-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 20:52:53 2024 UTC
-- File 20240302_115145_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.960731 to 75.691598
-each separately from 14s to 75s and from 50637us to 961020us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329563 in 1 packets: 18
      10.759765 in 1 packets: 36
      8.309883 in 1 packets: 54
      18.260125 in 1 packets: 72
      11.069797 in 1 packets: 90
MIU packet time from 2024-03-02T11:52:02.960.731.000 to 2024-03-02T11:53:03.691.598.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329563 in 1 packets: 18
      10.759765 in 1 packets: 36
      8.309883 in 1 packets: 54
      18.260125 in 1 packets: 72
      11.069797 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7734714 to 76.7732605
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999549 in 1 packets: 18
      10.9999619 in 1 packets: 36
      7.9999724 in 1 packets: 54
      17.9999379 in 1 packets: 72
      10.9999620 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 108 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_115145_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      103 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7342022 to 16.7344912
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3295190  -- OK for SL --
   MIU time from 2024-03-02T11:52:02.960.731.000 to 2024-03-02T11:52:02.961.020.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.329563

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0640102 to 29.0642992
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7597270  -- OK for SL --
   MIU time from 2024-03-02T11:52:15.290.583.000 to 2024-03-02T11:52:15.290.872.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.759765

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8240262 to 39.8243152
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3098550  -- OK for SL --
   MIU time from 2024-03-02T11:52:26.050.637.000 to 2024-03-02T11:52:26.050.926.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.309883

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      102 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.1341702 to 48.1344592
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2600630  -- OK for SL --
   MIU time from 2024-03-02T11:52:34.360.809.000 to 2024-03-02T11:52:34.361.098.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.260125

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.3945222 to 66.3948112
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0697590  -- OK for SL --
   MIU time from 2024-03-02T11:52:52.621.223.000 to 2024-03-02T11:52:52.621.512.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.069797

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.4645702 to 77.4648592
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T11:53:03.691.309.000 to 2024-03-02T11:53:03.691.598.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.734202200 corrected by 40bit 10MHz counter of 15.773471400 obtained at 1Hz counter of 14. Result: 14.960734163
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-02T11:52:02.960.590.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T11:53:03.691.457.725
   Min 1st sample XO time among all SL packets: 14.960593887
   Max 1st sample XO time among all SL packets: 75.691460845
   TM values from -1182 to 995 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T11:52:02.960.590.725 (XO: 14.960593887 - 14.960899888, t=35.6 degC)  Rank: 00103
   TM values: Max =   959  Min = -1182  Ave =    -4.48  Std =    31.12
   Min TM value:-1182 at an offset of 88.175 usec -> 2024-03-02T11:52:02.960.678.900 (XO: 14.960682062)
   Max TM value:959 at an offset of 88.238 usec -> 2024-03-02T11:52:02.960.678.962 (XO: 14.960682125)
   Max dTM/dt [1/us]:1.67e+04 at an offset of 88.225 usec -> 2024-03-02T11:52:02.960.678.950 (XO: 14.960682112)
   Min dTM/dt [1/us]:-1.32e+04 at an offset of 88.425 usec -> 2024-03-02T11:52:02.960.679.150 (XO: 14.960682312)

 - Block number: 2 starting at 2024-03-02T11:52:15.290.442.725 (XO: 27.290444642 - 27.290750643, t=35.6 degC)  Rank: 00063
   TM values: Max =   201  Min =  -182  Ave =    -4.46  Std =    14.80
   Min TM value:-182 at an offset of 97.263 usec -> 2024-03-02T11:52:15.290.539.987 (XO: 27.290541904)
   Max TM value:201 at an offset of 96.938 usec -> 2024-03-02T11:52:15.290.539.662 (XO: 27.290541579)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 99.713 usec -> 2024-03-02T11:52:15.290.542.437 (XO: 27.290544354)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 97.038 usec -> 2024-03-02T11:52:15.290.539.762 (XO: 27.290541679)

 - Block number: 3 starting at 2024-03-02T11:52:26.050.496.725 (XO: 38.050497902 - 38.050803903, t=35.6 degC)  Rank: 00064
   TM values: Max =   203  Min =  -194  Ave =    -4.48  Std =    15.19
   Min TM value:-194 at an offset of 124.888 usec -> 2024-03-02T11:52:26.050.621.612 (XO: 38.050622790)
   Max TM value:203 at an offset of 124.625 usec -> 2024-03-02T11:52:26.050.621.350 (XO: 38.050622527)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 124.538 usec -> 2024-03-02T11:52:26.050.621.262 (XO: 38.050622440)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 124.725 usec -> 2024-03-02T11:52:26.050.621.450 (XO: 38.050622627)

 - Block number: 4 starting at 2024-03-02T11:52:34.360.668.725 (XO: 46.360670551 - 46.360976552, t=35.6 degC)  Rank: 00102
   TM values: Max =   995  Min = -1159  Ave =    -4.49  Std =    31.70
   Min TM value:-1159 at an offset of 87.525 usec -> 2024-03-02T11:52:34.360.756.250 (XO: 46.360758077)
   Max TM value:995 at an offset of 87.588 usec -> 2024-03-02T11:52:34.360.756.312 (XO: 46.360758139)
   Max dTM/dt [1/us]:1.75e+04 at an offset of 87.575 usec -> 2024-03-02T11:52:34.360.756.300 (XO: 46.360758127)
   Min dTM/dt [1/us]:-1.35e+04 at an offset of 87.775 usec -> 2024-03-02T11:52:34.360.756.500 (XO: 46.360758327)

 - Block number: 5 starting at 2024-03-02T11:52:52.621.082.725 (XO: 64.621085599 - 64.621391600, t=35.6 degC)  Rank: 00067
   TM values: Max =   194  Min =  -187  Ave =    -4.53  Std =    18.00
   Min TM value:-187 at an offset of 108.475 usec -> 2024-03-02T11:52:52.621.191.200 (XO: 64.621194074)
   Max TM value:194 at an offset of 110.625 usec -> 2024-03-02T11:52:52.621.193.350 (XO: 64.621196224)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 113.375 usec -> 2024-03-02T11:52:52.621.196.100 (XO: 64.621198974)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 110.725 usec -> 2024-03-02T11:52:52.621.193.450 (XO: 64.621196324)

 - Block number: 6 starting at 2024-03-02T11:53:03.691.168.725 (XO: 75.691171844 - 75.691477845, t=35.6 degC)  Rank: 00064
   TM values: Max =   191  Min =  -183  Ave =    -4.51  Std =    21.32
   Min TM value:-183 at an offset of 123.650 usec -> 2024-03-02T11:53:03.691.292.375 (XO: 75.691295495)
   Max TM value:191 at an offset of 123.338 usec -> 2024-03-02T11:53:03.691.292.062 (XO: 75.691295182)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 126.138 usec -> 2024-03-02T11:53:03.691.294.862 (XO: 75.691297982)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 180.413 usec -> 2024-03-02T11:53:03.691.349.138 (XO: 75.691352257)
