-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 13:09:27 2024 UTC
-- File 20240301_213640_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.718740 to 17.538396
-each separately from 1s to 17s and from 538107us to 719029us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.819078 in 1 packets: 18
MIU packet time from 2024-03-01T21:36:44.718.740.000 to 2024-03-01T21:37:00.538.396.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.819078 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.6186320 to 18.6185765
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999445 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_213640_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3373722 to 3.3376612
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8190230  -- OK for SL --
   MIU time from 2024-03-01T21:36:44.718.740.000 to 2024-03-01T21:36:44.719.029.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.819078

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1566842 to 19.1569732
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T21:37:00.538.107.000 to 2024-03-01T21:37:00.538.396.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.337372200 corrected by 40bit 10MHz counter of 2.618632000 obtained at 1Hz counter of 1. Result: 1.718742716
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-01T21:36:44.718.599.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T21:37:00.538.255.725
   Min 1st sample XO time among all SL packets: 1.718602440
   Max 1st sample XO time among all SL packets: 17.538258309
   TM values from -1156 to 785 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T21:36:44.718.599.725 (XO: 1.718602440 - 1.718908441, t=35.6 degC)  Rank: 00085
   TM values: Max =   785  Min = -1156  Ave =    -4.46  Std =    30.74
   Min TM value:-1156 at an offset of 117.763 usec -> 2024-03-01T21:36:44.718.717.487 (XO: 1.718720203)
   Max TM value:785 at an offset of 118.000 usec -> 2024-03-01T21:36:44.718.717.725 (XO: 1.718720441)
   Max dTM/dt [1/us]:1.75e+04 at an offset of 117.813 usec -> 2024-03-01T21:36:44.718.717.537 (XO: 1.718720253)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 118.025 usec -> 2024-03-01T21:36:44.718.717.750 (XO: 1.718720466)

 - Block number: 2 starting at 2024-03-01T21:37:00.537.966.725 (XO: 17.537969308 - 17.538275309, t=35.6 degC)  Rank: 00057
   TM values: Max =   176  Min =  -162  Ave =    -4.47  Std =    18.80
   Min TM value:-162 at an offset of 130.975 usec -> 2024-03-01T21:37:00.538.097.700 (XO: 17.538100283)
   Max TM value:176 at an offset of 128.288 usec -> 2024-03-01T21:37:00.538.095.012 (XO: 17.538097596)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 128.225 usec -> 2024-03-01T21:37:00.538.094.950 (XO: 17.538097533)
   Min dTM/dt [1/us]:-1.37e+03 at an offset of 128.450 usec -> 2024-03-01T21:37:00.538.095.175 (XO: 17.538097758)
