-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Feb 24 01:02:26 2024 UTC
-- File 20240223_052236_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.555482 to 16.606035
-each separately from 9s to 16s and from 555482us to 606035us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.049975 in 1 packets: 18
MIU packet time from 2024-02-23T05:22:48.555.482.000 to 2024-02-23T05:22:55.606.035.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.049975 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8595657 to 17.8595415
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999758 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240223_052236_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.4150480 to 11.4153370
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.0499510  -- OK for SL --
   MIU time from 2024-02-23T05:22:48.555.482.000 to 2024-02-23T05:22:48.555.771.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.049975

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4652880 to 18.4655770
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-23T05:22:55.605.746.000 to 2024-02-23T05:22:55.606.035.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.415048000 corrected by 40bit 10MHz counter of 10.859565700 obtained at 1Hz counter of 9. Result: 9.555484244
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-02-23T05:22:48.555.341.725
   Max 1st sample MIU time among all SL packets: 2024-02-23T05:22:55.605.894.725
   Min 1st sample XO time among all SL packets: 9.555343969
   Max 1st sample XO time among all SL packets: 16.605897346
   TM values from -275 to 284 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-23T05:22:48.555.341.725 (XO: 9.555343969 - 9.555649970, t=35.6 degC)  Rank: 00081
   TM values: Max =   263  Min =  -234  Ave =    -4.48  Std =    21.33
   Min TM value:-234 at an offset of 126.638 usec -> 2024-02-23T05:22:48.555.468.362 (XO: 9.555470607)
   Max TM value:263 at an offset of 123.975 usec -> 2024-02-23T05:22:48.555.465.700 (XO: 9.555467944)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 126.763 usec -> 2024-02-23T05:22:48.555.468.487 (XO: 9.555470732)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 124.113 usec -> 2024-02-23T05:22:48.555.465.837 (XO: 9.555468082)

 - Block number: 2 starting at 2024-02-23T05:22:55.605.605.725 (XO: 16.605608345 - 16.605914346, t=35.6 degC)  Rank: 00081
   TM values: Max =   284  Min =  -275  Ave =    -4.52  Std =    30.74
   Min TM value:-275 at an offset of 206.238 usec -> 2024-02-23T05:22:55.605.811.963 (XO: 16.605814583)
   Max TM value:284 at an offset of 206.001 usec -> 2024-02-23T05:22:55.605.811.725 (XO: 16.605814345)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 120.463 usec -> 2024-02-23T05:22:55.605.726.187 (XO: 16.605728808)
   Min dTM/dt [1/us]:-2.92e+03 at an offset of 120.663 usec -> 2024-02-23T05:22:55.605.726.387 (XO: 16.605729008)
