-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 10:20:37 2024 UTC
-- File 20240301_144128_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.458015 to 59.516839
-each separately from 11s to 59s and from 458015us to 738144us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.159325 in 1 packets: 18
      5.119937 in 1 packets: 36
      24.778406 in 1 packets: 54
MIU packet time from 2024-03-01T14:41:42.458.015.000 to 2024-03-01T14:42:30.516.839.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.159325 in 1 packets: 18
      5.119937 in 1 packets: 36
      24.778406 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8822767 to 60.8821098
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999375 in 1 packets: 18
      4.9999825 in 1 packets: 36
      24.9999131 in 1 packets: 54
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 37 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 37 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_144128_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      97 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3402922 to 13.3405812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1592630  -- OK for SL --
   MIU time from 2024-03-01T14:41:42.458.015.000 to 2024-03-01T14:41:42.458.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.159325

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      62 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4998442 to 31.5001332
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1199190  -- OK for SL --
   MIU time from 2024-03-01T14:42:00.617.629.000 to 2024-03-01T14:42:00.617.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.119937

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.6200522 to 36.6203412
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.7783190  -- OK for SL --
   MIU time from 2024-03-01T14:42:05.737.855.000 to 2024-03-01T14:42:05.738.144.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.778406

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      94 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.3986602 to 61.3989492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T14:42:30.516.550.000 to 2024-03-01T14:42:30.516.839.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.340292200 corrected by 40bit 10MHz counter of 12.882276700 obtained at 1Hz counter of 11. Result: 11.458017103
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-03-01T14:41:42.457.874.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T14:42:30.516.698.725
   Min 1st sample XO time among all SL packets: 11.457876828
   Max 1st sample XO time among all SL packets: 59.516700933
   TM values from -1426 to 946 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T14:41:42.457.874.725 (XO: 11.457876828 - 11.458182829, t=35.6 degC)  Rank: 00097
   TM values: Max =   911  Min = -1380  Ave =    -4.48  Std =    33.89
   Min TM value:-1380 at an offset of 119.675 usec -> 2024-03-01T14:41:42.457.994.400 (XO: 11.457996503)
   Max TM value:911 at an offset of 119.900 usec -> 2024-03-01T14:41:42.457.994.625 (XO: 11.457996728)
   Max dTM/dt [1/us]:1.90e+04 at an offset of 119.725 usec -> 2024-03-01T14:41:42.457.994.450 (XO: 11.457996553)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 119.950 usec -> 2024-03-01T14:41:42.457.994.675 (XO: 11.457996778)

 - Block number: 2 starting at 2024-03-01T14:42:00.617.488.725 (XO: 29.617491886 - 29.617797887, t=35.6 degC)  Rank: 00062
   TM values: Max =   216  Min =  -199  Ave =    -4.50  Std =    16.69
   Min TM value:-199 at an offset of 123.763 usec -> 2024-03-01T14:42:00.617.612.487 (XO: 29.617615649)
   Max TM value:216 at an offset of 121.150 usec -> 2024-03-01T14:42:00.617.609.875 (XO: 29.617613037)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 121.063 usec -> 2024-03-01T14:42:00.617.609.787 (XO: 29.617612949)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 123.650 usec -> 2024-03-01T14:42:00.617.612.375 (XO: 29.617615537)

 - Block number: 3 starting at 2024-03-01T14:42:05.737.714.725 (XO: 34.737717733 - 34.738023734, t=35.6 degC)  Rank: 00063
   TM values: Max =   231  Min =  -198  Ave =    -4.51  Std =    14.92
   Min TM value:-198 at an offset of 121.825 usec -> 2024-03-01T14:42:05.737.836.550 (XO: 34.737839559)
   Max TM value:231 at an offset of 119.238 usec -> 2024-03-01T14:42:05.737.833.962 (XO: 34.737836971)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 119.125 usec -> 2024-03-01T14:42:05.737.833.850 (XO: 34.737836859)
   Min dTM/dt [1/us]:-1.52e+03 at an offset of 116.913 usec -> 2024-03-01T14:42:05.737.831.637 (XO: 34.737834646)

 - Block number: 4 starting at 2024-03-01T14:42:30.516.409.725 (XO: 59.516411932 - 59.516717934, t=35.6 degC)  Rank: 00094
   TM values: Max =   946  Min = -1426  Ave =    -4.59  Std =    34.80
   Min TM value:-1426 at an offset of 113.100 usec -> 2024-03-01T14:42:30.516.522.825 (XO: 59.516525033)
   Max TM value:946 at an offset of 113.338 usec -> 2024-03-01T14:42:30.516.523.062 (XO: 59.516525270)
   Max dTM/dt [1/us]:1.88e+04 at an offset of 113.138 usec -> 2024-03-01T14:42:30.516.522.862 (XO: 59.516525070)
   Min dTM/dt [1/us]:-1.36e+04 at an offset of 113.375 usec -> 2024-03-01T14:42:30.516.523.100 (XO: 59.516525308)
