-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 18:12:27 2024 UTC
-- File 20240207_132931_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.987670 to 28.878242
-each separately from 6s to 28s and from 877953us to 987959us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.889994 in 1 packets: 18
MIU packet time from 2024-02-07T13:29:40.987.670.000 to 2024-02-07T13:30:02.878.242.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.889994 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7840996 to 29.7840239
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999243 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240207_132931_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7717693 to 8.7720583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8899190  -- OK for SL --
   MIU time from 2024-02-07T13:29:40.987.670.000 to 2024-02-07T13:29:40.987.959.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.889994

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.6619773 to 30.6622663
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-07T13:30:02.877.953.000 to 2024-02-07T13:30:02.878.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.771769300 corrected by 40bit 10MHz counter of 7.784099600 obtained at 1Hz counter of 6. Result: 6.987673157
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-07T13:29:40.987.529.725
   Max 1st sample MIU time among all SL packets: 2024-02-07T13:30:02.878.101.725
   Min 1st sample XO time among all SL packets: 6.987532881
   Max 1st sample XO time among all SL packets: 28.878105111
   TM values from -249 to 206 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-07T13:29:40.987.529.725 (XO: 6.987532881 - 6.987838882, t=35.6 degC)  Rank: 00082
   TM values: Max =   196  Min =  -233  Ave =    -4.46  Std =    21.84
   Min TM value:-233 at an offset of 122.613 usec -> 2024-02-07T13:29:40.987.652.337 (XO: 6.987655494)
   Max TM value:196 at an offset of 122.338 usec -> 2024-02-07T13:29:40.987.652.062 (XO: 6.987655219)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 125.150 usec -> 2024-02-07T13:29:40.987.654.875 (XO: 6.987658032)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 122.450 usec -> 2024-02-07T13:29:40.987.652.175 (XO: 6.987655332)

 - Block number: 2 starting at 2024-02-07T13:30:02.877.812.725 (XO: 28.877816110 - 28.878122111, t=35.6 degC)  Rank: 00079
   TM values: Max =   206  Min =  -249  Ave =    -4.58  Std =    20.14
   Min TM value:-249 at an offset of 123.675 usec -> 2024-02-07T13:30:02.877.936.400 (XO: 28.877939785)
   Max TM value:206 at an offset of 123.363 usec -> 2024-02-07T13:30:02.877.936.087 (XO: 28.877939472)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 126.163 usec -> 2024-02-07T13:30:02.877.938.887 (XO: 28.877942273)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 123.475 usec -> 2024-02-07T13:30:02.877.936.200 (XO: 28.877939585)
