-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Wed Apr 17 04:06:12 2024 UTC
-- File 20240303_122823_1229.dat, 4118  packets found.
APID:
      1229 (EW) in 4118 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 4118 packets (all packets)
Total number of packets: 
      18564 in 4118 packets (all packets)
Packet No from 1 to 4118
!!!ERROR: Maximum packet number must be = 18564
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 18564
Hz and MHz counters from 15.396066 to 15.448086
-each separately from 15s to 15s and from 396066us to 448086us.
   Differences in neigboring packets: 
      0.000000 in 4116 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      0.052020 in 1 packets: 3060
MIU packet time from 2024-03-03T12:28:41.396.066.000 to 2024-03-03T12:28:41.448.086.000
   Differences in neigboring packets: 
      0.000000 in 4116 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      0.052020 in 1 packets: 3060
MIU orbit number: 
      60000 in 4118 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 4118 packets (all packets)
MIU orbit information / number of events: 
      0 in 4118 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 4118 packets (all packets)
Instrument anomalies: 
      0 (OK) in 4118 packets (all packets)
FPGA configuration version: 
      211 in 4118 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.7650442 to 16.7650442
   Differences in neigboring packets: 
      0.0000000 in 4117 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 4118 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 4118 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 4118 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 4118 packets (all packets)
Number of external events: 
      1 in 3060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1058 packets: 3061, 3062, 3063, 3064, 3065, 3066, 3067, 3068, 3069, 3070, 3071, 3072, 3073, 3074, 3075, 3076, 3077, 3078, ...
Number of generated alerts: 
      1 in 3060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      7 in 1058 packets: 3061, 3062, 3063, 3064, 3065, 3066, 3067, 3068, 3069, 3070, 3071, 3072, 3073, 3074, 3075, 3076, 3077, 3078, ...
Survey memory error: 
      0 in 4118 packets (all packets)
Event memory error: 
      0 in 4118 packets (all packets)
Configuration error: 
      0 in 4118 packets (all packets)
SDRAM CRC error: 
      0 in 4118 packets (all packets)
Radio gain: 
      0 in 4118 packets (all packets)
Radio bandwidth: 
      0 in 4118 packets (all packets)
Digital gain: 
      0 in 4118 packets (all packets)
Analog gain: 
      0 in 4118 packets (all packets)

EVENT DATA READER
-- File 20240303_122823_1229.dat, 20 EW pages found. Extra packets: 38.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 2
      1 in 3060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1020 packets: 3061, 3062, 3063, 3064, 3065, 3066, 3067, 3068, 3069, 3070, 3071, 3072, 3073, 3074, 3075, 3076, 3077, 3078, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 17.1056226 to 17.1749656
      Difference between the neighboring packets: 
      0.0000170 in 4079 packets (all packets)
   Memory page number from 91 to 110
      Difference between the neighboring packets: 
      0 in 4060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 19 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 15
      Difference between the neighboring packets: 
      0 in 4060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3264, 3468, 3672, 3876
      -14 in 1 packets: 3060
   MIU event flag 
      1 in 4080 packets (all packets)
   Time triggered event flag 
      0 in 4080 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-03-03T12:28:41.340.577.631
   Min EW header abs. time found among all events            : 2024-03-03T12:28:41.396.066.000
   Max EW event (1st sample) abs. time found among all events: 2024-03-03T12:28:41.392.597.631
   Max EW header abs. time found among all events            : 2024-03-03T12:28:41.448.086.000
   TM values from -719 to 942 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 3060
    Page no. 1:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 15  ordered packets: 3060
   TM values from -161 to 174 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=15 C1MHz=0.396066
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3059 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 17.1056226
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-03-03T12:28:41.396.066.000
   Event 1st sample abs. time : 2024-03-03T12:28:41.340.577.631)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 15, 32bit 10MHz counter = 17.105622600 corrected by 40bit 10MHz counter of 14.765051200 obtained at 1Hz counter of 13. Result: 15.340579592
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              15.340579417
   Event data descriptor XO time (reconstructed seconds from XP start)    :              15.340579592
   Sampling frequency correction factor for this event:     0.999996500

 Event no. 2 total number of packets: 1020
    Page no. 1:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 5  ordered packets: 1020
   TM values from -719 to 942 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=15 C1MHz=0.448086
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1019 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 17.1576426
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-03-03T12:28:41.448.086.000
   Event 1st sample abs. time : 2024-03-03T12:28:41.392.597.631)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 15, 32bit 10MHz counter = 17.157642600 corrected by 40bit 10MHz counter of 14.765051200 obtained at 1Hz counter of 13. Result: 15.392599774
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              15.392599599
   Event data descriptor XO time (reconstructed seconds from XP start)    :              15.392599774
   Sampling frequency correction factor for this event:     0.999996500
--
Extremes in separate events:

 - Event number: 1 from 2024-03-03T12:28:41.340.577.631 to 2024-03-03T12:28:41.392.597.813 (XO: 15.340579417 - 15.392599599)
   Min TM value:-161 at an offset of 11033.964 usec -> 2024-03-03T12:28:41.351.611.594 (XO: 15.351613381)
   Max TM value:174 at an offset of 11024.276 usec -> 2024-03-03T12:28:41.351.601.907 (XO: 15.351603693)
   Max dTM/dt [1/us]:1.28e+03 at an offset of 11024.189 usec -> 2024-03-03T12:28:41.351.601.819 (XO: 15.351603606)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 11024.376 usec -> 2024-03-03T12:28:41.351.602.007 (XO: 15.351603793)
   Med: -4.00000   Ave: -4.4454196   Std: 5.9293976

 - Event number: 2 from 2024-03-03T12:28:41.392.597.631 to 2024-03-03T12:28:41.409.937.691 (XO: 15.392599599 - 15.409939660)
   Min TM value:-719 at an offset of 5463.907 usec -> 2024-03-03T12:28:41.398.061.537 (XO: 15.398063506)
   Max TM value:942 at an offset of 3222.411 usec -> 2024-03-03T12:28:41.395.820.042 (XO: 15.395822010)
   Max dTM/dt [1/us]:8.73e+03 at an offset of 5463.957 usec -> 2024-03-03T12:28:41.398.061.587 (XO: 15.398063556)
   Min dTM/dt [1/us]:-9.06e+03 at an offset of 3454.025 usec -> 2024-03-03T12:28:41.396.051.655 (XO: 15.396053624)
   Med: -4.00000   Ave: -4.4203489   Std: 17.050719
