-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Apr 17 00:05:02 2024 UTC
-- File 20240302_190713_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.646857 to 37.649208
-each separately from 10s to 37s and from 646857us to 984822us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.337387 in 1 packets: 18
      18.664097 in 1 packets: 36
MIU packet time from 2024-03-02T19:07:26.646.857.000 to 2024-03-02T19:07:53.649.208.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.337387 in 1 packets: 18
      18.664097 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6122343 to 38.6121409
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999724 in 1 packets: 18
      18.9999342 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_190713_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      115 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2590921 to 12.2593811
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3373590  -- OK for SL --
   MIU time from 2024-03-02T19:07:26.646.857.000 to 2024-03-02T19:07:26.647.146.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.337387

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      127 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5967401 to 20.5970291
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6640310  -- OK for SL --
   MIU time from 2024-03-02T19:07:34.984.533.000 to 2024-03-02T19:07:34.984.822.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.664097

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.2610601 to 39.2613491
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T19:07:53.648.919.000 to 2024-03-02T19:07:53.649.208.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.259092100 corrected by 40bit 10MHz counter of 11.612234300 obtained at 1Hz counter of 10. Result: 10.646859999
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-02T19:07:26.646.716.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T19:07:53.649.067.725
   Min 1st sample XO time among all SL packets: 10.646719724
   Max 1st sample XO time among all SL packets: 37.649070197
   TM values from -1279 to 1164 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T19:07:26.646.716.725 (XO: 10.646719724 - 10.647025725, t=35.6 degC)  Rank: 00115
   TM values: Max =  1095  Min = -1207  Ave =    -4.48  Std =    38.61
   Min TM value:-1207 at an offset of 124.438 usec -> 2024-03-02T19:07:26.646.841.162 (XO: 10.646844162)
   Max TM value:1095 at an offset of 124.500 usec -> 2024-03-02T19:07:26.646.841.225 (XO: 10.646844224)
   Max dTM/dt [1/us]:1.72e+04 at an offset of 124.488 usec -> 2024-03-02T19:07:26.646.841.212 (XO: 10.646844212)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 124.675 usec -> 2024-03-02T19:07:26.646.841.400 (XO: 10.646844399)

 - Block number: 2 starting at 2024-03-02T19:07:34.984.392.725 (XO: 18.984396570 - 18.984702571, t=35.6 degC)  Rank: 00127
   TM values: Max =  1164  Min = -1279  Ave =    -4.53  Std =    37.90
   Min TM value:-1279 at an offset of 118.375 usec -> 2024-03-02T19:07:34.984.511.100 (XO: 18.984514946)
   Max TM value:1164 at an offset of 118.438 usec -> 2024-03-02T19:07:34.984.511.162 (XO: 18.984515008)
   Max dTM/dt [1/us]:1.75e+04 at an offset of 118.425 usec -> 2024-03-02T19:07:34.984.511.150 (XO: 18.984514996)
   Min dTM/dt [1/us]:-1.33e+04 at an offset of 118.613 usec -> 2024-03-02T19:07:34.984.511.337 (XO: 18.984515183)

 - Block number: 3 starting at 2024-03-02T19:07:53.648.778.725 (XO: 37.648781196 - 37.649087197, t=35.6 degC)  Rank: 00064
   TM values: Max =   195  Min =  -228  Ave =    -4.38  Std =    20.47
   Min TM value:-228 at an offset of 121.638 usec -> 2024-03-02T19:07:53.648.900.362 (XO: 37.648902834)
   Max TM value:195 at an offset of 121.350 usec -> 2024-03-02T19:07:53.648.900.075 (XO: 37.648902546)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 121.238 usec -> 2024-03-02T19:07:53.648.899.962 (XO: 37.648902434)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 121.463 usec -> 2024-03-02T19:07:53.648.900.187 (XO: 37.648902659)
