-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Feb 14 00:29:24 2024 UTC
-- File 20240213_152232_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.855034 to 19.035035
-each separately from 14s to 19s and from 34746us to 855323us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.179423 in 1 packets: 18
MIU packet time from 2024-02-13T15:22:49.855.034.000 to 2024-02-13T15:22:54.035.035.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.179423 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.1106311 to 21.1106138
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999827 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240213_152232_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9656651 to 16.9659541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1794070  -- OK for SL --
   MIU time from 2024-02-13T15:22:49.855.034.000 to 2024-02-13T15:22:49.855.323.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.179423

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1453611 to 21.1456501
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-13T15:22:54.034.746.000 to 2024-02-13T15:22:54.035.035.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.965665100 corrected by 40bit 10MHz counter of 16.110631100 obtained at 1Hz counter of 14. Result: 14.855036907
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-13T15:22:49.854.893.725
   Max 1st sample MIU time among all SL packets: 2024-02-13T15:22:54.034.894.725
   Min 1st sample XO time among all SL packets: 14.854896632
   Max 1st sample XO time among all SL packets: 19.034896144
   TM values from -242 to 212 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-13T15:22:49.854.893.725 (XO: 14.854896632 - 14.855202633, t=35.6 degC)  Rank: 00079
   TM values: Max =   212  Min =  -242  Ave =    -4.54  Std =    21.14
   Min TM value:-242 at an offset of 126.200 usec -> 2024-02-13T15:22:49.855.019.925 (XO: 14.855022832)
   Max TM value:212 at an offset of 113.988 usec -> 2024-02-13T15:22:49.855.007.712 (XO: 14.855010620)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 113.875 usec -> 2024-02-13T15:22:49.855.007.600 (XO: 14.855010507)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 114.075 usec -> 2024-02-13T15:22:49.855.007.800 (XO: 14.855010707)

 - Block number: 2 starting at 2024-02-13T15:22:54.034.605.725 (XO: 19.034607143 - 19.034913144, t=35.6 degC)  Rank: 00079
   TM values: Max =   212  Min =  -238  Ave =    -4.50  Std =    19.28
   Min TM value:-238 at an offset of 126.775 usec -> 2024-02-13T15:22:54.034.732.500 (XO: 19.034733918)
   Max TM value:212 at an offset of 126.400 usec -> 2024-02-13T15:22:54.034.732.125 (XO: 19.034733543)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 112.450 usec -> 2024-02-13T15:22:54.034.718.175 (XO: 19.034719593)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 126.513 usec -> 2024-02-13T15:22:54.034.732.237 (XO: 19.034733656)
