-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 12 07:11:22 2024 UTC
-- File 20240209_033559_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.480272 to 28.141054
-each separately from 1s to 28s and from 140765us to 480561us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.660204 in 1 packets: 18
MIU packet time from 2024-02-09T03:36:03.480.272.000 to 2024-02-09T03:36:30.141.054.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.660204 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8569693 to 29.8568764
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999071 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240209_033559_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3372425 to 3.3375315
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.6601110  -- OK for SL --
   MIU time from 2024-02-09T03:36:03.480.272.000 to 2024-02-09T03:36:03.480.561.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.660204

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9976425 to 29.9979315
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-09T03:36:30.140.765.000 to 2024-02-09T03:36:30.141.054.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.337242500 corrected by 40bit 10MHz counter of 2.856969300 obtained at 1Hz counter of 1. Result: 1.480274833
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-09T03:36:03.480.131.725
   Max 1st sample MIU time among all SL packets: 2024-02-09T03:36:30.140.913.725
   Min 1st sample XO time among all SL packets: 1.480134557
   Max 1st sample XO time among all SL packets: 28.140915304
   TM values from -291 to 264 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-09T03:36:03.480.131.725 (XO: 1.480134557 - 1.480440558, t=35.6 degC)  Rank: 00090
   TM values: Max =   264  Min =  -291  Ave =    -4.58  Std =    26.99
   Min TM value:-291 at an offset of 127.575 usec -> 2024-02-09T03:36:03.480.259.300 (XO: 1.480262133)
   Max TM value:264 at an offset of 127.325 usec -> 2024-02-09T03:36:03.480.259.050 (XO: 1.480261883)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 127.250 usec -> 2024-02-09T03:36:03.480.258.975 (XO: 1.480261808)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 127.425 usec -> 2024-02-09T03:36:03.480.259.150 (XO: 1.480261983)

 - Block number: 2 starting at 2024-02-09T03:36:30.140.624.725 (XO: 28.140626303 - 28.140932304, t=35.6 degC)  Rank: 00081
   TM values: Max =   252  Min =  -268  Ave =    -4.49  Std =    26.79
   Min TM value:-268 at an offset of 130.925 usec -> 2024-02-09T03:36:30.140.755.650 (XO: 28.140757229)
   Max TM value:252 at an offset of 130.625 usec -> 2024-02-09T03:36:30.140.755.350 (XO: 28.140756929)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 128.375 usec -> 2024-02-09T03:36:30.140.753.100 (XO: 28.140754679)
   Min dTM/dt [1/us]:-3.18e+03 at an offset of 130.738 usec -> 2024-02-09T03:36:30.140.755.462 (XO: 28.140757041)
