-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Apr 19 17:49:30 2024 UTC
-- File 20240401_213443_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000093 to 75.055168
-each separately from 0s to 75s and from 93us to 55168us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010610 in 1 packets: 18
      15.010707 in 1 packets: 36
      15.010659 in 2 packets: 54, 90
      15.010706 in 1 packets: 72
MIU packet time from 2024-04-01T21:34:46.000.093.000 to 2024-04-01T21:36:01.055.168.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010610 in 1 packets: 18
      15.010707 in 1 packets: 36
      15.010659 in 2 packets: 54, 90
      15.010706 in 1 packets: 72
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4745537 to 76.4742958
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999484 in 3 packets: 18, 54, 90
      14.9999483 in 1 packets: 36
      14.9999486 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      BF -> 34.0 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240401_213443_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4746477 to 1.4749367
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-04-01T21:34:46.000.093.000 to 2024-04-01T21:34:46.000.382.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010610

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4854957 to 16.4857847
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-01T21:35:01.010.992.000 to 2024-04-01T21:35:01.011.281.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4964397 to 31.4967287
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-01T21:35:16.021.988.000 to 2024-04-01T21:35:16.022.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.5073357 to 46.5076247
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-04-01T21:35:31.032.936.000 to 2024-04-01T21:35:31.033.225.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.5182797 to 61.5185687
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-04-01T21:35:46.043.931.000 to 2024-04-01T21:35:46.044.220.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.5291757 to 76.5294647
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-04-01T21:36:01.054.879.000 to 2024-04-01T21:36:01.055.168.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.474647700 corrected by 40bit 10MHz counter of 1.474553700 obtained at 1Hz counter of 0. Result: 0.000094000
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-04-01T21:34:45.999.952.725
   Max 1st sample MIU time among all SL packets: 2024-04-01T21:36:01.055.027.725
   Min 1st sample XO time among all SL packets: -0.000046275
   Max 1st sample XO time among all SL packets: 75.055028818
   TM values from -8 to -1 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-04-01T21:34:45.999.952.725 (XO: -0.000046275 - 0.000259726, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.53  Std =     0.50
   Min TM value:-6 at an offset of 41.813 usec -> 2024-04-01T21:34:45.999.994.537 (XO: -0.000004463)
   Max TM value:-3 at an offset of 14.563 usec -> 2024-04-01T21:34:45.999.967.287 (XO: -0.000031713)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 142.575 usec -> 2024-04-01T21:34:46.000.095.300 (XO: 0.000096300)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 41.763 usec -> 2024-04-01T21:34:45.999.994.487 (XO: -0.000004513)

 - Block number: 2 starting at 2024-04-01T21:35:01.010.851.725 (XO: 15.010853363 - 15.011159364, t=34.0 degC)  Rank: 00001
   TM values: Max =    -1  Min =    -8  Ave =    -4.61  Std =     0.57
   Min TM value:-8 at an offset of 15.613 usec -> 2024-04-01T21:35:01.010.867.337 (XO: 15.010868976)
   Max TM value:-1 at an offset of 31.388 usec -> 2024-04-01T21:35:01.010.883.112 (XO: 15.010884751)
   Max dTM/dt [1/us]:3.90e+01 at an offset of 31.338 usec -> 2024-04-01T21:35:01.010.883.062 (XO: 15.010884701)
   Min dTM/dt [1/us]:-4.10e+01 at an offset of 31.513 usec -> 2024-04-01T21:35:01.010.883.237 (XO: 15.010884876)

 - Block number: 3 starting at 2024-04-01T21:35:16.021.847.725 (XO: 30.021849099 - 30.022155100, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.48  Std =     0.50
   Min TM value:-6 at an offset of 43.688 usec -> 2024-04-01T21:35:16.021.891.412 (XO: 30.021892787)
   Max TM value:-3 at an offset of 10.200 usec -> 2024-04-01T21:35:16.021.857.925 (XO: 30.021859299)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 116.288 usec -> 2024-04-01T21:35:16.021.964.012 (XO: 30.021965387)
   Min dTM/dt [1/us]:-1.10e+01 at an offset of 8.338 usec -> 2024-04-01T21:35:16.021.856.062 (XO: 30.021857437)

 - Block number: 4 starting at 2024-04-01T21:35:31.032.795.725 (XO: 45.032796737 - 45.033102738, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.38  Std =     0.49
   Min TM value:-6 at an offset of 43.300 usec -> 2024-04-01T21:35:31.032.839.025 (XO: 45.032840037)
   Max TM value:-3 at an offset of 36.913 usec -> 2024-04-01T21:35:31.032.832.637 (XO: 45.032833649)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 43.350 usec -> 2024-04-01T21:35:31.032.839.075 (XO: 45.032840087)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 54.963 usec -> 2024-04-01T21:35:31.032.850.687 (XO: 45.032851699)

 - Block number: 5 starting at 2024-04-01T21:35:46.043.790.725 (XO: 60.043792178 - 60.044098179, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.56  Std =     0.50
   Min TM value:-6 at an offset of 24.450 usec -> 2024-04-01T21:35:46.043.815.175 (XO: 60.043816628)
   Max TM value:-3 at an offset of 60.638 usec -> 2024-04-01T21:35:46.043.851.362 (XO: 60.043852816)
   Max dTM/dt [1/us]:1.20e+01 at an offset of 124.238 usec -> 2024-04-01T21:35:46.043.914.962 (XO: 60.043916416)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 303.064 usec -> 2024-04-01T21:35:46.044.093.788 (XO: 60.044095242)

 - Block number: 6 starting at 2024-04-01T21:36:01.054.738.725 (XO: 75.054739817 - 75.055045818, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.50  Std =     0.50
   Min TM value:-6 at an offset of 47.788 usec -> 2024-04-01T21:36:01.054.786.512 (XO: 75.054787604)
   Max TM value:-3 at an offset of 93.900 usec -> 2024-04-01T21:36:01.054.832.625 (XO: 75.054833717)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 47.838 usec -> 2024-04-01T21:36:01.054.786.562 (XO: 75.054787654)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 134.513 usec -> 2024-04-01T21:36:01.054.873.237 (XO: 75.054874330)
