-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 17:41:55 2024 UTC
-- File 20240207_124950_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.059552 to 30.020357
-each separately from 1s to 30s and from 20068us to 59841us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.960227 in 1 packets: 18
MIU packet time from 2024-02-07T12:49:54.059.552.000 to 2024-02-07T12:50:23.020.357.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.960227 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8549573 to 31.8548576
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.9999003 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240207_124950_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9145107 to 2.9147997
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.9601270  -- OK for SL --
   MIU time from 2024-02-07T12:49:54.059.552.000 to 2024-02-07T12:49:54.059.841.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.960227

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8749267 to 31.8752157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-07T12:50:23.020.068.000 to 2024-02-07T12:50:23.020.357.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.914510700 corrected by 40bit 10MHz counter of 2.854957300 obtained at 1Hz counter of 1. Result: 1.059553602
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-07T12:49:54.059.411.725
   Max 1st sample MIU time among all SL packets: 2024-02-07T12:50:23.020.216.725
   Min 1st sample XO time among all SL packets: 1.059413327
   Max 1st sample XO time among all SL packets: 30.020217996
   TM values from -248 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-07T12:49:54.059.411.725 (XO: 1.059413327 - 1.059719328, t=35.6 degC)  Rank: 00081
   TM values: Max =   211  Min =  -248  Ave =    -4.47  Std =    19.92
   Min TM value:-248 at an offset of 111.563 usec -> 2024-02-07T12:49:54.059.523.287 (XO: 1.059524890)
   Max TM value:211 at an offset of 111.263 usec -> 2024-02-07T12:49:54.059.522.987 (XO: 1.059524590)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 114.088 usec -> 2024-02-07T12:49:54.059.525.812 (XO: 1.059527415)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 111.375 usec -> 2024-02-07T12:49:54.059.523.100 (XO: 1.059524702)

 - Block number: 2 starting at 2024-02-07T12:50:23.019.927.725 (XO: 30.019928995 - 30.020234996, t=35.6 degC)  Rank: 00080
   TM values: Max =   183  Min =  -239  Ave =    -4.47  Std =    17.50
   Min TM value:-239 at an offset of 120.313 usec -> 2024-02-07T12:50:23.020.048.037 (XO: 30.020049308)
   Max TM value:183 at an offset of 119.988 usec -> 2024-02-07T12:50:23.020.047.712 (XO: 30.020048983)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 122.850 usec -> 2024-02-07T12:50:23.020.050.575 (XO: 30.020051845)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 120.113 usec -> 2024-02-07T12:50:23.020.047.837 (XO: 30.020049108)
