-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 20:40:23 2024 UTC
-- File 20240302_113659_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.195723 to 40.718243
-each separately from 1s to 40s and from 195723us to 718243us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.090483 in 1 packets: 18
      24.431170 in 1 packets: 36
MIU packet time from 2024-03-02T11:37:03.195.723.000 to 2024-03-02T11:37:42.718.243.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.090483 in 1 packets: 18
      24.431170 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8052690 to 41.8051341
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999481 in 1 packets: 18
      23.9999170 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_113659_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      105 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0009927 to 3.0012817
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0904310  -- OK for SL --
   MIU time from 2024-03-02T11:37:03.195.723.000 to 2024-03-02T11:37:03.196.012.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.090483

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0917127 to 18.0920017
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.4310870  -- OK for SL --
   MIU time from 2024-03-02T11:37:18.286.495.000 to 2024-03-02T11:37:18.286.784.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.431170

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      94 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.5230887 to 42.5233777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T11:37:42.717.954.000 to 2024-03-02T11:37:42.718.243.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.000992700 corrected by 40bit 10MHz counter of 2.805269000 obtained at 1Hz counter of 1. Result: 1.195724385
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-02T11:37:03.195.582.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T11:37:42.718.102.725
   Min 1st sample XO time among all SL packets: 1.195584110
   Max 1st sample XO time among all SL packets: 40.718105767
   TM values from -1076 to 960 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T11:37:03.195.582.725 (XO: 1.195584110 - 1.195890111, t=35.6 degC)  Rank: 00105
   TM values: Max =   960  Min = -1076  Ave =    -4.38  Std =    33.28
   Min TM value:-1076 at an offset of 90.638 usec -> 2024-03-02T11:37:03.195.673.362 (XO: 1.195674747)
   Max TM value:960 at an offset of 90.700 usec -> 2024-03-02T11:37:03.195.673.425 (XO: 1.195674810)
   Max dTM/dt [1/us]:1.73e+04 at an offset of 90.675 usec -> 2024-03-02T11:37:03.195.673.400 (XO: 1.195674785)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 90.875 usec -> 2024-03-02T11:37:03.195.673.600 (XO: 1.195674985)

 - Block number: 2 starting at 2024-03-02T11:37:18.286.354.725 (XO: 16.286356299 - 16.286662300, t=35.6 degC)  Rank: 00064
   TM values: Max =   204  Min =  -194  Ave =    -4.59  Std =    15.15
   Min TM value:-194 at an offset of 127.575 usec -> 2024-03-02T11:37:18.286.482.300 (XO: 16.286483874)
   Max TM value:204 at an offset of 127.275 usec -> 2024-03-02T11:37:18.286.482.000 (XO: 16.286483574)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 127.188 usec -> 2024-03-02T11:37:18.286.481.912 (XO: 16.286483487)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 127.388 usec -> 2024-03-02T11:37:18.286.482.112 (XO: 16.286483687)

 - Block number: 3 starting at 2024-03-02T11:37:42.717.813.725 (XO: 40.717816766 - 40.718122767, t=35.6 degC)  Rank: 00094
   TM values: Max =   924  Min = -1029  Ave =    -4.53  Std =    34.05
   Min TM value:-1029 at an offset of 104.950 usec -> 2024-03-02T11:37:42.717.918.675 (XO: 40.717921716)
   Max TM value:924 at an offset of 105.025 usec -> 2024-03-02T11:37:42.717.918.750 (XO: 40.717921791)
   Max dTM/dt [1/us]:1.66e+04 at an offset of 105.000 usec -> 2024-03-02T11:37:42.717.918.725 (XO: 40.717921766)
   Min dTM/dt [1/us]:-1.29e+04 at an offset of 105.200 usec -> 2024-03-02T11:37:42.717.918.925 (XO: 40.717921966)
