-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 10:56:11 2024 UTC
-- File 20240301_154613_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.492610 to 41.424285
-each separately from 1s to 41s and from 423996us to 853508us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.360320 in 1 packets: 18
      11.570488 in 1 packets: 36
MIU packet time from 2024-03-01T15:46:17.492.610.000 to 2024-03-01T15:46:57.424.285.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.360320 in 1 packets: 18
      11.570488 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9020073 to 42.9018690
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999032 in 1 packets: 18
      11.9999585 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_154613_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3946175 to 3.3949065
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.3602230  -- OK for SL --
   MIU time from 2024-03-01T15:46:17.492.610.000 to 2024-03-01T15:46:17.492.899.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.360320

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7551295 to 31.7554185
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5704470  -- OK for SL --
   MIU time from 2024-03-01T15:46:45.853.219.000 to 2024-03-01T15:46:45.853.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.570488

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.3258655 to 43.3261545
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T15:46:57.423.996.000 to 2024-03-01T15:46:57.424.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.394617500 corrected by 40bit 10MHz counter of 2.902007300 obtained at 1Hz counter of 1. Result: 1.492611924
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-01T15:46:17.492.469.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T15:46:57.424.144.725
   Min 1st sample XO time among all SL packets: 1.492471649
   Max 1st sample XO time among all SL packets: 41.424146710
   TM values from -1411 to 948 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T15:46:17.492.469.725 (XO: 1.492471649 - 1.492777650, t=35.6 degC)  Rank: 00100
   TM values: Max =   948  Min = -1411  Ave =    -4.57  Std =    31.05
   Min TM value:-1411 at an offset of 125.588 usec -> 2024-03-01T15:46:17.492.595.312 (XO: 1.492597237)
   Max TM value:948 at an offset of 125.825 usec -> 2024-03-01T15:46:17.492.595.550 (XO: 1.492597474)
   Max dTM/dt [1/us]:1.97e+04 at an offset of 125.638 usec -> 2024-03-01T15:46:17.492.595.362 (XO: 1.492597287)
   Min dTM/dt [1/us]:-1.31e+04 at an offset of 125.863 usec -> 2024-03-01T15:46:17.492.595.587 (XO: 1.492597512)

 - Block number: 2 starting at 2024-03-01T15:46:45.853.078.725 (XO: 29.853081625 - 29.853387627, t=35.6 degC)  Rank: 00067
   TM values: Max =   225  Min =  -203  Ave =    -4.56  Std =    18.29
   Min TM value:-203 at an offset of 124.425 usec -> 2024-03-01T15:46:45.853.203.150 (XO: 29.853206051)
   Max TM value:225 at an offset of 123.813 usec -> 2024-03-01T15:46:45.853.202.537 (XO: 29.853205438)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 123.713 usec -> 2024-03-01T15:46:45.853.202.437 (XO: 29.853205338)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 126.400 usec -> 2024-03-01T15:46:45.853.205.125 (XO: 29.853208026)

 - Block number: 3 starting at 2024-03-01T15:46:57.423.855.725 (XO: 41.423857709 - 41.424163710, t=35.6 degC)  Rank: 00068
   TM values: Max =   217  Min =  -191  Ave =    -4.50  Std =    19.27
   Min TM value:-191 at an offset of 122.113 usec -> 2024-03-01T15:46:57.423.977.837 (XO: 41.423979821)
   Max TM value:217 at an offset of 121.525 usec -> 2024-03-01T15:46:57.423.977.250 (XO: 41.423979234)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 109.288 usec -> 2024-03-01T15:46:57.423.965.012 (XO: 41.423966996)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 121.650 usec -> 2024-03-01T15:46:57.423.977.375 (XO: 41.423979359)
