-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 19:19:51 2024 UTC
-- File 20240302_092943_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.392506 to 40.702459
-each separately from 3s to 40s and from 382210us to 702459us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.989415 in 1 packets: 18
      16.319671 in 1 packets: 36
MIU packet time from 2024-03-02T09:29:49.392.506.000 to 2024-03-02T09:30:26.702.459.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.989415 in 1 packets: 18
      16.319671 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8495983 to 41.8494706
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999276 in 1 packets: 18
      15.9999447 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_092943_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2421048 to 5.2423938
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9893430  -- OK for SL --
   MIU time from 2024-03-02T09:29:49.392.506.000 to 2024-03-02T09:29:49.392.795.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.989415

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      95 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.2317368 to 26.2320258
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3196150  -- OK for SL --
   MIU time from 2024-03-02T09:30:10.382.210.000 to 2024-03-02T09:30:10.382.499.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.319671

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.5516408 to 42.5519298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T09:30:26.702.170.000 to 2024-03-02T09:30:26.702.459.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.242104800 corrected by 40bit 10MHz counter of 4.849598300 obtained at 1Hz counter of 3. Result: 3.392507835
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-02T09:29:49.392.365.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T09:30:26.702.318.725
   Min 1st sample XO time among all SL packets: 3.392367559
   Max 1st sample XO time among all SL packets: 40.702321313
   TM values from -1099 to 927 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T09:29:49.392.365.725 (XO: 3.392367559 - 3.392673560, t=35.6 degC)  Rank: 00065
   TM values: Max =   178  Min =  -179  Ave =    -4.58  Std =    15.59
   Min TM value:-179 at an offset of 118.638 usec -> 2024-03-02T09:29:49.392.484.362 (XO: 3.392486197)
   Max TM value:178 at an offset of 106.163 usec -> 2024-03-02T09:29:49.392.471.887 (XO: 3.392473722)
   Max dTM/dt [1/us]:1.27e+03 at an offset of 106.075 usec -> 2024-03-02T09:29:49.392.471.800 (XO: 3.392473634)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 106.275 usec -> 2024-03-02T09:29:49.392.472.000 (XO: 3.392473834)

 - Block number: 2 starting at 2024-03-02T09:30:10.382.069.725 (XO: 24.382071924 - 24.382377925, t=35.6 degC)  Rank: 00095
   TM values: Max =   921  Min = -1042  Ave =    -4.52  Std =    32.11
   Min TM value:-1042 at an offset of 94.475 usec -> 2024-03-02T09:30:10.382.164.200 (XO: 24.382166399)
   Max TM value:921 at an offset of 94.550 usec -> 2024-03-02T09:30:10.382.164.275 (XO: 24.382166474)
   Max dTM/dt [1/us]:1.70e+04 at an offset of 94.525 usec -> 2024-03-02T09:30:10.382.164.250 (XO: 24.382166449)
   Min dTM/dt [1/us]:-1.27e+04 at an offset of 94.725 usec -> 2024-03-02T09:30:10.382.164.450 (XO: 24.382166649)

 - Block number: 3 starting at 2024-03-02T09:30:26.702.029.725 (XO: 40.702032312 - 40.702338313, t=35.6 degC)  Rank: 00090
   TM values: Max =   927  Min = -1099  Ave =    -4.58  Std =    29.40
   Min TM value:-1099 at an offset of 126.438 usec -> 2024-03-02T09:30:26.702.156.162 (XO: 40.702158750)
   Max TM value:927 at an offset of 126.500 usec -> 2024-03-02T09:30:26.702.156.225 (XO: 40.702158812)
   Max dTM/dt [1/us]:1.53e+04 at an offset of 126.488 usec -> 2024-03-02T09:30:26.702.156.212 (XO: 40.702158800)
   Min dTM/dt [1/us]:-1.23e+04 at an offset of 126.688 usec -> 2024-03-02T09:30:26.702.156.412 (XO: 40.702159000)
