> 2024-04-02T21:19:40.197.416.562 (XO: 270.197417853)
   Max TM value:-3 at an offset of 80.538 usec -> 2024-04-02T21:19:40.197.476.262 (XO: 270.197477554)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 120.925 usec -> 2024-04-02T21:19:40.197.516.650 (XO: 270.197517941)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 31.238 usec -> 2024-04-02T21:19:40.197.426.962 (XO: 270.197428253)

 - Block number: 20 starting at 2024-04-02T21:19:55.208.342.725 (XO: 285.208344533 - 285.208650534, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.48  Std =     0.50
   Min TM value:-6 at an offset of 36.113 usec -> 2024-04-02T21:19:55.208.378.837 (XO: 285.208380646)
   Max TM value:-3 at an offset of 4.463 usec -> 2024-04-02T21:19:55.208.347.187 (XO: 285.208348996)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 221.151 usec -> 2024-04-02T21:19:55.208.563.875 (XO: 285.208565684)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 236.926 usec -> 2024-04-02T21:19:55.208.579.650 (XO: 285.208581459)

 - Block number: 21 starting at 2024-04-02T21:20:10.219.338.725 (XO: 300.219340093 - 300.219646094, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.53  Std =     0.50
   Min TM value:-6 at an offset of 59.688 usec -> 2024-04-02T21:20:10.219.398.412 (XO: 300.219399780)
   Max TM value:-3 at an offset of 250.776 usec -> 2024-04-02T21:20:10.219.589.500 (XO: 300.219590869)
   Max dTM/dt [1/us]:1.10e+01 at an offset of 2.300 usec -> 2024-04-02T21:20:10.219.341.025 (XO: 300.219342393)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 250.826 usec -> 2024-04-02T21:20:10.219.589.550 (XO: 300.219590919)
