-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 12 06:28:02 2024 UTC
-- File 20240209_015347_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.311055 to 33.519757
-each separately from 6s to 33s and from 311055us to 910229us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.598596 in 1 packets: 18
      6.609239 in 1 packets: 36
MIU packet time from 2024-02-09T01:53:56.311.055.000 to 2024-02-09T01:54:23.519.757.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.598596 in 1 packets: 18
      6.609239 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8910870 to 34.8909942
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999313 in 1 packets: 18
      6.9999759 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240209_015347_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2021427 to 8.2024317
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5985270  -- OK for SL --
   MIU time from 2024-02-09T01:53:56.311.055.000 to 2024-02-09T01:53:56.311.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.598596

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8009587 to 28.8012477
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6092150  -- OK for SL --
   MIU time from 2024-02-09T01:54:16.909.940.000 to 2024-02-09T01:54:16.910.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.609239

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4104627 to 35.4107517
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-09T01:54:23.519.468.000 to 2024-02-09T01:54:23.519.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.202142700 corrected by 40bit 10MHz counter of 7.891087000 obtained at 1Hz counter of 6. Result: 6.311056758
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-09T01:53:56.310.914.725
   Max 1st sample MIU time among all SL packets: 2024-02-09T01:54:23.519.616.725
   Min 1st sample XO time among all SL packets: 6.310916482
   Max 1st sample XO time among all SL packets: 33.519619044
   TM values from -266 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-09T01:53:56.310.914.725 (XO: 6.310916482 - 6.311222483, t=35.6 degC)  Rank: 00080
   TM values: Max =   235  Min =  -255  Ave =    -4.48  Std =    20.99
   Min TM value:-255 at an offset of 122.713 usec -> 2024-02-09T01:53:56.311.037.437 (XO: 6.311039195)
   Max TM value:235 at an offset of 122.450 usec -> 2024-02-09T01:53:56.311.037.175 (XO: 6.311038933)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 122.375 usec -> 2024-02-09T01:53:56.311.037.100 (XO: 6.311038858)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 122.575 usec -> 2024-02-09T01:53:56.311.037.300 (XO: 6.311039058)

 - Block number: 2 starting at 2024-02-09T01:54:16.909.799.725 (XO: 26.909803218 - 26.910109219, t=35.6 degC)  Rank: 00085
   TM values: Max =   230  Min =  -266  Ave =    -4.56  Std =    32.74
   Min TM value:-266 at an offset of 179.013 usec -> 2024-02-09T01:54:16.909.978.738 (XO: 26.909982231)
   Max TM value:230 at an offset of 206.401 usec -> 2024-02-09T01:54:16.910.006.125 (XO: 26.910009619)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 126.350 usec -> 2024-02-09T01:54:16.909.926.075 (XO: 26.909929569)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 192.501 usec -> 2024-02-09T01:54:16.909.992.225 (XO: 26.909995719)

 - Block number: 3 starting at 2024-02-09T01:54:23.519.327.725 (XO: 33.519330043 - 33.519636044, t=35.6 degC)  Rank: 00082
   TM values: Max =   207  Min =  -227  Ave =    -4.51  Std =    30.13
   Min TM value:-227 at an offset of 125.863 usec -> 2024-02-09T01:54:23.519.453.587 (XO: 33.519455906)
   Max TM value:207 at an offset of 123.125 usec -> 2024-02-09T01:54:23.519.450.850 (XO: 33.519453168)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 125.963 usec -> 2024-02-09T01:54:23.519.453.687 (XO: 33.519456006)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 123.263 usec -> 2024-02-09T01:54:23.519.450.987 (XO: 33.519453306)
