-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 17:44:16 2024 UTC
-- File 20240207_125139_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.434209 to 29.624837
-each separately from 0s to 29s and from 434209us to 624837us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.190050 in 1 packets: 18
MIU packet time from 2024-02-07T12:51:42.434.209.000 to 2024-02-07T12:52:11.624.837.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.190050 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8872042 to 30.8871047
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.9999005 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240207_125139_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3214136 to 2.3217026
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.1899510  -- OK for SL --
   MIU time from 2024-02-07T12:51:42.434.209.000 to 2024-02-07T12:51:42.434.498.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.190050

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5116536 to 31.5119426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-07T12:52:11.624.548.000 to 2024-02-07T12:52:11.624.837.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.321413600 corrected by 40bit 10MHz counter of 1.887204200 obtained at 1Hz counter of 0. Result: 0.434210876
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-07T12:51:42.434.068.725
   Max 1st sample MIU time among all SL packets: 2024-02-07T12:52:11.624.696.725
   Min 1st sample XO time among all SL packets: 0.434070601
   Max 1st sample XO time among all SL packets: 29.624699749
   TM values from -250 to 178 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-07T12:51:42.434.068.725 (XO: 0.434070601 - 0.434376602, t=35.6 degC)  Rank: 00077
   TM values: Max =   178  Min =  -216  Ave =    -4.47  Std =    22.33
   Min TM value:-216 at an offset of 123.225 usec -> 2024-02-07T12:51:42.434.191.950 (XO: 0.434193826)
   Max TM value:178 at an offset of 120.463 usec -> 2024-02-07T12:51:42.434.189.187 (XO: 0.434191064)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 123.325 usec -> 2024-02-07T12:51:42.434.192.050 (XO: 0.434193926)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 120.638 usec -> 2024-02-07T12:51:42.434.189.362 (XO: 0.434191239)

 - Block number: 2 starting at 2024-02-07T12:52:11.624.407.725 (XO: 29.624410748 - 29.624716749, t=35.6 degC)  Rank: 00079
   TM values: Max =   177  Min =  -250  Ave =    -4.54  Std =    19.04
   Min TM value:-250 at an offset of 118.488 usec -> 2024-02-07T12:52:11.624.526.212 (XO: 29.624529236)
   Max TM value:177 at an offset of 106.250 usec -> 2024-02-07T12:52:11.624.513.975 (XO: 29.624516998)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 121.050 usec -> 2024-02-07T12:52:11.624.528.775 (XO: 29.624531798)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 106.338 usec -> 2024-02-07T12:52:11.624.514.062 (XO: 29.624517086)
