-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 20:01:44 2024 UTC
-- File 20240302_102553_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.822061 to 67.755346
-each separately from 10s to 67s and from 591042us to 910083us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.768692 in 1 packets: 18
      9.318463 in 1 packets: 36
      23.976801 in 1 packets: 54
      12.867884 in 1 packets: 72
MIU packet time from 2024-03-02T10:26:06.822.061.000 to 2024-03-02T10:27:03.755.346.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.768692 in 1 packets: 18
      9.318463 in 1 packets: 36
      23.976801 in 1 packets: 54
      12.867884 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8847706 to 68.8845746
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999622 in 1 packets: 18
      8.9999691 in 1 packets: 36
      23.9999174 in 1 packets: 54
      12.9999553 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      BF -> 34.0 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_102553_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.7068318 to 12.7071208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7686550  -- OK for SL --
   MIU time from 2024-03-02T10:26:06.822.061.000 to 2024-03-02T10:26:06.822.350.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.768692

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.4757758 to 23.4760648
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3184310  -- OK for SL --
   MIU time from 2024-03-02T10:26:17.591.042.000 to 2024-03-02T10:26:17.591.331.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.318463

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7944958 to 32.7947848
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.9767190  -- OK for SL --
   MIU time from 2024-03-02T10:26:26.909.794.000 to 2024-03-02T10:26:26.910.083.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.976801

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.7715038 to 56.7717928
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8678390  -- OK for SL --
   MIU time from 2024-03-02T10:26:50.886.884.000 to 2024-03-02T10:26:50.887.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.867884

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.6396318 to 69.6399208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T10:27:03.755.057.000 to 2024-03-02T10:27:03.755.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.706831800 corrected by 40bit 10MHz counter of 11.884770600 obtained at 1Hz counter of 10. Result: 10.822063995
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-02T10:26:06.821.920.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T10:27:03.755.205.725
   Min 1st sample XO time among all SL packets: 10.821923720
   Max 1st sample XO time among all SL packets: 67.755208568
   TM values from -1195 to 980 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T10:26:06.821.920.725 (XO: 10.821923720 - 10.822229721, t=35.6 degC)  Rank: 00065
   TM values: Max =   198  Min =  -181  Ave =    -4.59  Std =    15.58
   Min TM value:-181 at an offset of 125.438 usec -> 2024-03-02T10:26:06.822.046.162 (XO: 10.822049157)
   Max TM value:198 at an offset of 125.138 usec -> 2024-03-02T10:26:06.822.045.862 (XO: 10.822048857)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 125.038 usec -> 2024-03-02T10:26:06.822.045.762 (XO: 10.822048757)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 125.238 usec -> 2024-03-02T10:26:06.822.045.962 (XO: 10.822048957)

 - Block number: 2 starting at 2024-03-02T10:26:17.590.901.725 (XO: 21.590904734 - 21.591210735, t=35.6 degC)  Rank: 00091
   TM values: Max =   939  Min = -1195  Ave =    -4.54  Std =    33.12
   Min TM value:-1195 at an offset of 85.950 usec -> 2024-03-02T10:26:17.590.987.675 (XO: 21.590990684)
   Max TM value:939 at an offset of 86.013 usec -> 2024-03-02T10:26:17.590.987.737 (XO: 21.590990747)
   Max dTM/dt [1/us]:1.62e+04 at an offset of 85.988 usec -> 2024-03-02T10:26:17.590.987.712 (XO: 21.590990722)
   Min dTM/dt [1/us]:-1.25e+04 at an offset of 86.200 usec -> 2024-03-02T10:26:17.590.987.925 (XO: 21.590990934)

 - Block number: 3 starting at 2024-03-02T10:26:26.909.653.725 (XO: 30.909656718 - 30.909962719, t=35.6 degC)  Rank: 00100
   TM values: Max =   980  Min = -1068  Ave =    -4.50  Std =    31.50
   Min TM value:-1068 at an offset of 105.975 usec -> 2024-03-02T10:26:26.909.759.700 (XO: 30.909762693)
   Max TM value:980 at an offset of 106.050 usec -> 2024-03-02T10:26:26.909.759.775 (XO: 30.909762768)
   Max dTM/dt [1/us]:1.76e+04 at an offset of 106.025 usec -> 2024-03-02T10:26:26.909.759.750 (XO: 30.909762743)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 106.225 usec -> 2024-03-02T10:26:26.909.759.950 (XO: 30.909762943)

 - Block number: 4 starting at 2024-03-02T10:26:50.886.743.725 (XO: 54.886747240 - 54.887053241, t=34.0 degC)  Rank: 00064
   TM values: Max =   182  Min =  -173  Ave =    -4.47  Std =    14.91
   Min TM value:-173 at an offset of 113.550 usec -> 2024-03-02T10:26:50.886.857.275 (XO: 54.886860790)
   Max TM value:182 at an offset of 110.788 usec -> 2024-03-02T10:26:50.886.854.512 (XO: 54.886858028)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 113.625 usec -> 2024-03-02T10:26:50.886.857.350 (XO: 54.886860865)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 110.925 usec -> 2024-03-02T10:26:50.886.854.650 (XO: 54.886858165)

 - Block number: 5 starting at 2024-03-02T10:27:03.754.916.725 (XO: 67.754919567 - 67.755225568, t=35.6 degC)  Rank: 00063
   TM values: Max =   212  Min =  -196  Ave =    -4.53  Std =    16.96
   Min TM value:-196 at an offset of 112.175 usec -> 2024-03-02T10:27:03.755.028.900 (XO: 67.755031743)
   Max TM value:212 at an offset of 114.475 usec -> 2024-03-02T10:27:03.755.031.200 (XO: 67.755034043)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 114.363 usec -> 2024-03-02T10:27:03.755.031.087 (XO: 67.755033930)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 114.575 usec -> 2024-03-02T10:27:03.755.031.300 (XO: 67.755034143)
