-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 22:43:09 2024 UTC
-- File 20240302_150523_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.624418 to 29.174875
-each separately from 8s to 29s and from 174586us to 624707us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.549879 in 1 packets: 18
MIU packet time from 2024-03-02T15:05:34.624.418.000 to 2024-03-02T15:05:55.174.875.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.549879 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8533402 to 30.8532673
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999271 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_150523_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4777585 to 10.4780475
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5498070  -- OK for SL --
   MIU time from 2024-03-02T15:05:34.624.418.000 to 2024-03-02T15:05:34.624.707.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.549879

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      140 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.0278545 to 31.0281435
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T15:05:55.174.586.000 to 2024-03-02T15:05:55.174.875.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.477758500 corrected by 40bit 10MHz counter of 9.853340200 obtained at 1Hz counter of 8. Result: 8.624420485
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-02T15:05:34.624.277.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T15:05:55.174.734.725
   Min 1st sample XO time among all SL packets: 8.624280210
   Max 1st sample XO time among all SL packets: 29.174736537
   TM values from -1486 to 1241 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T15:05:34.624.277.725 (XO: 8.624280210 - 8.624586211, t=35.6 degC)  Rank: 00065
   TM values: Max =   218  Min =  -211  Ave =    -4.53  Std =    16.65
   Min TM value:-211 at an offset of 128.763 usec -> 2024-03-02T15:05:34.624.406.487 (XO: 8.624408973)
   Max TM value:218 at an offset of 128.500 usec -> 2024-03-02T15:05:34.624.406.225 (XO: 8.624408710)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 128.438 usec -> 2024-03-02T15:05:34.624.406.162 (XO: 8.624408648)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 128.600 usec -> 2024-03-02T15:05:34.624.406.325 (XO: 8.624408810)

 - Block number: 2 starting at 2024-03-02T15:05:55.174.445.725 (XO: 29.174447536 - 29.174753537, t=35.6 degC)  Rank: 00140
   TM values: Max =  1241  Min = -1486  Ave =    -4.58  Std =    39.30
   Min TM value:-1486 at an offset of 129.400 usec -> 2024-03-02T15:05:55.174.575.125 (XO: 29.174576936)
   Max TM value:1241 at an offset of 129.463 usec -> 2024-03-02T15:05:55.174.575.187 (XO: 29.174576999)
   Max dTM/dt [1/us]:1.96e+04 at an offset of 129.438 usec -> 2024-03-02T15:05:55.174.575.162 (XO: 29.174576974)
   Min dTM/dt [1/us]:-1.56e+04 at an offset of 129.638 usec -> 2024-03-02T15:05:55.174.575.362 (XO: 29.174577174)
