-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 10:53:04 2024 UTC
-- File 20240301_154259_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.362420 to 33.691699
-each separately from 6s to 33s and from 92055us to 691699us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.729346 in 1 packets: 18
      17.599066 in 1 packets: 36
MIU packet time from 2024-03-01T15:43:08.362.420.000 to 2024-03-01T15:43:35.691.699.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.729346 in 1 packets: 18
      17.599066 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7219706 to 34.7218772
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999654 in 1 packets: 18
      16.9999412 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_154259_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.0843918 to 8.0846808
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7293110  -- OK for SL --
   MIU time from 2024-03-01T15:43:08.362.420.000 to 2024-03-01T15:43:08.362.709.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.729346

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      98 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8139918 to 17.8142808
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5990070  -- OK for SL --
   MIU time from 2024-03-01T15:43:18.092.055.000 to 2024-03-01T15:43:18.092.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.599066

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      97 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4132878 to 35.4135768
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T15:43:35.691.410.000 to 2024-03-01T15:43:35.691.699.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.084391800 corrected by 40bit 10MHz counter of 7.721970600 obtained at 1Hz counter of 6. Result: 6.362422468
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-01T15:43:08.362.279.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T15:43:35.691.558.725
   Min 1st sample XO time among all SL packets: 6.362282193
   Max 1st sample XO time among all SL packets: 33.691561745
   TM values from -1551 to 1012 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T15:43:08.362.279.725 (XO: 6.362282193 - 6.362588194, t=35.6 degC)  Rank: 00065
   TM values: Max =   256  Min =  -189  Ave =    -4.44  Std =    20.17
   Min TM value:-189 at an offset of 131.113 usec -> 2024-03-01T15:43:08.362.410.837 (XO: 6.362413306)
   Max TM value:256 at an offset of 130.488 usec -> 2024-03-01T15:43:08.362.410.212 (XO: 6.362412681)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 130.363 usec -> 2024-03-01T15:43:08.362.410.087 (XO: 6.362412556)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 130.588 usec -> 2024-03-01T15:43:08.362.410.312 (XO: 6.362412781)

 - Block number: 2 starting at 2024-03-01T15:43:18.091.914.725 (XO: 16.091915847 - 16.092221848, t=35.6 degC)  Rank: 00098
   TM values: Max =  1012  Min = -1551  Ave =    -4.39  Std =    35.13
   Min TM value:-1551 at an offset of 105.750 usec -> 2024-03-01T15:43:18.092.020.475 (XO: 16.092021597)
   Max TM value:1012 at an offset of 105.975 usec -> 2024-03-01T15:43:18.092.020.700 (XO: 16.092021822)
   Max dTM/dt [1/us]:2.06e+04 at an offset of 105.800 usec -> 2024-03-01T15:43:18.092.020.525 (XO: 16.092021647)
   Min dTM/dt [1/us]:-1.44e+04 at an offset of 106.025 usec -> 2024-03-01T15:43:18.092.020.750 (XO: 16.092021872)

 - Block number: 3 starting at 2024-03-01T15:43:35.691.269.725 (XO: 33.691272744 - 33.691578746, t=35.6 degC)  Rank: 00097
   TM values: Max =  1010  Min = -1531  Ave =    -4.48  Std =    36.68
   Min TM value:-1531 at an offset of 117.088 usec -> 2024-03-01T15:43:35.691.386.812 (XO: 33.691389832)
   Max TM value:1010 at an offset of 117.313 usec -> 2024-03-01T15:43:35.691.387.037 (XO: 33.691390057)
   Max dTM/dt [1/us]:2.05e+04 at an offset of 117.125 usec -> 2024-03-01T15:43:35.691.386.850 (XO: 33.691389870)
   Min dTM/dt [1/us]:-1.41e+04 at an offset of 117.350 usec -> 2024-03-01T15:43:35.691.387.075 (XO: 33.691390095)
