-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 13:56:33 2024 UTC
-- File 20240301_234538_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.120948 to 101.142447
-each separately from 1s to 101s and from 120948us to 582602us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.246471 in 1 packets: 18
      27.214316 in 1 packets: 36
      10.898101 in 1 packets: 54
      12.667628 in 1 packets: 72
      18.406142 in 1 packets: 90
      14.586818 in 1 packets: 108
MIU packet time from 2024-03-01T23:45:42.120.948.000 to 2024-03-01T23:47:22.142.447.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.246471 in 1 packets: 18
      27.214316 in 1 packets: 36
      10.898101 in 1 packets: 54
      12.667628 in 1 packets: 72
      18.406142 in 1 packets: 90
      14.586818 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.6719163 to 102.6715704
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999448 in 1 packets: 18
      26.9999066 in 1 packets: 36
      10.9999619 in 1 packets: 54
      12.9999550 in 1 packets: 72
      17.9999376 in 1 packets: 90
      14.9999482 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      07 -> 45.9 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C0 -> 35.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_234538_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      55 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7928656 to 2.7931546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.2464150  -- OK for SL --
   MIU time from 2024-03-01T23:45:42.120.948.000 to 2024-03-01T23:45:42.121.237.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.246471

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.0395696 to 19.0398586
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.2142230  -- OK for SL --
   MIU time from 2024-03-01T23:45:58.367.708.000 to 2024-03-01T23:45:58.367.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.214316

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.2540816 to 46.2543706
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8980630  -- OK for SL --
   MIU time from 2024-03-01T23:46:25.582.313.000 to 2024-03-01T23:46:25.582.602.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.898101

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.1524336 to 57.1527226
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6675830  -- OK for SL --
   MIU time from 2024-03-01T23:46:36.480.703.000 to 2024-03-01T23:46:36.480.992.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.667628

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      56 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.8203056 to 69.8205946
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4060790  -- OK for SL --
   MIU time from 2024-03-01T23:46:49.148.620.000 to 2024-03-01T23:46:49.148.909.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.406142

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 88.2266736 to 88.2269626
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5867670  -- OK for SL --
   MIU time from 2024-03-01T23:47:07.555.051.000 to 2024-03-01T23:47:07.555.340.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.586818

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 102.8137296 to 102.8140186
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T23:47:22.142.158.000 to 2024-03-01T23:47:22.142.447.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.792865600 corrected by 40bit 10MHz counter of 2.671916300 obtained at 1Hz counter of 1. Result: 1.120949711
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-01T23:45:42.120.807.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T23:47:22.142.306.725
   Min 1st sample XO time among all SL packets: 1.120809436
   Max 1st sample XO time among all SL packets: 101.142308423
   TM values from -1157 to 756 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T23:45:42.120.807.725 (XO: 1.120809436 - 1.121115437, t=35.6 degC)  Rank: 00055
   TM values: Max =   191  Min =  -157  Ave =    -4.51  Std =    16.02
   Min TM value:-157 at an offset of 124.700 usec -> 2024-03-01T23:45:42.120.932.425 (XO: 1.120934136)
   Max TM value:191 at an offset of 151.738 usec -> 2024-03-01T23:45:42.120.959.463 (XO: 1.120961174)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 151.613 usec -> 2024-03-01T23:45:42.120.959.338 (XO: 1.120961049)
   Min dTM/dt [1/us]:-1.44e+03 at an offset of 151.838 usec -> 2024-03-01T23:45:42.120.959.563 (XO: 1.120961274)

 - Block number: 2 starting at 2024-03-01T23:45:58.367.567.725 (XO: 17.367569511 - 17.367875513, t=35.6 degC)  Rank: 00079
   TM values: Max =   756  Min = -1140  Ave =    -4.55  Std =    29.17
   Min TM value:-1140 at an offset of 91.438 usec -> 2024-03-01T23:45:58.367.659.162 (XO: 17.367660949)
   Max TM value:756 at an offset of 91.500 usec -> 2024-03-01T23:45:58.367.659.225 (XO: 17.367661012)
   Max dTM/dt [1/us]:1.70e+04 at an offset of 91.475 usec -> 2024-03-01T23:45:58.367.659.200 (XO: 17.367660987)
   Min dTM/dt [1/us]:-1.30e+04 at an offset of 91.700 usec -> 2024-03-01T23:45:58.367.659.425 (XO: 17.367661212)

 - Block number: 3 starting at 2024-03-01T23:46:25.582.172.725 (XO: 44.582175663 - 44.582481664, t=35.6 degC)  Rank: 00087
   TM values: Max =   751  Min = -1155  Ave =    -4.58  Std =    28.54
   Min TM value:-1155 at an offset of 122.425 usec -> 2024-03-01T23:46:25.582.295.150 (XO: 44.582298088)
   Max TM value:751 at an offset of 122.500 usec -> 2024-03-01T23:46:25.582.295.225 (XO: 44.582298163)
   Max dTM/dt [1/us]:1.63e+04 at an offset of 122.463 usec -> 2024-03-01T23:46:25.582.295.187 (XO: 44.582298126)
   Min dTM/dt [1/us]:-1.31e+04 at an offset of 122.688 usec -> 2024-03-01T23:46:25.582.295.412 (XO: 44.582298351)

 - Block number: 4 starting at 2024-03-01T23:46:36.480.562.725 (XO: 55.480565359 - 55.480871360, t=35.6 degC)  Rank: 00058
   TM values: Max =   187  Min =  -180  Ave =    -4.49  Std =    17.85
   Min TM value:-180 at an offset of 122.725 usec -> 2024-03-01T23:46:36.480.685.450 (XO: 55.480688084)
   Max TM value:187 at an offset of 122.138 usec -> 2024-03-01T23:46:36.480.684.862 (XO: 55.480687497)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 109.725 usec -> 2024-03-01T23:46:36.480.672.450 (XO: 55.480675084)
   Min dTM/dt [1/us]:-1.39e+03 at an offset of 124.675 usec -> 2024-03-01T23:46:36.480.687.400 (XO: 55.480690034)

 - Block number: 5 starting at 2024-03-01T23:46:49.148.479.725 (XO: 68.148481230 - 68.148787231, t=35.6 degC)  Rank: 00056
   TM values: Max =   178  Min =  -146  Ave =    -4.52  Std =    19.21
   Min TM value:-146 at an offset of 117.275 usec -> 2024-03-01T23:46:49.148.597.000 (XO: 68.148598505)
   Max TM value:178 at an offset of 114.613 usec -> 2024-03-01T23:46:49.148.594.337 (XO: 68.148595843)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 114.538 usec -> 2024-03-01T23:46:49.148.594.262 (XO: 68.148595768)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 114.750 usec -> 2024-03-01T23:46:49.148.594.475 (XO: 68.148595980)

 - Block number: 6 starting at 2024-03-01T23:47:07.554.910.725 (XO: 86.554913067 - 86.555219068, t=35.6 degC)  Rank: 00081
   TM values: Max =   749  Min = -1157  Ave =    -4.48  Std =    28.27
   Min TM value:-1157 at an offset of 97.488 usec -> 2024-03-01T23:47:07.555.008.212 (XO: 86.555010555)
   Max TM value:749 at an offset of 97.713 usec -> 2024-03-01T23:47:07.555.008.437 (XO: 86.555010780)
   Max dTM/dt [1/us]:1.72e+04 at an offset of 97.525 usec -> 2024-03-01T23:47:07.555.008.250 (XO: 86.555010593)
   Min dTM/dt [1/us]:-1.29e+04 at an offset of 97.750 usec -> 2024-03-01T23:47:07.555.008.475 (XO: 86.555010818)

 - Block number: 7 starting at 2024-03-01T23:47:22.142.017.725 (XO: 101.142019422 - 101.142325423, t=35.6 degC)  Rank: 00061
   TM values: Max =   191  Min =  -175  Ave =    -4.58  Std =    20.24
   Min TM value:-175 at an offset of 124.500 usec -> 2024-03-01T23:47:22.142.142.225 (XO: 101.142143923)
   Max TM value:191 at an offset of 123.900 usec -> 2024-03-01T23:47:22.142.141.625 (XO: 101.142143323)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 123.838 usec -> 2024-03-01T23:47:22.142.141.562 (XO: 101.142143260)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 124.050 usec -> 2024-03-01T23:47:22.142.141.775 (XO: 101.142143473)
