-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Feb 17 03:11:41 2024 UTC
-- File 20240216_141056_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.011977 to 85.753724
-each separately from 1s to 85s and from 11977us to 823294us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.300238 in 1 packets: 18
      10.990021 in 1 packets: 36
      18.519902 in 1 packets: 54
      14.859987 in 1 packets: 72
      17.069865 in 1 packets: 90
MIU packet time from 2024-02-16T14:11:00.011.977.000 to 2024-02-16T14:12:24.753.724.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.300238 in 1 packets: 18
      10.990021 in 1 packets: 36
      18.519902 in 1 packets: 54
      14.859987 in 1 packets: 72
      17.069865 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.5489778 to 86.5486877
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999207 in 1 packets: 18
      10.9999620 in 1 packets: 36
      17.9999379 in 1 packets: 54
      14.9999481 in 1 packets: 72
      16.9999412 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C0 -> 35.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240216_141056_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5609556 to 2.5612446
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3001590  -- OK for SL --
   MIU time from 2024-02-16T14:11:00.011.977.000 to 2024-02-16T14:11:00.012.266.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.300238

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8614036 to 25.8616926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9899830  -- OK for SL --
   MIU time from 2024-02-16T14:11:23.312.504.000 to 2024-02-16T14:11:23.312.793.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.990021

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.8516756 to 36.8519646
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5198390  -- OK for SL --
   MIU time from 2024-02-16T14:11:34.302.814.000 to 2024-02-16T14:11:34.303.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.519902

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.3718036 to 55.3720926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8599350  -- OK for SL --
   MIU time from 2024-02-16T14:11:52.823.005.000 to 2024-02-16T14:11:52.823.294.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.859987

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.2320276 to 70.2323166
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0698070  -- OK for SL --
   MIU time from 2024-02-16T14:12:07.683.281.000 to 2024-02-16T14:12:07.683.570.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.069865

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 87.3021236 to 87.3024126
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-16T14:12:24.753.435.000 to 2024-02-16T14:12:24.753.724.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.560955600 corrected by 40bit 10MHz counter of 2.548977800 obtained at 1Hz counter of 1. Result: 1.011977841
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-16T14:11:00.011.836.725
   Max 1st sample MIU time among all SL packets: 2024-02-16T14:12:24.753.583.725
   Min 1st sample XO time among all SL packets: 1.011837565
   Max 1st sample XO time among all SL packets: 85.753587187
   TM values from -265 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-16T14:11:00.011.836.725 (XO: 1.011837565 - 1.012143566, t=35.6 degC)  Rank: 00081
   TM values: Max =   216  Min =  -265  Ave =    -4.44  Std =    21.40
   Min TM value:-265 at an offset of 120.050 usec -> 2024-02-16T14:11:00.011.956.775 (XO: 1.011957616)
   Max TM value:216 at an offset of 107.788 usec -> 2024-02-16T14:11:00.011.944.512 (XO: 1.011945353)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 105.638 usec -> 2024-02-16T14:11:00.011.942.362 (XO: 1.011943203)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 107.913 usec -> 2024-02-16T14:11:00.011.944.637 (XO: 1.011945478)

 - Block number: 2 starting at 2024-02-16T14:11:23.312.363.725 (XO: 24.312365918 - 24.312671919, t=35.6 degC)  Rank: 00078
   TM values: Max =   226  Min =  -251  Ave =    -4.54  Std =    18.15
   Min TM value:-251 at an offset of 122.600 usec -> 2024-02-16T14:11:23.312.486.325 (XO: 24.312488519)
   Max TM value:226 at an offset of 122.300 usec -> 2024-02-16T14:11:23.312.486.025 (XO: 24.312488219)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 125.088 usec -> 2024-02-16T14:11:23.312.488.812 (XO: 24.312491006)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 122.450 usec -> 2024-02-16T14:11:23.312.486.175 (XO: 24.312488369)

 - Block number: 3 starting at 2024-02-16T14:11:34.302.673.725 (XO: 35.302675884 - 35.302981885, t=35.6 degC)  Rank: 00078
   TM values: Max =   188  Min =  -233  Ave =    -4.50  Std =    18.96
   Min TM value:-233 at an offset of 124.300 usec -> 2024-02-16T14:11:34.302.798.025 (XO: 35.302800185)
   Max TM value:188 at an offset of 121.613 usec -> 2024-02-16T14:11:34.302.795.337 (XO: 35.302797497)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 124.400 usec -> 2024-02-16T14:11:34.302.798.125 (XO: 35.302800285)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 121.738 usec -> 2024-02-16T14:11:34.302.795.462 (XO: 35.302797622)

 - Block number: 4 starting at 2024-02-16T14:11:52.822.864.725 (XO: 53.822867805 - 53.823173806, t=35.6 degC)  Rank: 00078
   TM values: Max =   213  Min =  -225  Ave =    -4.50  Std =    22.04
   Min TM value:-225 at an offset of 124.538 usec -> 2024-02-16T14:11:52.822.989.262 (XO: 53.822992343)
   Max TM value:213 at an offset of 121.888 usec -> 2024-02-16T14:11:52.822.986.612 (XO: 53.822989693)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 124.650 usec -> 2024-02-16T14:11:52.822.989.375 (XO: 53.822992455)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 122.013 usec -> 2024-02-16T14:11:52.822.986.737 (XO: 53.822989818)

 - Block number: 5 starting at 2024-02-16T14:12:07.683.140.725 (XO: 68.683143216 - 68.683449217, t=35.6 degC)  Rank: 00082
   TM values: Max =   173  Min =  -242  Ave =    -4.54  Std =    22.13
   Min TM value:-242 at an offset of 122.100 usec -> 2024-02-16T14:12:07.683.262.825 (XO: 68.683265316)
   Max TM value:173 at an offset of 119.300 usec -> 2024-02-16T14:12:07.683.260.025 (XO: 68.683262516)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 122.163 usec -> 2024-02-16T14:12:07.683.262.887 (XO: 68.683265379)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 119.525 usec -> 2024-02-16T14:12:07.683.260.250 (XO: 68.683262741)

 - Block number: 6 starting at 2024-02-16T14:12:24.753.294.725 (XO: 85.753298186 - 85.753604187, t=35.6 degC)  Rank: 00078
   TM values: Max =   223  Min =  -262  Ave =    -4.46  Std =    17.54
   Min TM value:-262 at an offset of 119.363 usec -> 2024-02-16T14:12:24.753.414.087 (XO: 85.753417549)
   Max TM value:223 at an offset of 119.100 usec -> 2024-02-16T14:12:24.753.413.825 (XO: 85.753417287)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 121.850 usec -> 2024-02-16T14:12:24.753.416.575 (XO: 85.753420037)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 119.225 usec -> 2024-02-16T14:12:24.753.413.950 (XO: 85.753417412)
