-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Feb 17 01:56:32 2024 UTC
-- File 20240216_120257_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.783712 to 48.133461
-each separately from 14s to 48s and from 133172us to 784001us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.679579 in 1 packets: 18
      5.709572 in 1 packets: 36
      14.959442 in 1 packets: 54
MIU packet time from 2024-02-16T12:03:14.783.712.000 to 2024-02-16T12:03:48.133.461.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.679579 in 1 packets: 18
      5.709572 in 1 packets: 36
      14.959442 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.6861076 to 49.6859905
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999552 in 1 packets: 18
      5.9999794 in 1 packets: 36
      14.9999483 in 1 packets: 54
Temperature FPGA: 
      06 -> 47.2 degC in 72 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240216_120257_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4698198 to 16.4701088
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6795350  -- OK for SL --
   MIU time from 2024-02-16T12:03:14.783.712.000 to 2024-02-16T12:03:14.784.001.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.679579

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.1496438 to 29.1499328
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.7095510  -- OK for SL --
   MIU time from 2024-02-16T12:03:27.463.580.000 to 2024-02-16T12:03:27.463.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.709572

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8594838 to 34.8597728
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9593910  -- OK for SL --
   MIU time from 2024-02-16T12:03:33.173.441.000 to 2024-02-16T12:03:33.173.730.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.959442

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.8191638 to 49.8194528
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-16T12:03:48.133.172.000 to 2024-02-16T12:03:48.133.461.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.469819800 corrected by 40bit 10MHz counter of 15.686107600 obtained at 1Hz counter of 14. Result: 14.783714865
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-16T12:03:14.783.571.725
   Max 1st sample MIU time among all SL packets: 2024-02-16T12:03:48.133.320.725
   Min 1st sample XO time among all SL packets: 14.783574589
   Max 1st sample XO time among all SL packets: 48.133322478
   TM values from -265 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-16T12:03:14.783.571.725 (XO: 14.783574589 - 14.783880590, t=35.6 degC)  Rank: 00080
   TM values: Max =   227  Min =  -253  Ave =    -4.46  Std =    21.14
   Min TM value:-253 at an offset of 110.138 usec -> 2024-02-16T12:03:14.783.681.862 (XO: 14.783684727)
   Max TM value:227 at an offset of 109.850 usec -> 2024-02-16T12:03:14.783.681.575 (XO: 14.783684440)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 153.651 usec -> 2024-02-16T12:03:14.783.725.375 (XO: 14.783728240)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 109.963 usec -> 2024-02-16T12:03:14.783.681.687 (XO: 14.783684552)

 - Block number: 2 starting at 2024-02-16T12:03:27.463.439.725 (XO: 27.463442301 - 27.463748302, t=35.6 degC)  Rank: 00082
   TM values: Max =   232  Min =  -265  Ave =    -4.45  Std =    18.07
   Min TM value:-265 at an offset of 122.350 usec -> 2024-02-16T12:03:27.463.562.075 (XO: 27.463564651)
   Max TM value:232 at an offset of 122.088 usec -> 2024-02-16T12:03:27.463.561.812 (XO: 27.463564389)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 121.988 usec -> 2024-02-16T12:03:27.463.561.712 (XO: 27.463564289)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 122.188 usec -> 2024-02-16T12:03:27.463.561.912 (XO: 27.463564489)

 - Block number: 3 starting at 2024-02-16T12:03:33.173.300.725 (XO: 33.173301932 - 33.173607933, t=35.6 degC)  Rank: 00085
   TM values: Max =   189  Min =  -234  Ave =    -4.46  Std =    19.30
   Min TM value:-234 at an offset of 118.625 usec -> 2024-02-16T12:03:33.173.419.350 (XO: 33.173420557)
   Max TM value:189 at an offset of 105.800 usec -> 2024-02-16T12:03:33.173.406.525 (XO: 33.173407732)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 121.138 usec -> 2024-02-16T12:03:33.173.421.862 (XO: 33.173423069)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 105.913 usec -> 2024-02-16T12:03:33.173.406.637 (XO: 33.173407844)

 - Block number: 4 starting at 2024-02-16T12:03:48.133.031.725 (XO: 48.133033477 - 48.133339478, t=35.6 degC)  Rank: 00078
   TM values: Max =   201  Min =  -213  Ave =    -4.49  Std =    14.56
   Min TM value:-213 at an offset of 107.488 usec -> 2024-02-16T12:03:48.133.139.212 (XO: 48.133140965)
   Max TM value:201 at an offset of 107.163 usec -> 2024-02-16T12:03:48.133.138.887 (XO: 48.133140640)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 109.850 usec -> 2024-02-16T12:03:48.133.141.575 (XO: 48.133143328)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 107.275 usec -> 2024-02-16T12:03:48.133.139.000 (XO: 48.133140753)
