-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 10:57:56 2024 UTC
-- File 20240301_154806_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.616380 to 24.796926
-each separately from 10s to 24s and from 616380us to 796926us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.179968 in 1 packets: 18
MIU packet time from 2024-03-01T15:48:19.616.380.000 to 2024-03-01T15:48:33.796.926.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.179968 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8173697 to 25.8173213
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999516 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_154806_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.4337504 to 12.4340394
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1799190  -- OK for SL --
   MIU time from 2024-03-01T15:48:19.616.380.000 to 2024-03-01T15:48:19.616.669.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.179968

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      104 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6139584 to 26.6142474
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T15:48:33.796.637.000 to 2024-03-01T15:48:33.796.926.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.433750400 corrected by 40bit 10MHz counter of 11.817369700 obtained at 1Hz counter of 10. Result: 10.616382857
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-01T15:48:19.616.239.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T15:48:33.796.785.725
   Min 1st sample XO time among all SL packets: 10.616242582
   Max 1st sample XO time among all SL packets: 24.796788614
   TM values from -1463 to 993 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T15:48:19.616.239.725 (XO: 10.616242582 - 10.616548583, t=35.6 degC)  Rank: 00080
   TM values: Max =   908  Min = -1396  Ave =    -4.46  Std =    38.10
   Min TM value:-1396 at an offset of 96.313 usec -> 2024-03-01T15:48:19.616.336.037 (XO: 10.616338895)
   Max TM value:908 at an offset of 96.538 usec -> 2024-03-01T15:48:19.616.336.262 (XO: 10.616339120)
   Max dTM/dt [1/us]:1.88e+04 at an offset of 96.363 usec -> 2024-03-01T15:48:19.616.336.087 (XO: 10.616338945)
   Min dTM/dt [1/us]:-1.31e+04 at an offset of 96.588 usec -> 2024-03-01T15:48:19.616.336.312 (XO: 10.616339170)

 - Block number: 2 starting at 2024-03-01T15:48:33.796.496.725 (XO: 24.796499613 - 24.796805614, t=35.6 degC)  Rank: 00104
   TM values: Max =   993  Min = -1463  Ave =    -4.46  Std =    34.92
   Min TM value:-1463 at an offset of 124.475 usec -> 2024-03-01T15:48:33.796.621.200 (XO: 24.796624088)
   Max TM value:993 at an offset of 124.713 usec -> 2024-03-01T15:48:33.796.621.437 (XO: 24.796624326)
   Max dTM/dt [1/us]:2.00e+04 at an offset of 124.525 usec -> 2024-03-01T15:48:33.796.621.250 (XO: 24.796624138)
   Min dTM/dt [1/us]:-1.39e+04 at an offset of 124.750 usec -> 2024-03-01T15:48:33.796.621.475 (XO: 24.796624363)
