-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 16:21:35 2024 UTC
-- File 20240302_035628_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.422189 to 40.762646
-each separately from 1s to 40s and from 122307us to 762646us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.699829 in 1 packets: 18
      14.639761 in 1 packets: 36
MIU packet time from 2024-03-02T03:56:32.422.189.000 to 2024-03-02T03:57:11.762.646.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.699829 in 1 packets: 18
      14.639761 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.2436280 to 42.2434928
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999135 in 1 packets: 18
      13.9999513 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      BF -> 34.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240302_035628_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6658177 to 3.6661067
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.6997430  -- OK for SL --
   MIU time from 2024-03-02T03:56:32.422.189.000 to 2024-03-02T03:56:32.422.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.699829

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3658497 to 28.3661387
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6397110  -- OK for SL --
   MIU time from 2024-03-02T03:56:57.122.307.000 to 2024-03-02T03:56:57.122.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.639761

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0058497 to 43.0061387
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-02T03:57:11.762.357.000 to 2024-03-02T03:57:11.762.646.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.665817700 corrected by 40bit 10MHz counter of 3.243628000 obtained at 1Hz counter of 1. Result: 1.422191178
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-02T03:56:32.422.048.725
   Max 1st sample MIU time among all SL packets: 2024-03-02T03:57:11.762.505.725
   Min 1st sample XO time among all SL packets: 1.422050902
   Max 1st sample XO time among all SL packets: 40.762508294
   TM values from -1055 to 772 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-02T03:56:32.422.048.725 (XO: 1.422050902 - 1.422356903, t=34.0 degC)  Rank: 00082
   TM values: Max =   772  Min = -1053  Ave =    -4.43  Std =    27.57
   Min TM value:-1053 at an offset of 100.925 usec -> 2024-03-02T03:56:32.422.149.650 (XO: 1.422151828)
   Max TM value:772 at an offset of 101.000 usec -> 2024-03-02T03:56:32.422.149.725 (XO: 1.422151903)
   Max dTM/dt [1/us]:1.55e+04 at an offset of 100.975 usec -> 2024-03-02T03:56:32.422.149.700 (XO: 1.422151878)
   Min dTM/dt [1/us]:-1.21e+04 at an offset of 101.188 usec -> 2024-03-02T03:56:32.422.149.912 (XO: 1.422152090)

 - Block number: 2 starting at 2024-03-02T03:56:57.122.166.725 (XO: 26.122168353 - 26.122474354, t=34.0 degC)  Rank: 00086
   TM values: Max =   771  Min = -1055  Ave =    -4.59  Std =    32.73
   Min TM value:-1055 at an offset of 121.388 usec -> 2024-03-02T03:56:57.122.288.112 (XO: 26.122289741)
   Max TM value:771 at an offset of 121.450 usec -> 2024-03-02T03:56:57.122.288.175 (XO: 26.122289803)
   Max dTM/dt [1/us]:1.62e+04 at an offset of 121.425 usec -> 2024-03-02T03:56:57.122.288.150 (XO: 26.122289778)
   Min dTM/dt [1/us]:-1.23e+04 at an offset of 121.638 usec -> 2024-03-02T03:56:57.122.288.362 (XO: 26.122289991)

 - Block number: 3 starting at 2024-03-02T03:57:11.762.216.725 (XO: 40.762219293 - 40.762525294, t=34.0 degC)  Rank: 00077
   TM values: Max =   737  Min = -1032  Ave =    -4.45  Std =    28.90
   Min TM value:-1032 at an offset of 88.613 usec -> 2024-03-02T03:57:11.762.305.337 (XO: 40.762307906)
   Max TM value:737 at an offset of 88.688 usec -> 2024-03-02T03:57:11.762.305.412 (XO: 40.762307981)
   Max dTM/dt [1/us]:1.54e+04 at an offset of 88.663 usec -> 2024-03-02T03:57:11.762.305.387 (XO: 40.762307956)
   Min dTM/dt [1/us]:-1.20e+04 at an offset of 88.875 usec -> 2024-03-02T03:57:11.762.305.600 (XO: 40.762308168)
