-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr 16 09:38:09 2024 UTC
-- File 20240301_132441_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.138842 to 28.797355
-each separately from 5s to 28s and from 138842us to 797355us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.657935 in 1 packets: 18
MIU packet time from 2024-03-01T13:24:49.138.842.000 to 2024-03-01T13:25:12.797.355.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.657935 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8268018 to 29.8267225
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999207 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240301_132441_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      97 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9656447 to 6.9659337
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.6578550  -- OK for SL --
   MIU time from 2024-03-01T13:24:49.138.842.000 to 2024-03-01T13:24:49.139.131.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.657935

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.6237887 to 30.6240777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-03-01T13:25:12.797.066.000 to 2024-03-01T13:25:12.797.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.965644700 corrected by 40bit 10MHz counter of 6.826801800 obtained at 1Hz counter of 5. Result: 5.138843386
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-03-01T13:24:49.138.701.725
   Max 1st sample MIU time among all SL packets: 2024-03-01T13:25:12.797.214.725
   Min 1st sample XO time among all SL packets: 5.138703110
   Max 1st sample XO time among all SL packets: 28.797217715
   TM values from -1305 to 905 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-03-01T13:24:49.138.701.725 (XO: 5.138703110 - 5.139009112, t=35.6 degC)  Rank: 00097
   TM values: Max =   878  Min = -1305  Ave =    -4.41  Std =    32.19
   Min TM value:-1305 at an offset of 113.075 usec -> 2024-03-01T13:24:49.138.814.800 (XO: 5.138816186)
   Max TM value:878 at an offset of 113.313 usec -> 2024-03-01T13:24:49.138.815.037 (XO: 5.138816423)
   Max dTM/dt [1/us]:1.85e+04 at an offset of 113.125 usec -> 2024-03-01T13:24:49.138.814.850 (XO: 5.138816236)
   Min dTM/dt [1/us]:-1.35e+04 at an offset of 113.350 usec -> 2024-03-01T13:24:49.138.815.075 (XO: 5.138816461)

 - Block number: 2 starting at 2024-03-01T13:25:12.796.925.725 (XO: 28.796928714 - 28.797234715, t=35.6 degC)  Rank: 00084
   TM values: Max =   905  Min = -1255  Ave =    -4.44  Std =    28.69
   Min TM value:-1255 at an offset of 108.388 usec -> 2024-03-01T13:25:12.797.034.112 (XO: 28.797037102)
   Max TM value:905 at an offset of 108.625 usec -> 2024-03-01T13:25:12.797.034.350 (XO: 28.797037340)
   Max dTM/dt [1/us]:1.86e+04 at an offset of 108.438 usec -> 2024-03-01T13:25:12.797.034.162 (XO: 28.797037152)
   Min dTM/dt [1/us]:-1.34e+04 at an offset of 108.663 usec -> 2024-03-01T13:25:12.797.034.387 (XO: 28.797037377)
